- Research article
- Open Access
- Open Peer Review
This article has Open Peer Review reports available.
Effects of inaccuracies in arterial path length measurement on differences in MRI and tonometry measured pulse wave velocity
© The Author(s). 2017
Received: 14 February 2017
Accepted: 28 April 2017
Published: 10 May 2017
Carotid-femoral pulse wave velocity (cf-PWV) and aortic PWV measured using MRI (MRI-PWV) show good correlation, but with a significant and consistent bias across studies. The aim of the current study was to evaluate whether the differences between cf.-PWV and MRI-PWV can be accounted for by inaccuracies of currently used distance measurements.
One hundred fourteen study participants were recruited into one of 4 groups: Type 2 diabetes melltus (T2DM) with cardiovascular disease (CVD) (n = 23), T2DM without CVD (n = 41), CVD without T2DM (n = 25) and a control group (n = 25). All participants underwent cf.-PWV, cardiac MRI and whole body MR angiography(WB-MRA). 90 study participants also underwent aortic PWV using MRI. cf.-PWVEXT was performed using a SphygmoCor device (Atcor Medical, West Ryde, Australia). The true intra-arterial pathlength was measured using the WB-MRA and then used to recalculate the cf.-PWVEXT to give a cf.-PWVMRA.
Distance measurements were significantly lower on WB-MRA than on external tape measure (mean diff = −85.4 ± 54.0 mm,p < 0.001). MRI-PWV was significantly lower than cf.-PWVEXT (MRI-PWV = 8.1 ± 2.9 vs. cf.-PWVEXT = 10.9 ± 2.7 ms−1,p < 0.001). When cf.-PWV was recalculated using the inter-arterial distance from WB-MRA, this difference was significantly reduced but not lost (MRI-PWV = 8.1 ± 2.9 ms−1 vs. cf.-PWVMRA 9.1 ± 2.1 ms−1, mean diff = −0.96 ± 2.52 ms−1,p = 0.001). Recalculation of the PWV increased correlation with age and pulse pressure.
Differences in cf.-PWV and MRI PWV can be predominantly but not entirely explained by inaccuracies introduced by the use of simple surface measurements to represent the convoluted arterial path between the carotid and femoral arteries.
Arteriosclerosis is the process of arterial stiffening that has significant pathophysiological implications and is strongly associated with cardiovascular events . It is predominantly a product of age and pulse pressure, reflecting the effects of repetitive strain on the elastic fibers of the arterial wall . Pulse wave velocity is a well established marker of arterial stiffening with important prognostic implications, and as a result of such has now been incorporated into management guidelines for the treatment of hypertension . Carotid femoral pulse wave velocity (cf-PWV) is the most commonly used technique for the measurement of PWV, however it relies on the use of surface measures of distance to represent the convoluted path of the underlying arterial tree, introducing potential errors in PWV calculation.
Use of Magnetic Resonance Imaging (MRI) for the measurement of PWV (MRI-PWV) has become increasingly prominent due to its ability to measure directly the central aortic PWV coupled to recent advances in the technique allowing high temporal resolution imaging and reduced image acquisition time . The two techniques correlate well with each other, but with a significant and consistent bias between the two techniques across studies with cf.-PWV consistently being approximately 1.6–1.7 ms−1 higher than MRI-PWV [5, 6]. This is consistent with observations of 1.9 ms−1 difference between the aortic pulse wave velocity measured invasively and the cf-PWV, however this previous study also used a direct carotid-femoral measurement which is known to overestimate distance travelled and thus PWV by approximately 25% . Previous work has demonstrated both a clear discrepancy in the distances measured using a tape measure and the actual arterial pathlength travelled by the pulse wave due to the effects of age and obesity confounding both arterial length and body contour measurements . However the muscular and elastic properties of the aorta and its branch vessels is also known to change throughout their length which in turn affects their stiffness and by extension PWV [6, 9].
It is thus not currently clear whether the differences in PWV between the two techniques are due to inaccuracies in distance measurement in cf-PWV, the lower temporal resolution of MRI-PWV or the changing elastic properties throughout the arterial tree. The effects of this bias are also not clear. The aim of the current study was therefore to evaluate whether the differences between cf.-PWV and MRI-PWV in a mixed cohort of patients with and without diabetes mellitus and symptomatic cardiovascular disease can be accounted for by inaccuracies of currently used distance measurements and whether recalculation of cf.-PWV using accurate distance measurements results in improved detection of cardiovascular disease.
The study was a single centre observational sub-study of the multicentre SUMMIT (SUrrogate markers for Micro- and Macrovascular hard endpoints for Innovative diabetes Tools) study which was designed to analyse cardiovascular biomarkers in diabetes. Recruitment criteria, strategy and study protocol have been described in detail previously [10, 11], but in summary, subjects were recruited and categorised into 4 groups based on their history of type 2 diabetes and cardiovascular disease (CVD) as follows: Group 1: Type 2 diabetes mellitus (T2DM) with a prior clinical diagnosis of cardiovascular disease that included coronary artery disease (CAD), cerebrovascular disease and/or lower extremity arterial disease (LEAD); Group 2: Type 2 diabetes mellitus with no clinical evidence of cardiovascular disease; Group 3: Absence of diabetes mellitus with clinical evidence of CAD, cerebrovascular disease and/or LEAD; Group 4: Healthy controls, with no biochemical evidence of diabetes mellitus (see below) and no clinical evidence of cardiovascular disease. All participants had a detailed clinical history and examination performed, bloods taken for renal function, cholesterol and HbA1c, and underwent whole body magnetic resonance angiography (MRA), cardiac magnetic resonance imaging (CMR) and cf.-PWV.
Carotid-femoral pulse wave velocity
Carotid-femoral PWV was measured using a SphygmoCor device (Atcor Medical, West Ryde, Australia). A blood pressure (BP) cuff was attached to the left arm, and three electrocardiogram (ECG) electrodes (lead I) were attached. External distance measurements were performed using a tape measure, and a proximal (carotid to sternal notch) and distal (sternal notch to umbilicus and umbilicus to femoral) measure with final distance used for calculation being the proximal distance subtracted from the distal distance. After resting for 5 min, the BP was measured three times at one-minute intervals, and the mean value of the two final measurements was recorded. The carotid and femoral pulses were captured using the SphygmoCor device with cf.-PWV automatically calculated as the measured distance divided by the differences in time between the R wave of the ECG and the foot of the carotid and femoral pulse curves. For the remainder of the paper, cf.-PWV calculated using the external tape measure distance shall be referred to as cf.-PWVEXT.
Magnetic resonance imaging
Images were acquired on a 32 RF receiver channel, 3 Tesla MRI scanner (Magnetom Trio, Siemens, Erlangen, Germany). For whole body coverage, a combination of six RF coils including head matrix, neck matrix, spine matrix, two body matrix and peripheral angiography phased array RF surface coils were used. Subjects where placed supine, head first into the magnet bore. The imaging protocol was carried out in 3 phases: i) MRA of the thoracic and neck, and distal lower limbs, ii) CMR including late gadolinium enhancement (LGE) and iii) MRA of the abdomen, pelvis and proximal lower limb.
Whole body magnetic resonance angiography protocol
Whole body magnetic resonance angiography (WB-MRA) was performed using 4 overlapping data sets covering: head, neck and thorax (station 1), abdomen and pelvis (station 2), upper legs (stations 3) and lower legs (station 4). All stations were acquired using a coronal spoiled FLASH (Fast Low Angle Shot) sequence (repetition time TR = 2.6–3.47 ms; echo time TE = 0.96–1.21 ms; flip angle = 16–37°; pixel area = 1.1–1.5 mm2 and slice thickness = 1–1.4 mm, slight variation according to station and participant body habitus). .
First, station 1 and 4 pre-contrast images were acquired. Following this an injection of 10 ml of 0.5 M Gadoterate meglumine (Guerbet, Villipinte, France) followed by a 20 ml saline flush were administered at a rate of 1.5 ml/s. Acquisition of station 1 was triggered when the contrast reached the aortic arch (timed using a fluoro sequence through the aortic arch), following which three sequential acquisitions of station 4 were performed. Cardiac MRI was performed after this first injection (see next section for details) before stations 2 and 3 image acquisition began. After the cardiac acquisition was completed, pre contrast images were acquired of both stations following which post-contrast images were acquired after an injection of 15 ml gadoterate meglumine and 20 ml saline flush, both administered at a rate of 1.5 ml/s.
Cardiac magnetic resonance (CMR) protocols
Cardiac magnetic resonance (CMR) imaging utilised a spine matrix and six-element body array matrix RF coils. TurboFLASH two-chamber, four-chamber and short axis localiser, and two-chamber and four-chamber cine images were acquired. Left ventricular assessment involved the acquisition of a horizontal long axis, vertical long axis and stacked short axis cine images using a retrospectively gated TrueFISP sequence (repetition time TR = 3.4 ms; echo time TE = 1.48 ms; flip angle = 50–60°; pixel area = 1.4 mm by 1.9 mm; slice thickness = 6 mm; inter-slice gap = 4 mm). Left ventricular mass (LVM), end diastolic volume (EDV), end systolic volume (ESV), stroke volume (SV) and ejection fraction (EF) were calculated from the short axis stack.
MRI pulse wave velocity measurement
A retrospective ECG-gated gradient-echo pulse sequence with velocity encoding was applied to measure the through plane flow at two predefined locations in the ascending and abdominal aorta. The first slice was positioned axial through the aortic arch at the level of the pulmonary bifurcation, and the second slice was placed axial through the descending aorta immediately proximal to the renal arteries.
Imaging parameters included the following: echo time of 4.83 ms, repetition time of 14 ms, flip angle 20°, slice thickness of 8 mm, field of view at 350 mm, matrix size 256 × 256. The temporal resolution was optimised to ensure that 128 phases per cardiac cycle were obtained and a VENC of 150 cm/s. To determine the distance between the two aortic slices, a 2D gradient echo FLASH (fast low angle shot) was acquired of the aorta in a ‘candy stick’ double-oblique orientation. TR/TE 40/1.2 ms; flip angle 15°, slice thickness of 8 mm, 23 cardiac phases, 1 averages, a pixel size of 1.5 × 1.5 mm2, bandwidth of 475 Hz/ pix and breath hold scan time of average 9 s. Aortic PWV was calculated from the MRI images using the transit time method . Image analysis was performed by a single observer using Segment version 1.9 R4339 (http://segment.heiberg.se) , blinded to the clinical status of the subjects. The up-slope of the arriving pulse wave at each location was calculated from the flow curves. The distance was measured along the aorta between the two analyses planes using candy stick FLASH and the time delay calculated as the time delay between the arrival of the foot of the pulse wave at the ascending aorta and abdominal aorta.
Descriptive statistics were used for the analysis of the demographic and clinical features of the cohorts with data expressed as mean ± standard deviation (SD) for continuously distributed variables, and n(%) for nominal data.. Normality of distribution was tested using Shapiro Wilk test; if the test failed, where possible standard transformations such as square root, reciprocal or logarithmic transforms were used to generate a Gaussian distribution. To test the null hypothesis that samples originate from the same population, a paired sample t-test was used for comparison of distance measurements and between technique PWV measurements. Spearman correlation coefficient and Bland-Altman plots were performed to further examine the differences between carotid-femoral and MRI PWV. One-way analysis of variance (ANOVA) and Kruskal–Wallis ANOVA by ranks was used for the non-parametric data to compare demographic data and PWV measurements between the four groups. Backward stepwise linear regression was used to determine the contribution of body metrics and contours to discordances between tape measure and MRI measured arterial pathlengths. All data were analysed using SPSS statistical package (version 21.0, SPSS Inc. Chicago, Illinois). Significance was assumed when p < 0.05.
Population demographics of the participants
64 ± 7
62 ± 9
68 ± 9**
62 ± 8
1.71 ± 0.09
1.68 ± 0.09
1.66 ± 0.08
1.67 ± 0.10
106 ± 11^
104 ± 11^
98 ± 12
94 ± 14
108.9 ± 8.5^
109.8 ± 9.3^
103.8 ± 8.7
104.2 ± 8.3
30 ± 4
30 ± 5
29 ± 3
28 ± 4
Systolic BP (mmHg)
135 ± 12
135 ± 13
137 ± 17
135 ± 15
Diastolic BP (mmHg)
73 ± 8
77 ± 7
76 ± 8
78 ± 9
Total cholesterol (mmmo/L)
3.85 ± 0.89*
3.92 ± 0.78*
4.08 ± 0.67*
5.07 ± 1.0*
LDL Cholesterol (mmol/L)
1.71 ± 0.51*
1.97 ± 0.74*
2.07 ± 0.66*
2.69 ± 0.79
HDL Cholesterol (mmol/L)
1.14 ± 0.33*
1.21 ± 0.29*
1.29 ± 0.45*
1.53 ± 0.44
2.11 ± 1.01
1.64 ± 0.72
1.55 ± 0.79
1.86 ± 1.07
81 ± 21
73 ± 19
82 ± 19
71 ± 16
7.6 ± 1.3^
7.3 ± 1.3^
5.5 ± 0.2
5.7 ± 0.2
Prior cardiovascular eventa
61 ± 9**
55 ± 9
59 ± 9*
53 ± 9
71 ± 10
67 ± 12
73 ± 12
68 ± 10
25 ± 11
23 ± 8
26 ± 11
23 ± 7
65 ± 11
65 ± 8
65 ± 10
67 ± 7
Comparison of the proximal and distal measurements using an external tape measure technique and arterial centerline technique
p-value (A - B)
92 ± 15
160 ± 17
629 ± 45
611 ± 39
Distance for PWV calculation (mm)
537 ± 48
451 ± 33
Comparison of the effects of recalculation of PWV on correlation with cardiovascular risk factors
Total cholesterol (mmmo/L)
LDL Cholesterol (mmol/L)
HDL Cholesterol (mmol/L)
Comparison of PWV between the four groups calculated using tonometry with an external tape measure technique or an arterial centerline technique for the distance measurement, and separately using MRI
11.3 ± 3.0
11.2 ± 2.7
11.1 ± 2.4
10.2 ± 2.4
9.2 ± 2.3
9.2 ± 1.9
9.7 ± 2.1
8.8 ± 1.8
9.2 ± 2.3
7.6 ± 2.4
9.2 ± 3.6
6.7 ± 1.6
In the current study we have shown a significant difference between the external distance measurement used for the calculation of PWV and the actual arterial pathlength. When this distance discrepancy is corrected for using intra-arterial distances measured on MRI there is a significant improvement in the agreement between PWV measured using tonometry or MRI.
The continued difference of approximately 1 ms−1 between the two techniques after accounting for this difference in distance measurements suggests part of the differences in the PWV is due to differences in the underlying arterial stiffness between the underlying vessels. Previous work has shown the PWV to increase throughout the aorta, although even in the distal aorta where the PWV was the highest, this was still 1 ms−1 lower than cfPWV . Thus differences in the two measurements are almost certainly secondary to differences in carotid and iliac stiffness. Of these the iliac stiffness is the most likely to contribute to this difference as the carotid artery demonstrates a similar distensibility to the abdominal aorta, while the femoral arteries are significantly stiffer than both [14, 15]. The distorting effect of the iliac stiffness on this measurement has clinically significant implications, as even the corrected cfPWV did not differentiate between those with and without cardiovascular disease, whereas the aortic stiffness as measured on MRI was able to detect a significant difference in the PWV between these.
Our difference between cf.-PWV and MRI is consistent with prior studies, although is even more pronounced at 2.8 ms−1 than previous reports of 1.8 ms−1 [5, 6]. The most likely explanatory factor for this is the much higher BMI of the recruited participants in the current study compared with these two previous studies. Our findings of an 85 mm discordance between the external measurement technique and the actual arterial pathlength taken by the pulse wave is in agreement with prior work by Huybrechts et al. who found a difference in distance measurement of 81.8 mm between an arterial centreline and external tape measure distance . This previous study found a significant confounding effect from age on pathlength measurement with only 4% of the discrepancy being accounted for by BMI. This is in contradistinction to our study where weight and hip circumference accounted for 42% of the measurement differences with only 1% of differences explained by age. These differences in findings are likely due to the much narrower age range combined with the more obese nature of our study participants compared to the wide age range and relatively normal weight in the previous study, as obesity is known to result in increased tape measure distances [16, 17]. However as those who are most likely to be assessed by PWV for risk assessment are likely to be older and overweight, rather than young and slim, our findings are likely to be more clinically relevant, and add weight to the proposal that direct caliper measured distance between the carotid artery and femoral artery should be the measurement technique of choice as these are least affected by body contours . Creating a consistent technique for the accurate calculation of travelled pulse wave distance is crucial to the interpretation of these studies, and also to allow comparison between different techniques. We have shown that differences in PWV between MRI and tonometric measures can be largely accounted for by inaccuracies in these distance measurement techniques. Thus creation of suitable techniques may allow not just accurate PWV measurements but improved cross centre reproducibility and the ability to combine data from multiple techniques in systematic reviews and meta-analysis adding greater power to these. It may be possible to generate these measurements automatically from simple patient allometric measures such as sex, age, and height, and indeed several groups have tried exactly this [18, 19]. However one of these used the tape measure as the gold standard while the other used the aortic root to aortic bifurcation measurement, neither of which reflect the true path of the pulse wave as measured by carotid-femoral PWV. One option may be to use large whole body imaging studies such as the TASCFORCE and SHIP studies that would allow for the generation of an accurate aortic pathlength model based on thousands of individuals [20–22]. While this would require substantial effort and external validation, the benefits may be justified in terms of improved inter-centre and inter-individual reproducibility due to the removal of an extra step in the process of PWV and its incumbent potential for human error.
In our study we observed no significant difference between the groups using either cf.-PWV or MRI PWV. This is likely due to the relatively small sample size of our study, as using the same carotid-femoral technique in a larger population, of which the current group is a sub-study of, significant differences were observed between all groups . Our lack of correlation between PWV and left ventricular mass is interesting as it has previously been posited that the importance of arterial stiffening is due to its effects on the afterload of the ventricle resulting in an increased work load with resultant adverse remodelling . Previous studies demonstrating a correlation between the two have been performed in healthy volunteers or those with cardiovascular risk factors but no overt clinical cardiovascular disease [24, 25]. Thus PWV may have a greater impact on left ventricular mass earlier in the arteriosclerotic disease process with later left ventricular remodelling being affected by other processes such as coronary atherosclerosis, however longitudinal studies will be required to further elucidate this relationship.
There are several limitations within our current study. We used a three point technique for measuring the PWV distance externally, while current ESC guidelines would advocate the use of a direct carotid-femoral distance measurement with a correction factor of 0.8 to account for the shorter internal pathlength of the pulse wave and the simplicity of this technique . However this is largely based on a single MRI study of 98 individuals, and other studies have suggested the currently used technique provides superior correlation with invasive aortic PWV over a direct distance measurement technique . Our cf.-PWV and MRI scans were performed on different days, which is known to affect PWV measurement due to differing haemodynamic states , and may account for the lower correlation between the two techniques than has been seen in previous studies, but this does not change the fact that recalculation of PWV using an accurate distance measurement improves agreement between techniques. Finally, the MRI PWV and cf.-PWV use different techniques for determining wave arrival time which is known to affect the measurement and reproducibility,  however each technique was optimised to maximise its own intrinsic accuracy and reproducibility.
In conclusion, differences in PWV measurement between carotid-femoral PWV and MRI measured central aortic PWV can be predominantly explained by inaccuracies introduced by the use of simple surface measurements to represent the convoluted arterial path between the carotid and femoral arteries. Correction for this may in future allow more direct comparison between the techniques thus strengthening systematic reviews and meta-analyses.
This is a sub study of the multicentre SUMMIT study. The research leading to these results has received support from the Innovative Medicines Initiative Joint Undertaking under grant agreement n°115,006, resources of which are composed of financial contribution from the European Union’s Seventh Framework Programme (FP7/2007–2013) and EFPIA companies in kind contribution. JRWM is supported by the Wellcome Trust through the Scottish Translational Medicine and Therapeutics Initiative (Grant no. WT 085664) in the form of a Clinical Research Fellowship. Neither groups had any role in: study design, the collection, analysis, and interpretation of data; in the writing of the manuscript; nor in the decision to submit the manuscript for publication.
JWM, AT, DC, FK, HMC, and GH conceived the study. JWM, AT, FD, FA, SZM, and DC collected data for the study. JWM, AT, and DC analysed and interpreted the data. JWM drafted the manuscript. All authors revised the manuscript critically for important intellectual content, and read and approved the final manuscript version.
Availability of data and materials
The datasets used and/or analysed during the current study are available from the corresponding author on reasonable request.
Consent for publication
No relevant conflicts of interest to declare.
Ethical approval and consent to participate
The East of Scotland Research Ethics Committee approved the protocols. All study participants gave written informed consent to participate in this study, which was conducted in accordance with the Declaration of Helsinki.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Ben-Shlomo Y, Spears M, Boustred C, May M, Anderson SG, Benjamin EJ, et al. Aortic pulse wave velocity improves cardiovascular event prediction: an individual participant meta-analysis of prospective observational data from 17,635 subjects. J Am Coll Cardiol [Internet]. 2014;63:636–46. Available from: http://linkinghub.elsevier.com/retrieve/pii/S0735109713059743 View ArticleGoogle Scholar
- Cecelja M, Chowienczyk P. Dissociation of aortic pulse wave velocity with risk factors for cardiovascular disease other than hypertension: a systematic review. Hypertension. 2009;54:1328–36.View ArticlePubMedGoogle Scholar
- Mancia G, Fagard R, Narkiewicz K, Redon J, Zanchetti A, Böhm M, et al. ESH/ESC guidelines for the management of arterial hypertension: the task force for the management of arterial hypertension of the European Society of Hypertension (ESH) and of the European Society of Cardiology (ESC). Eur Heart J. 2013;34:2159–219.View ArticlePubMedGoogle Scholar
- Ibrahim E-SH, Johnson KR, Miller AB, Shaffer JM, White RD. Measuring aortic pulse wave velocity using high-field cardiovascular magnetic resonance: comparison of techniques. J Cardiovasc Magn Reson. 2010;12:26. Available from: http://www.pubmedcentral.nih.gov/articlerender.fcgi?artid=2874535&tool=pmcentrez&rendertype=abstract View ArticlePubMed CentralGoogle Scholar
- Parikh JD, Hollingsworth KG, Kunadian V, Blamire A, MacGowan GA. Measurement of pulse wave velocity in normal ageing: comparison of Vicorder and magnetic resonance phase contrast imaging. BMC Cardiovasc Disord. 2016;16:50. Available from: http://www.ncbi.nlm.nih.gov/pubmed/26892669 View ArticlePubMedPubMed CentralGoogle Scholar
- Hickson SS, Butlin M, Graves M, Taviani V, Avolio AP, McEniery CM, et al. The relationship of age with regional aortic stiffness and diameter. JACC Cardiovasc Imaging. 2010;3:1247–55. Available from: http://dx.doi.org/10.1016/j.jcmg.2010.09.016 View ArticlePubMedGoogle Scholar
- Van Bortel LM, Laurent S, Boutouyrie P, Chowienczyk P, Cruickshank JK, De Backer T, et al. Expert consensus document on the measurement of aortic stiffness in daily practice using carotid-femoral pulse wave velocity. J Hypertens. 2012;30:445–8.View ArticlePubMedGoogle Scholar
- Huybrechts SAM, Devos DG, Vermeersch SJ, Mahieu D, Achten E, de Backer TLM, et al. Carotid to femoral pulse wave velocity: a comparison of real travelled aortic path lengths determined by MRI and superficial measurements. J Hypertens. 2011;29:1577–82. Available from: http://www.ncbi.nlm.nih.gov/pubmed/21666491 View ArticlePubMedGoogle Scholar
- Latham RD, Westerhof N, Sipkema P, Rubal BJ, Reuderink P, Murgo JP. Regional wave travel and reflections along the human aorta: a study with six simultaneous micromanometric pressures. Circulation. 1985;72:1257–69. Available from: http://www.ncbi.nlm.nih.gov/pubmed/4064270 View ArticlePubMedGoogle Scholar
- Duce SL, Weir-McCall JR, Gandy SJ, Matthew SZ, Cassidy DB, McCormick L, et al. Cohort comparison study of cardiac disease and atherosclerotic burden in type 2 diabetic adults using whole body cardiovascular magnetic resonance imaging. Cardiovasc Diabetol BioMed Central; 2015;14:122. Available from: http://www.cardiab.com/content/14/1/122
- Shore AC, Colhoun HM, Natali A, Palombo C, Östling G, Aizawa K, et al. Measures of atherosclerotic burden are associated with clinically manifest cardiovascular disease in type 2 diabetes: a European cross-sectional study. J Intern Med, Available from: http://www.ncbinlmnihgov/pubmed/25752315. 2015;278:291–302.
- Waugh SA, Ramkumar PG, Gandy SJ, Nicholas RS, Martin P, Belch JJF, et al. Optimization of the contrast dose and injection rates in whole-body MR angiography at 3.0T. J Magn Reson Imaging [Internet]. 2009 [cited 2013 Dec 19];30:1059–67. Available from: http://www.ncbi.nlm.nih.gov/pubmed/19856438.
- Heiberg E, Sjögren J, Ugander M, Carlsson M, Engblom H, Arheden H. Design and validation of Segment--freely available software for cardiovascular image analysis. BMC Med Imaging [Internet]. 2010 [cited 2014 May 26];10:1. Available from: http://www.pubmedcentral.nih.gov/articlerender.fcgi?artid=2822815&tool=pmcentrez&rendertype=abstract.
- Cavalcante JL, Lima JAC, Redheuil A, Al-Mallah MH. Aortic stiffness: Current understanding and future directions. J Am Coll Cardiol [Internet]. Elsevier Inc.; 2011;57:1511–22. Available from: http://dx.doi.org/10.1016/j.jacc.2010.12.017.
- Van Sloten TT, Schram MT, Van Den Hurk K, Dekker JM, Nijpels G, Henry RMA, et al. Local stiffness of the carotid and femoral artery is associated with incident cardiovascular events and all-cause mortality: the hoorn study. J Am Coll Cardiol. 2014;63:1739–47.View ArticlePubMedGoogle Scholar
- Desamericq G, Tissot CM, Akakpo S, Tropeano AI, Millasseau S, Macquin-Mavier I. Carotid-femoral pulse wave velocity is not increased in obesity. Am J Hypertens. 2015;28:546–51.View ArticlePubMedGoogle Scholar
- Levi-Marpillat N, Desamericq G, Akakpo S, Affes-Ayadi H, Tropeano A-I, Millasseau S, et al. Crucial importance of using a sliding calliper to measure distance for carotid-femoral pulse wave velocity assessment. J Hypertens [Internet] 2013 [cited 2016 Sep 6];31:940–5. Available from: http://www.ncbi.nlm.nih.gov/pubmed/23337470.
- Weber T, Ammer M, Rammer M, Adji A, O’Rourke MF, Wassertheurer S, et al. Noninvasive determination of carotid-femoral pulse wave velocity depends critically on assessment of travel distance: a comparison with invasive measurement. J Hypertens 2009 [cited 2016 Jun 28];27:1624–30. Available from: http://www.ncbi.nlm.nih.gov/pubmed/19531964.
- Filipovský J, Mayer O, Dolejšová M, Seidlerová J. The assessment of carotid–femoral distance for aortic pulse wave velocity: should it be estimated from body height? Artery Res. 2010;4:19–23. Available from: http://linkinghub.elsevier.com/retrieve/pii/S1872931210000049 View ArticleGoogle Scholar
- Gandy SJ, Lambert M, Belch JJF, Cavin ID, Crowe E, Littleford R, et al. Technical assessment of whole body angiography and cardiac function within a single MRI examination. Clin Radiol. 2015;70:595–603. Available from: http://www.ncbi.nlm.nih.gov/pubmed/25791202 View ArticlePubMedPubMed CentralGoogle Scholar
- Gandy SJ, Lambert M, Belch J, Cavin I, Crowe E, Littleford R, et al. 3T MRI investigation of cardiac left ventricular structure and function in a UK population: The tayside screening for the prevention of cardiac events (TASCFORCE) study. J Magn Reson Imaging [Internet]. 2016;1–11. Available from: http://doi.wiley.com/10.1002/jmri.25267.
- Hegenscheid K, Kühn JP, Völzke H, Biffar R, Hosten N, Puls R. Whole-body magnetic resonance imaging of healthy volunteers: pilot study results from the population-based SHIP study. Rofo [Internet]. 2009 [cited 2016 Sep 8];181:748–59. Available from: http://www.ncbi.nlm.nih.gov/pubmed/19598074.
- Laurent S, Cockcroft J, Van Bortel L, Boutouyrie P, Giannattasio C, Hayoz D, et al. Expert consensus document on arterial stiffness: methodological issues and clinical applications. Eur Heart J. 2006;27:2588–605. Available from: http://www.ncbi.nlm.nih.gov/pubmed/17000623 View ArticlePubMedGoogle Scholar
- Brandts A, van Elderen SGC, Westenberg JJM, van der Grond J, van Buchem MA, Huisman MV, et al. Association of aortic arch pulse wave velocity with left ventricular mass and lacunar brain infarcts in hypertensive patients: assessment with MR imaging. Radiology. 2009;253:681–8. Available from: http://www.ncbi.nlm.nih.gov/pubmed/19864506 View ArticlePubMedGoogle Scholar
- Schillaci G, Mannarino MR, Pucci G, Pirro M, Helou J, Savarese G, et al. Age-specific relationship of aortic pulse wave velocity with left ventricular geometry and function in hypertension. Hypertension. 2007;49:317–21.View ArticlePubMedGoogle Scholar
- Bodlaj G, Berg J, Biesenbach G. Diurnal variation of pulse wave velocity assessed non-invasively by applanation tonometry in young healthy men. Yonsei Med. J. 2007. p. 665–70.Google Scholar
- Bachler M, Mayer C, Hametner B, Wassertheurer S. Increasing Stability of Real-Time Pulse Wave Velocity Estimation by Combining Established and New Approaches. 2013 8th EUROSIM Congr. Model. Simul. [Internet]. IEEE; 2013 [cited 2017 Mar 30]. p. 47–51. Available from: http://ieeexplore.ieee.org/document/7004916/.