- Research article
- Open Access
Comparison of the effect of membrane sizes and fibre arrangements of two membrane oxygenators on the inflammatory response, oxygenation and decarboxylation in a rat model of extracorporeal membrane oxygenation
BMC Cardiovascular Disorders volume 20, Article number: 294 (2020)
Extracorporeal membrane oxygenation (ECMO) has gained widespread acceptance for the treatment of critically ill patients suffering from cardiac and/or respiratory failure. Various animal models have been developed to investigate the adverse effects induced by ECMO. Different membrane oxygenators have been used with varying priming volumes and membrane surfaces (Micro-1, small animal membrane oxygenator (SAMO)).
Sixteen male Lewis rats (350–400 g) were randomly assigned to receive ECMO with Micro-1 or SAMO (n = 8, respectively). Venoarterial ECMO was established after cannulation of the femoral artery and the jugular vein. The cardiac output was measured using a left-ventricular conductance catheter. The oxygen fraction of the ECMO was set to 1.0, 0.75, 0.5 and 0.21 after a stabilisation period of 15 min. Further, arterial blood gas analyses were performed at baseline, and during the first hour every 15 min after commencing the ECMO, and subsequently every 30 min. Dilutional anaemia was calculated using haemoglobin concentration at baseline, and 15 min after the start of ECMO therapy. Moreover, inflammation was determined by measuring tumour necrosis factor alpha, interleukin-6 and -10 at baseline and every 30 min.
Animals of the Micro-1 group showed a significantly lower dilutional anaemia (ΔHaemoglobin t0 – t0.25: SAMO 6.3 [5.6–7.5] g/dl vs. Micro-1 5.6 [4.6–5.8] g/dl; p = 0.028). Further, significantly higher oxygen partial pressure was measured in the SAMO group, at an oxygen fraction of 0.75, 0.5 and 0.21 (380 [356–388] vs. 314 [263–352] mmHg, p = 0.002; 267 [249–273] mmHg vs. 197 [140–222] mmHg, p = 0.002; 87 [82–106] mmHg vs. 76 [60–79] mmHg, p = 0.021, respectively). However, no differences were found regarding the oxygen fraction of 1.0, in terms of carbon-dioxide partial pressure and cardiac output. Moreover, in the Micro-1 group tumour necrosis factor alpha was increased after 60 min and interleukin-6 after 120 min.
While the dilutional anaemia was increased after commencing the ECMO, the oxygenation was augmented in the SAMO group. The inflammatory response was elevated in the Micro-1 group.
Extracorporeal membrane oxygenation (ECMO) was first performed by Gibbon et al. in 1954, during cardiac surgery . The potential risks during cardiopulmonary bypass (CPB) with extracorporeal membrane oxygenation include cerebral stroke, inflammatory response and contact activation of the coagulation system [2, 3].Various CPB animal models have been described to investigate the adverse effects of CBP . Large animal models of CPB – in lambs, pigs and dogs – have been established, but they are limited by personal requirements and high costs of operation and handling . Thus, rodent models of CPB facilitate large sample sizes and can be conducted by a single experimenter. During CPB, cardiac arrest is often generated for cardiac surgery, and the lungs are not perfused. In contrast to CPB, the heart is still beating during ECMO, producing a continuous blood flow through the lungs, which prevents ischemia-reperfusion injuries.
The first description of a CPB model in rats was published by Popovic et al. in 1967 . Subsequently, several modified models of CPB were established . Most of them used oversized bubble or membrane oxygenators with large priming volumes . Therefore, a priming of the extracorporeal circuit with the blood of donor rats was required. Further, investigations on the inflammatory response induced by extracorporeal oxygenation were compromised by the transfusion-donor blood . In 2005 and 2006, two small rat membrane oxygenators with reduced priming volumes were developed. These oxygenators were used without the need for blood transfusion [8, 9]. The Micro-1 oxygenator (Micro-1, Kewei Rising Medical, Shenzhen, China) has a gas exchange area of 50 cm2 and a priming volume of 3.5 ml . In contrast, the small animal membrane oxygenator (SAMO, M. Humbs, Valley, Germany) consists of a three-layered hollow-fibre membrane with a gas exchange area of 500 cm2 and a priming volume of 7 ml . Both oxygenators have been used successfully in different models [8,9,10,11,12,13,14,15]. Models without a requirement for blood transfusion offer the opportunity to measure the extracorporeal oxygenation-induced inflammatory response. In addition to the priming volume, the oxygenation performance is an important parameter, which can be quantified by oxygen partial pressure (pO2).
The aim of our study was to examine the impact of different membrane sizes and fibre arrangements on inflammatory response, oxygenation and decarboxylation. Furthermore, we aimed to investigate the effect of the different priming volumes on the dilutional anaemia affecting the oxygen content (arterial saturation × haemoglobin × 1.34 + pO2 × 0.003) of the rat.
All procedures involving animals were conducted in compliance with the standards for animal care and were approved by the local committee for animal care (GI 20/26 G45/2018; Regierungspraesidium Giessen, Germany) .
Male Lewis rats (350–400 g), obtained from Janvier Labs (Le Genest St. Isle, France), were stored in conditions of 22 °C, 55% relative humidity and a day/night cycle of 14/10 h, with access to standard chow and water ad libitum. Rats were randomly divided into two groups to undergo ECMO with SAMO (n = 8) or Micro-1 (n = 8). Inclusion criteria contained body weight between 350 and 400 g. Unfortunately, two animals died during the surgical procedure and two animals had to be excluded due to severe blood loss during the surgical procedure. Therefore, four rats were replaced with additional animals.
Anaesthesia and surgery
After inhalative induction of anaesthesia (5% isoflurane balanced with 100% oxygen), rats were intubated endotracheally (16 G cannula, B. Braun, Melsungen, Germany) and ventilated volume controlled (Harvard Inspira, Harvard Apparatus, Cambridge, UK) in a weight-adjusted manner. The animals were placed on an automated heating pad and a rectal temperature probe was inserted. Monitoring included end-tidal carbon-dioxide (CO2, MicroCapStar, CWE, Ardmore, Pennsylvania, USA), continuous electrocardiogram measurements, heart rate (HR), cardiac output (CO), stroke volume (SV), left-ventricular end diastolic volume (LVEDV), left-ventricular end diastolic pressure (LVEDP), left-ventricular ejection fraction (LVEF) and arterial blood pressure (systolic, diastolic and mean). Vascular accesses were placed surgically and consisted of the lateral tail vein for infusion and anaesthesia (24 G cannula, B. Braun, Melsungen, Germany), the tail artery for measurement of the arterial blood pressure and intermittent blood gas analysis (24 G B. Braun, Melsungen, Germany), the right femoral artery for ECMO inflow (22 G catheter, Terumo, Eschborn, Germany) and the right jugular vein for ECMO outflow (modified multi-orifice 17 G cannula, B. Braun, Melsungen, Germany). Further, a 2 F pressure–volume catheter (SPR-838, Millar, Houston, TX, USA) was inserted into the left ventricle through the right carotid artery for the continuous measurement of LVEDP and LVEDV.
Extracorporeal membrane oxygenation
The ECMO circuit consisted of a venous reservoir (M. Humbs, Valley, Germany), a roller-pump (Verderflex Vantage 3000, Castleford, UK) and a membrane oxygenator. Both the SAMO and the Micro-1 oxygenator were used, depending upon the randomised groups. The oxygenators were primed with 250 IE Heparin (Ratiopharm, Ulm, Germany) and hydroxyethyl starch 6% (Voluven, Fresenius Kabi, Bad Homburg, Germany), according to the different priming volume of 7 ml (SAMO) and 3.5 ml (Micro-1).
The blood flow was initiated at a flow rate of 45 ml/kg/min, and continuously increased to 90 ml/kg/min. To measure the dilutional anaemia, arterial blood gas analyses – including haemoglobin concentration (Hb) – were performed just before and 15 min (min) after the start of ECMO therapy.
At first, ECMO commenced an oxygen fraction (FiO2) of 1.0. After a stabilization period of 15 min, the FiO2 was adjusted to 0.75, 0.5 and 0.21.
Euthanasia at end of experiments
After the end of the experiments (t2) Isoflurane was adjusted at 5% and the blood flow rate on the ECMO was slowly decreased to 45 ml/kg/min and subsequently stopped. Accordingly, the rats were euthanised by exsanguination. Therefore, the whole blood was collected through the arterial and venous ECMO cannulas. Afterwards, the heart and lungs were resected.
ELISA and blood gas analysis
Blood samples were collected at baseline (t0), every 15 min after starting the ECMO till t1 (60 min) and subsequently every 30 min till t2 (120 min). At each observation point, pO2, carbon-dioxide partial pressure (pCO2), Hb, pH, bicarbonate, base excess (BE), lactate, glucose, natrium, potassium, calcium and chloride were measured (ABL800, Radiometer, Copenhagen, Denmark). In addition, blood samples were centrifuged at 3000 g for 5 min and the plasma was stored at − 80 °C for further analysis at baseline (t0), and every consecutive 30 min.
To quantify the inflammatory response, the tumour necrosis factor alpha (TNF-α), interleukin (IL)-6 and IL-10 were measured by enzyme-linked immunosorbent assays (ELISA Kits R6000B, RTA00 and R1000 R&D System, Wiesbaden, Germany) according to manufacturer’s instructions. The probes were unfrozen only once.
The sample size calculation was performed using G*Power version 188.8.131.52 (Heinrich-Heine-Universität, Düsseldorf, Germany) and revealed a group size of nine animals with an alpha and beta error of 0.05. Further, the underlying effect size of 1.25 was calculated with Hb values from previous experiments. According to animal welfare regulations, intermittent statistical analyses were performed. Based on the regulations of the animal welfare board, the experiments were stopped due to statistically significant results after 16 animals. All data were expressed as the median, with an interquartile range (25th and 75th percentile). The Wilcoxon–Mann–Whitney test was used to compare the groups at the same time point. All statistical analyses were performed using SPSS Version 22 (IBM, Stuttgart, Germany). GraphPad Prism Version 7 was used to present the data (GraphPad Software, San Diego, CA, USA).
Blood gas analyses
The dilutional anaemia was calculated using the baseline Hb and the Hb value 15 min after commencing the ECMO. Animals in the Micro-1 group showed a significantly lower extent of dilutional anaemia (ΔHb t0 – t0.25: SAMO 6.3 [5.6–7.5] g/dl vs. Micro-1 5.6 [4.6–5.8] g/dl; p = 0.028; Fig. 1). Further measurements of the Hb revealed significant differences at t0.25 (Table 1).
Moreover, SAMO reached significantly higher pO2 values at different FiO2 (Fig. 2). After t1 the FiO2 was adjusted to 0.5 and significantly higher pO2 were measured in the SAMO group at t1.5 and t2 (Table 1).
In addition, no significant differences between both groups were measured regarding the pCO2. The pH-value was significantly higher in the SAMO group between t0.5 and t2 (Table 1). Accordingly, SAMO resulted in increased bicarbonate levels and BE at t0.75, t1, t1.5 and t2 (Table 1).
Measurements of natrium, potassium, calcium, chloride, glucose and lactate revealed no differences between the two groups (Table 1).
Measurement of inflammatory parameters
The TNF-α levels were significantly higher in the Micro-1 group at t1 (SAMO 26 [16–32] pg/ml vs. Micro-1 39 [36–56] pg/ml, p = 0.005; Fig. 3). Moreover, the serum values of IL-6 were significantly increased in the Micro-1 group at t2 (SAMO 229 [147–269] pg/ml vs. Micro-1 314 [268–378] pg/ml, p = 0.028; Fig. 3). Further analysis, at other times, for TNF-α and IL-6, as well as the measurements of IL-10 did not reveal significant differences between the two groups.
Haemodynamic values were analysed every 15 min after commencing the ECMO during a two-hour observation period. No significant differences in HR, diastolic and mean arterial blood pressure were found between the two groups (Fig. 4). The systolic blood pressure was increased in the Micro-1 group at t1.75 and t2 (t1.75 SAMO 122 [118–132] mmHg vs. Micro-1132 [129–141] mmHg, p = 0.038; t2 SAMO 122 [120–125] mmHg vs. Micro-1126 [123–130] mmHg, p = 0.038).
Furthermore, measurements of CO, SV, LVEDV, LVEDP and LVEF did not reveal significant differences between the SAMO and Micro-1 group (Fig. 4).
This study compared two different membrane oxygenators (SAMO and Micro-1) in in-vivo rat models of ECMO. Hemodynamic measurements, oxygenation, haemodilution and ECMO-induced inflammation were investigated.
In the past, both oxygenators have been used in different rat models, including models of CPB and deep hypothermic cardiac arrest (DHCA) [8, 9, 12,13,14,15, 17]. To the best of our knowledge, they have not been compared regarding their oxygenation, decarboxylation, dilutional anaemia and inflammation response to date.
The haemoglobin concentration just after commencing the ECMO was significantly lower in the SAMO group, reflecting the different priming volumes of both oxygenators (SAMO 11 ml vs. Micro-1 9 ml). Moreover, the pO2 was significantly increased in the SAMO group, which could be caused by the larger membrane surface (SAMO 500 cm2 vs. Micro-1 50 cm2). After two-hours of ECMO therapy, the pO2 was significantly reduced in the Micro-1 group, suggesting the SAMO is more long-term stable. However, there was no difference in decarboxylation between both groups. Magnet et al. used the SAMO in a rat model of cardiac arrest. After 15 min of ECMO with a blood flow of 100 ml/kg/min and a FiO2 of 1.0, they measured a pO2 of 443 mmHg, which is similar to that in our findings . The Micro-1 oxygenator was validated in a rat model of CPB by Dong et al. . A pO2 of 286 ± 21 mmHg was measured with a blood flow of 100–150 ml/kg/min and a FiO2 of 1.0, which was lower than that found in our measurements (Micro-1 pO2 408 [394–475] mmHg). While the ventilation was continued during ECMO, Dong et al. stopped the mechanical ventilation of the lungs during CPB, which might be the cause of this difference.
The extracorporeal circuit, which consists of a venous reservoir, a roller pump, a membrane oxygenator and several tubes, must be primed. Further, the priming volume varies between 8 and 15 ml reflecting the different oxygenators, reservoirs and tubes [10, 11, 15]. For this reason, dilutional anaemia occurs after commencing the extracorporeal circulation. The reported haematocrit values during extracorporeal circulation vary between 22 and 32% [12, 13, 17]. In our model, haematocrits of 21% (SAMO) and 24% (Micro-1) were measured. Moreover, the blood volume of the rat also affects the dilutional anaemia and can be calculated by the following formula: blood volume (ml) = bodyweight (g) × 0.06 + 0,77 . The body weight of the rats used in literature on the topic vary between 300 g and 500 g, resulting in blood volumes between 19 ml and 31 ml [13, 15]. We used rats weighing 350–400 g to prevent heart failure, a phenomenon that also occurs more frequently in old animals.
Extracorporeal circulation causes systemic inflammation. Mechanical cell damage and contact with the extracorporeal circuit, especially with the membrane of the oxygenator, are assumed to be the cause of this . The systemic inflammatory response syndrome like cascade includes the intrinsic and extrinsic coagulation pathway, complement system and endothelia cells, leukocytes, platelets and cytokines . Other factors influencing the inflammatory response include the health of the animals and the surgical trauma induced by vascular access. Cannulation with an open chest procedure leads to an aggravation of the inflammatory response. Therefore, we chose a vascular access procedure without sternotomy.
Interestingly, we observed higher levels of the proinflammatory cytokines TNF-α and IL-6 in the Micro-1 group, although the membrane surface is smaller than that of the SAMO group (SAMO 500 cm2 vs. Micro-1 50 cm2). The Micro-1 oxygenator was gas sterilized prior to use and applied only once. In contrast, the SAMO consists of a plexiglass chassis, which was reused and cleaned with an enzyme cleaner (Helizyme, B. Braun, Melsungen, Germany), and a three-layer hollow-fibre membrane – applied only once – was also gas sterilized prior to use.
During CPB, the heart and the lungs are not regularly perfused, resulting in an ischemia-reperfusion injury . Therefore, our approach was to compare the SAMO and Micro-1 groups during ECMO with continuous ventilation and blood flow through heart and lungs to avoid ischemia and reperfusion. Hence, our results cannot be applied to other rat models using the SAMO or Micro-1 oxygenator during CPB or DHCA.
Furthermore, it is important to note the limitations of our work: first, the surgical vascular access results in a distal ischemia caused by vascular ligation. Furthermore, the commencement of the ECMO caused a dilutional anaemia with low haemoglobin concentrations. No elevation in lactate was measured, suggesting that the oxygen delivery was greater than the oxygen consumption. Second, the observation period during ECMO was only 2 h. Last, despite the sample size calculation, the study cohort of eight animals per group remains small.
Besides, shorter tubes could have led to reduced priming volumes with less dilutional anaemia. Further experiments should use SAMO or Micro-1 oxygenators regarding the underlying scientific question.
Herein, we compared the SAMO and Micro-1 oxygenators in terms of oxygenation, decarboxylation, dilutional anaemia and inflammation response. While the Hb concentration was impaired after commencing the ECMO, the oxygenation was increased in the SAMO group. Further, the proinflammatory cytokines – TNF-α and IL-6 – were elevated in the Micro-1 group.
In summary, the Micro-1 oxygenator induced a higher extent of inflammatory response, had a lower oxygenation capacity and showed less dilutional anaemia. Contrarily, the use of SAMO resulted in a weaker inflammatory reaction, higher oxygenation capacity and increased dilutional anaemia. According to our findings, further studies on ECMO-induced inflammation should be performed using the Micro-1 oxygenator, while oxygenation studies should be investigated with the SAMO.
Availability of data and materials
The datasets used and analysed during the current study are available from the corresponding author upon reasonable request.
Deep hypothermic cardiac arrest
Extracorporeal membrane oxygenation
- FiO2 :
Left ventricular end diastolic pressure
Left ventricular end diastolic volume
Left ventricular ejection fraction
Mean arterial pressure
- pCO2 :
Carbon-dioxide partial pressure
- pO2 :
Oxygen partial pressure
Small animal membrane oxygenator
Tumor necrosis factor alpha
Gibbon JH. Application of a mechanical heart and lung apparatus to cardiac surgery. Minn Med. 1954;37:171–85 passim.
Millar JE, Fanning JP, McDonald CI, McAuley DF, Fraser JF. The inflammatory response to extracorporeal membrane oxygenation (ECMO): a review of the pathophysiology. Crit Care. 2016;20:387. https://doi.org/10.1186/s13054-016-1570-4.
Roach GW, Kanchuger M, Mangano CM, Newman M, Nussmeier N, Wolman R, et al. Adverse cerebral outcomes after coronary bypass surgery. Multicenter study of Perioperative Ischemia Research Group and the Ischemia Research and Education Foundation Investigators. N Engl J Med. 1996;335:1857–63. https://doi.org/10.1056/NEJM199612193352501.
Jungwirth B, de Lange F. Animal models of cardiopulmonary bypass: development, applications, and impact. Semin Cardiothorac Vasc Anesth. 2010;14:136–40. https://doi.org/10.1177/1089253210370491.
Popovic P, Horecky J, Popovic VP. Instrumental responses in rats after hypothermic cardiopulmonary by-pass. Proc Soc Exp Biol Med. 1967;126:225–8. https://doi.org/10.3181/00379727-126-32408.
Ballaux PK, Gourlay T, Ratnatunga CP, Taylor KM. A literature review of cardiopulmonary bypass models for rats. Perfusion. 1999;14:411–7. https://doi.org/10.1177/026765919901400603.
Remy KE, Hall MW, Cholette J, Juffermans NP, Nicol K, Doctor A, et al. Mechanisms of red blood cell transfusion-related immunomodulation. Transfusion. 2018;58:804–15. https://doi.org/10.1111/trf.14488.
Dong G-H, Xu B, Wang C-T, Qian J-J, Liu H, Huang G, Jing H. A rat model of cardiopulmonary bypass with excellent survival. J Surg Res. 2005;123:171–5. https://doi.org/10.1016/j.jss.2004.08.007.
Jungwirth B, Mackensen GB, Blobner M, Neff F, Reichart B, Kochs EF, Nollert G. Neurologic outcome after cardiopulmonary bypass with deep hypothermic circulatory arrest in rats: description of a new model. J Thorac Cardiovasc Surg. 2006;131:805–12. https://doi.org/10.1016/j.jtcvs.2005.11.017.
Samarska IV, Henning RH, Buikema H, Bouma HR, Houwertjes MC, Mungroop H, et al. Troubleshooting the rat model of cardiopulmonary bypass: effects of avoiding blood transfusion on long-term survival, inflammation and organ damage. J Pharmacol Toxicol Methods. 2013;67:82–90. https://doi.org/10.1016/j.vascn.2013.01.002.
Peterss S, Guenther S, Kellermann K, Jungwirth B, Lichtinghagen R, Haverich A, et al. An experimental model of myocardial infarction and controlled reperfusion using a miniaturized cardiopulmonary bypass in rats. Interact Cardiovasc Thorac Surg. 2014;19:561–6. https://doi.org/10.1093/icvts/ivu187.
Koning NJ, de Lange F, Vonk ABA, Ahmed Y, van den Brom CE, Bogaards S, et al. Impaired microcirculatory perfusion in a rat model of cardiopulmonary bypass: the role of hemodilution. Am J Physiol Heart Circ Physiol. 2016;310:H550–8. https://doi.org/10.1152/ajpheart.00913.2015.
Magnet IAM, Ettl F, Schober A, Warenits A-M, Grassmann D, Wagner M, et al. Extracorporeal life support increases survival after prolonged ventricular fibrillation cardiac arrest in the rat. Shock. 2017;48:674–80. https://doi.org/10.1097/SHK.0000000000000909.
Sun Y, Song D, Wang M, Chen K, Zhang T. α7 nicotinic acetylcholine receptor agonist attenuates the cerebral injury in a rat model of cardiopulmonary bypass by activating the Akt/GSK3β pathway. Mol Med Rep. 2017;16:7979–86. https://doi.org/10.3892/mmr.2017.7600.
Bianchini EP, Sebestyen A, Abache T, Bourti Y, Fontayne A, Richard V, et al. Inactivated antithombin as anticoagulant reversal in a rat model of cardiopulmonary bypass: a potent and potentially safer alternative to protamine. Br J Haematol. 2018;180:715–20.https://doi.org/10.1111/bjh.15091.
National Academies Press (US). Guide for the care and use of laboratory animals. 8th ed. Washington: National Research Council (US) Committee for the Update of the Guide for the Care and Use of Laboratory Animals; 2011.
Zhou J, Zhou N, Wu X-N, Cao H-J, Sun Y-J, Zhang T-Z, et al. Role of the Toll-like receptor 3 signaling pathway in the neuroprotective effect of sevoflurane pre-conditioning during cardiopulmonary bypass in rats. Mol Med Rep. 2015;12:7859–68.https://doi.org/10.3892/mmr.2015.4420.
Lee HB, Blaufox MD. Blood volume in the rat. J Nucl Med. 1985;26:72–6.
Huffmyer JL, Groves DS. Pulmonary complications of cardiopulmonary bypass. Best Pract Res Clin Anaesthesiol. 2015;29:163–75. https://doi.org/10.1016/j.bpa.2015.04.002.
We thank Ilona Magel for her support in the laboratory and Ingo Lotz for taking care of the animals.
ES received grants from the von Behring Röntgen foundation (67–0007), which were used to purchase the rats, membrane oxygenators, ELISAs and conductance catheter. The von Behring Röntgen foundation had no influence on planning, implementation and evaluation of the study.
Ethics approval and consent to participate
All procedures involving animals were conducted in compliance with standards for animal care and were approved by the local committee for animal care (animal welfare commission at regional council Giessen - GI 20/26 G45/2018; Regierungspraesidium Giessen, Germany).
Consent for publication
The authors have no conflicts of interest relevant to this article.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Edinger, F., Schneck, E., Schulte, C. et al. Comparison of the effect of membrane sizes and fibre arrangements of two membrane oxygenators on the inflammatory response, oxygenation and decarboxylation in a rat model of extracorporeal membrane oxygenation. BMC Cardiovasc Disord 20, 294 (2020). https://doi.org/10.1186/s12872-020-01581-3
- Membrane oxygenator