Research article | Open | Open Peer Review | Published:
Cerebrovascular events, bleeding complications and device related thrombi in atrial fibrillation patients with chronic kidney disease and left atrial appendage closure with the WATCHMAN™ device
BMC Cardiovascular Disordersvolume 19, Article number: 112 (2019)
Impaired renal function increases the bleeding risk, leading to a conservative prescription and frequent discontinuation of oral anticoagulation in atrial fibrillation patients with chronic kidney disease (CKD). Interventional left atrial appendage closure (LAAC) might be an alternative therapeutic strategy for these patients.
We aimed to prospectively assess cerebrovascular (CE) and bleeding events, as well as peri-procedural and long-term complications in a cohort of consecutive patients undergoing interventional LAAC using the WATCHMAN™ device, with focus on CKD patients.
One hundred and eighty-nine consecutive patients undergoing interventional LAAC were included in this analysis; 171 (90.5%) patients had a reduced estimated glomerular filtration rate (eGFR; patients for each CKD stage: II = 66; IIIa = 32; IIIb = 43; IV = 18; V = 12). During a follow-up of 310 patient years three (1.0%) patients suffered a CE (two strokes, one transitory ischemic attack) and five (1.6%) other ones a bleeding complication. The observed stroke rate was more than two-thirds and the bleeding risk more than half lower than expected. Device related thrombi (DRT) were detected in twelve (6.5%) patients; women had significantly more DRT than men (12.5% vs. 2.6%; p = 0.009). Patients with an eGFR< 30 ml/min/1.73m2 showed a trend to a higher DRT rate as compared to the opposite group (13.3% vs. 5.1%; p = 0.10). Thrombus resolved with temporary oral anticoagulation therapy in ten patients without sequelae; thrombus consolidation was confirmed by serial TEE controls in the remaining two patients.
Atrial fibrillation patients with CKD have low CE and bleeding rates after LAAC with the WATCHMAN™ device. DRT risk is higher in female and patients with severe CKD. More frequent post-interventional TEE controls might be justified for early DRT detection and safe management of patients at high DRT risk.
(German Clinical Trials Register ID: DRKS00 010768; Registration Date 07.07.2016).
Non-valvular atrial fibrillation (NVAF) is the most common cardiac arrhythmia, which is associated with an increased morbidity and mortality, mainly due to a higher risk of stroke . The pathomechanism of the increased stroke risk in NVAF patients is thought to be the embolism of intracardiac thrombi, which are far more frequent in these patients and mostly found in the left atrial appendage (LAA) . A major corner stone in the management of NVAF patients at moderate or high risk of thromboembolic complications is the preventive oral anticoagulation (OAC) therapy tailored on a risk to benefit relation, which can both be estimated by appropriate risk scores (e.g. CHA2DS2Vasc and HAS-BLED scores) [3, 4]. Over the past decades vitamin K antagonists (VKA) were used as the gold standard medication for this purpose, resulting in a significant reduction of the stroke risk and mortality (by approximately two thirds and one fourth to one third, respectively) . The medication with the relatively new non-vitamin-K antagonist oral anticoagulants (NOACs) showed a more favorable profile with a further reduction of stroke risk, as compared to VKA [6,7,8,9,10]. Considering the actual data, the current guidelines recommend a preventive oral anticoagulation therapy (preferably NOACs) in patients with NVAF and a CHA2DS2-VASc-Score of ≥1 (≥ 2 in women) . However, several shortcomings of this therapy have been reported; at the time of VKAs being the only treatment option, just about half of the NVAF patients eligible for OAC therapy did receive a VKA  and in those treated with a VKA, the time within the standard target therapeutic range (INR 2.0–3.0) was just about two thirds of the total treatment time despite frequent monitoring . Undertreatment remains an important issue in the NOAC’s era, as well, with a drug discontinuation rate of up to 34%, even in the setting of large randomized trials [6,7,8,9]. Furthermore, NOAC therapy is not recommended in stage V chronic kidney disease (CKD) patients and the benefit of VKA-therapy has not been unambiguously demonstrated in that cohort . Since NVAF patients with CKD are at increased risk of both stroke and bleeding complications [15,16,17,18,19], the net clinical benefit of the OAC therapy in these patients needs to be carefully evaluated. LAA closure is an alternative therapeutic strategy to a lifelong anticoagulation in NVAF patients with moderate or high thromboembolic risk and a high bleeding risk or bleeding history [20,21,22,23], but data on short- and long-term efficacy or safety of interventional LAA-closure in patients with CKD are scarce.
We aimed to prospectively assess thromboembolic and bleeding events as well as periprocedural and long-term complications in a cohort of consecutive atrial fibrillation patients undergoing interventional closure of the LAA using the WATCHMAN™ device with focus on CKD patients.
Consecutive NVAF patients with clinical indication and suitable anatomy for interventional LAA closure, as assessed by TEE prior to the implantation procedure, were included in this study after giving their written informed consent; Fig. 1 shows a flowchart of patient inclusion, management and follow up. A diagnostic evaluation including inquiry of medical and drug history, physical examination and routine blood tests was accomplished before the implantation procedure. Effective LAA closure (defined as no or residual leakage < 5 mm between the occluder device and the LAA wall), potential device dislocation or embolization and device related thrombi (DRT) were assessed by transesophageal echocardiography (TEE) at the index procedure and the scheduled controls after six weeks and six months. Clinical safety and efficacy data after successful LAA occlusion were collected during the in-hospital treatment, at the above-mentioned outpatient visits and by phone contact prior to the final data analysis. Clinical efficacy data included rates of cerebrovascular events or peripheral embolism, whereas safety data included periprocedural complications, potential device dislocation or embolization, DRT and bleeding events; intracranial bleeding or those requiring transfusion were classified as major ones. In cases with thromboembolic or bleeding complications reports from the respective treating center were collected.
Renal function of the study subjects was classified according to the Kidney Disease Improving Global Outcomes (KDIGO) 2017 clinical practice guideline update . The estimated glomerular filtration (eGFR) rate was calculated using the Chronic Kidney Disease Epidemiology Collaboration (CKD-EPI) formula at the time of hospital admission for device implantation . The study was approved by the local Ethics Committee.
WATCHMAN™ device and implantation procedure
Interventional LAA closure using a 3rd (or 4th) generation WATCHMAN™ (WATCHMAN FLX™) device (Boston Scientific, Marlborough, USA) was performed under TEE and fluoroscopic guidance in our catheterization lab. We have described the device and implantation procedure previously in detail . Shortly, the size and morphology of the LAA were assessed by TEE standard views and fluoroscopy using a pigtail diagnostic catheter for contrast media injection after gaining transseptal access. Device size was selected upon manufacturer recommendation on the compression ratio. The device was deployed only if a residual leakage was absent or < 5 mm. In patients with a special LAA anatomy with proximal arborization into two or more dominant lobes a two-device implantation strategy was considered.
The post-implantation dual antiplatelet therapy with clopidogrel and aspirin was limited to aspirin monotherapy, after DRT exclusion at the last scheduled TEE control (after six-months).
Performance of the occluder device regarding effective LAA closure (= residual leakage < 5 mm) and potential complications such as device dislocation, DRT formation, pericardial effusion, presence and severity of mitral valve regurgitation were assessed by transesophageal echocardiography at the index procedure and scheduled controls after six weeks and six months. Furthermore, maximum and minimal device compression (DCmax/min) and compression rate (CRmax/min) as assessed by TEE measurements at the index procedure were investigated as potential predictive parameters of residual leakage. DCmax was defined as the ratio between the nominal device size and minimal device diameter, whereas DCmin as the ratio between the nominal device size and maximal device diameter as measured by TEE after deployment. Similarly, CRmax was defined as the ratio between nominal device size and minimal LAA ostium diameter, whereas CRmin as the ratio between nominal device size and maximal LAA ostium diameter as measured by TEE in the standard views (45°, 60°, 90° and 135°) before device deployment.
All patients included in the registry were considered for analysis of the data. Categorical parameters are presented as counts and percentages, whereas continuous variables as mean values ± standard deviation (SD). Categorical variables were compared by the chi-square test. Continuous variables were compared using the matched-pairs t-test after controlling for normal distribution by the Kolmogorov-Smirnov test. Estimated annual stroke or bleeding risks according to the CHA2DS2Vasc and HAS-BLED scores, respectively, are based on data from Lip and colleagues [3, 4, 27]. Statistical calculations were performed using the dedicated software SPSS Version 18 (IBM, Armonk, NY, USA).
One hundred ninety-eight consecutive patients, who were assigned to interventional LAA closure in our catheterization laboratory, were included in this registry. Interventional LAA closure was not attempted in nine patients because of unsuitable anatomy (n = 6; usually very small LAA dimensions) or detected thrombus within the LAA (n = 3), as assessed by TEE prior to the procedure. In three more patients the procedure was aborted because of ineffective LAA closure (= residual leakage ≥5 mm). Consequently, successful device implantation with confirmation of effective LAA closure occurred in 186 (98.4%) patients.
The mean patient age was 73.4 ± 7.9 years (61.4% men). The indication for interventional LAA closure was previous bleeding event(s) in 108 (57.1%), increased bleeding risk (as assessed by a HAS-BLED score ≥ 3) in 58 (30.7%), patient preference in 6 (3.2%) and contraindication to oral anticoagulation therapy in 17 (9.0%) patients. From the 108 patients with a bleeding history, 61 (56.5%) patients had suffered a gastrointestinal bleeding, 12 (11.1%) patients an intracranial bleeding and 35 (32.4%) patients other bleeding complications such as hematuria, hemoptysis, intraocular bleeding etc. Mean CHA2DS2−Vasc and HAS-BLED score were 3.88 ± 1.6 and 3.35 ± 1.0, respectively. Moderate or severe impairment of renal function with an eGFR of less than 45 ml/min/1.73m2 was present in 73 (38.6%) patients (number of patients for each CKD stage: II = 66; IIIa = 32; IIIb = 43; IV = 18 and V = 12). The baseline characteristics of the analyzed patient population as well as of the subgroups with eGFR < 45 vs. ≥ 45 ml/min/1.73m2 are shown in Table 1. Patients with moderate or severe eGFR reduction (CKD IIIb or worse) suffered more often from diabetes mellitus (61.6% vs. 34.5%, p < 0.01), had a higher CHA2DS2−Vasc-score (4.50 ± 1.42 vs. 3.40 ± 1.57; p < 0.01) and a higher HAS-BLED score (3.65 ± 1.0 vs. 3.13 ± 0.9; p < 0.01) compared to the opposite group (CKD IIIa or better), see Table 1 for details.
During a follow-up of 310 (1.6 ± 1.0) patient years three (1.0%) patients (all three with eGFR < 45 ml/min/1.73m2) suffered thromboembolic events including two ischemic strokes (one perioperative to a transcutaneous aortic valve replacement) and one transitory ischemic attack, see Table 2 for details. The estimated annual rate for cerebrovascular events according to a CHA2DS2Vasc score between 3 and 4 (3.88 in the analyzed population) ranged from 3.2 to 4.0%. The CHA2DS2Vasc-Score was higher in patients with reported thromboembolic events, as compared to those without (5.3 ± 0.5 vs. 3.7 ± 1.7, p = n.s.). Bleeding complications were observed in five (1.6%) patients, all with moderate or severe impairment of the renal function (eGFR < 45 ml/min/1.73m2); one patient suffered a gastrointestinal bleeding two days after a colonoscopy with polypectomy while being on aspirin monotherapy; in two other patients macrohematuria occurred, one patient developed a peri-renal capsule hematoma and another one post-interventional hemoptysis (all four patients were receiving a dual antiplatelet therapy). The estimated bleeding risk for anticoagulated atrial fibrillation patients with a HAS-BLED score between 3 and 4 (3.35 ± 1.0 in the analyzed population) ranged from 3.7 to 8.7%. Figure 2 shows the relative difference between the estimated and observed rates of cerebrovascular events and bleeding complications in our study population.
During follow up, six patients died in the subgroup with CKD IIIb or worse (cardiovascular death was diagnosed in four patients) and two other patients in the subgroup with CKD IIIa or better (both with cardiovascular death), see Table 2 for details.
After successful LAA closure DRT were detected in twelve (6.5%) patients (in eleven patients at the first scheduled TEE). Women had significantly more DRT than men (12.5% vs. 2.6%, p = 0.009). EGFR in the subgroup of patients with DRT was numerically lower as compared to those without (46.5 vs. 58.3 ml/min/1.73 m2, p = 0.18). On the other hand, patients with severe impairment of the renal function (eGFR < 30 ml/min/1.73m2) showed a clear trend to more likely develop DRT (13.3% vs. 5.1%, p = 0.10), as compared to the opposite group, Fig. 3 a and b. All but one patient were on dual antiplatelet therapy at the time point of DRT diagnosis; all DRT patients received a temporary anticoagulation therapy until thrombus resolution or consolidation was confirmed by TEE (at least 6 weeks). Thrombus resolved in ten patients without sequelae, whereas thrombus consolidation (endothelialization) was confirmed in the remaining two patients by serial TEE controls. In these two patients a thromboembolic event was considered very unlikely. Those patients were switched to aspirin monotherapy and showed no change of thrombus size in serial TEE controls. No cardioembolic events were reported in the subgroup of patients with documented DRT.
A residual peri-device leakage was detected by TEE in 61 (32.8%) patients immediately after release of the occluder device, in 50 (26.9%) patients after six weeks and in 39 (21.0%) patients after six months. In all patients residual leakage was < 5 mm and decreased from 2.9 ± 1.1 mm immediately after implantation to 2.5 ± 0.8 mm after six weeks and 2.2 ± 0.9 mm after six months. Patients with detected residual leakage at the index procedure had a significantly lower compression rate (CR), as defined above (see Table 3 for details), while device compression (DC) seemed to have no influence on residual leakage. With decreased number of patients with residual leakage after six weeks and six months, the CR differences between them and the opposite group were no longer significant.
Periprocedural complications occurred in four (2.1%) patients. A temporarily ST-segment elevation on the monitoring electrocardiogram (ECG) presumed air embolism into the coronary arteries in two patients. In both cases immediate angiography revealed no air or obstruction in the coronary arteries and ST-segment elevation resolved spontaneously within a few minutes. In two more patients vascular complications occurred, including a groin hematoma and an artery-venous fistula; the latter had to be treated surgically. No pericardial effusion, device embolization or cerebrovascular events were observed during the implantation procedures.
In this consecutive series of atrial fibrillation patients at moderate or high thromboembolic risk predominated by chronic kidney disease patients, we prospectively demonstrated low rates of cerebrovascular and bleeding events in a long-term follow up after successful LAA closure with the WATCHMAN™ device. Furthermore, the results of our study confirm low complication rates after interventional LAA closure with the WATCHMAN™ device and add important information to the limited data on atrial fibrillation patients with CKD undergoing this procedure.
According to the data of Lip and colleagues, the annual risk of stroke in atrial fibrillation patients increases with the CHA2DS2Vasc score, whereas the bleeding risk in anticoagulated atrial fibrillation patients increases with the HAS-BLED score [3, 4, 27]. The observed stroke rate in our study is comparable to that reported in the interventional groups of the randomized trials  and the same rate in the large series of patients after WATCHMAN™ device implantation included in the EWOLUTION registry . Considering an estimated annual stroke risk between 3.2 and 4.0% based on the CHA2DS2Vasc score in our study population, the observed stroke rate was 69 to 75% lower than expected. The observed reduction in stroke rate in our study is comparable to that reported for VKA use in atrial fibrillation patients , which confirms the efficacy of interventional LAA closure with the WATCHMAN™ device and the non-inferiority of this therapy when compared to VKA use as shown by the randomized trials [20,21,22]. At the same time, we observed a 56 to 82% reduction in the estimated bleeding rate (the latter calculated on the HAS-BLED score in our study population) [4, 27], which represents the net clinical benefit of interventional LAA closure over VKA use for stroke prevention in atrial fibrillation patients.
The confirmation of the efficacy and safety of LAA closure with the WATCHMAN™ device in a cohort of atrial fibrillation patients, predominated by those with impaired kidney function, should facilitate the decision on the appropriate individual stroke prevention therapy in these patients. Furthermore, our results support the interventional LAA closure in atrial fibrillation patients with absolute contraindication or refusal to oral anticoagulation therapy for stroke prevention. Previous studies reported controversial results regarding stroke risk reduction by VKA therapy in patients with severe CKD [17, 30, 31], while they unanimously reported significantly higher risk of major bleeding. Data on the benefit of NOAC use in these patients are even more limited, though that might be subject to actual investigation. Therefore, the amount of atrial fibrillation patients receiving no anticoagulation therapy for stroke prevention is considered to be higher among those with severe CKD. The results of our study should encourage the interventional LAA closure as an effective and safe alternative stroke prevention therapy in atrial fibrillation patients with severe CKD.
A larger operators’ experience with the device and a thorough preselection of the patients considering the LAA anatomy, as assessed by TEE prior to the implantation procedure, led to a high implantation success rate (98.4%), which was higher than that reported in the randomized trials [20, 32] and comparable to the recently published results on implantation success rate of the WATCHMAN™ device in the large multicentric EWOLUTION registry . Previously, Holmes et al. showed that increasing operator experience is not only related to a higher implantation success but also to lower rates of procedure related serious adverse events [21, 32]. In our study, a low periprocedural complication rate was observed, which is comparable to the reported recent experience with the WATCHMAN™ device . In previously published trials the majority of periprocedural adverse events could be accounted to accidental air embolism and pericardial effusion [20, 21, 29]. While a hemorrhagic pericardial effusion could be avoided in all patients in our study by using TEE guidance for transseptal puncture and with operators’ experience, accidental air embolism in the right coronary artery was observed in two patients despite accurate flushing of the delivery system and transseptal sheath. Both patients had a low left atrial pressure (though 500 ml saline were infused prior to transseptal puncture), which was intermittently negative because of a distinct negative intrathoracic pressure while snoring during deep sedation; the latter was necessary for tolerance of the TEE probe. The use of intracardiac echocardiography for guidance of transseptal puncture and device placement could avoid deep sedation and might be an advantageous alternative to TEE in these patients, especially in those with known sleep apnea.
Concerns are raised by a higher risk for DRT formation in female patients and those with severe CKD [33, 34]. In our study DRT rate was 4.8 times higher in female patients and 2.6 times higher in those with a CKD stage IV or V, as compared to the opposite groups. These findings confirm the importance of systematic TEE controls after the implantation procedure for early DRT detection and safe management of the patients and might justify a more intensive schedule of TEE controls in patients at high risk for DRT formation, such as female patients with severe CKD. Furthermore, a temporarily oral VKA therapy as used in the randomized trials (if not contraindicated) might lower the risk of DRT formation in the pre-endothelialization phase, which in CKD patients is presumably prolonged. These considerations as well as a potential clopidogrel resistance, which might play an important role in DRT formation as previously reported by Ketterer and colleagues , should be subject to further investigation.
Residual leakage between the occluder device and the LAA wall is a common finding immediately after the implantation procedure of a WATCHMAN™ device . However, in most cases a residual leakage diminishes with complete endothelialization of the device and according to actual data residual leaks < 5 mm are not associated with increased rates of thromboembolic complications . In our study, maximum and minimal compression rates as defined above were significantly lower in patients with residual peri-device leaks. In this context, choosing the correct implantation strategy and appropriate device size in relation to the LAA ostium morphology and diameter should avoid large residual peri-device leaks. A special LAA anatomy with a proximal arborization into two or more lobes might necessitate the implantation of more than one WATCHMAN™ device, whereas in patients with a residual leak of ≥5 mm percutaneous closure using a vascular plug device might be individually considered .
This is not a randomized trial and therefore prone to related bias. Furthermore, the proportion of patients with an eGFR below 60 and 30 ml/min/1.73m2 were only 55.6 and 38.6%, respectively. In this study of mostly patients with high bleeding risk or bleeding history we restricted the antithrombotic medication to antiplatelet therapy alone avoiding the initial recommendation for anticoagulation therapy by the manufacturer, except for temporarily anticoagulation in patients with observed DRT at TEE controls, which might have influenced the DRT rates. Collection of clinical data by phone-contact at the last follow-up is related to subjective or mnestic bias which might have influenced the results of our study.
Atrial fibrillation patients with CKD and interventional LAA closure using the WATCHMAN™ device have low rates of thromboembolic cerebrovascular events or bleeding complications. The risk of DRT formation is higher in female and patients with severe CKD. A temporarily oral anticoagulation therapy (if not contraindicated) and/or shorter time intervals for TEE controls after the implantation procedure might be justified for prevention, early DRT detection and safe management of these patients. A low compression rate, defined as the ratio between the nominal device size and LAA ostium diameter, is predictive for the presence of residual peri-device leaks immediately after the implantation procedure.
Chronic kidney disease
Chronic kidney disease epidemiology collaboration
- CRmax :
Maximum compression rate
- CRmin :
Minimal compression rate
- DCmax :
Maximum device compression
- DCmin :
Minimal device compression
Device related thrombi
Estimated glomerular filtration rate
Kidney disease improving global outcomes
Left atrial appendage
Non-vitamin-K antagonist oral anticoagulant
Non-valvular atrial fibrillation
Vitamin K antagonists
Andersson T, Magnuson A, Bryngelsson IL, Frøbert O, Henriksson KM, Edvardsson N, Poçi D. All-cause mortality in 272,186 patients hospitalized with incident atrial fibrillation 1995-2008: a Swedish nationwide long-term case-control study. Eur Heart J. 2013;34:1061–7. https://doi.org/10.1093/eurheartj/ehs469. Epub 2013 Jan 14.
Wang F, Zhu M, Wang X, Zhang W, Su Y, Lu Y, Pan X, Gao D, Zhang X, Chen W, Xu Y, Sun Y, Xu D. Predictive value of left atrial appendage lobes on left atrial thrombus or spontaneous echo contrast in patients with non-valvular atrial fibrillation. BMC Cardiovasc Disord. 2018;18:153. https://doi.org/10.1186/s12872-018-0889-y.
Lip GYH, Nieuwlaat R, Pisters R, Lane DA, Crijns HJGM. Refining clinical risk stratification for predicting stroke and thromboembolism in atrial fibrillation using a novel risk factor-based approach: the euro heart survey on atrial fibrillation. Chest. 2010;137:263–72.
Pisters R, Lane DA, Nieuwlaat R, de Vos CB, Crijns HJGM, Lip GYH. A novel user-friendly score (HAS-BLED) to assess 1-year risk of major bleeding in patients with atrial fibrillation: the euro heart survey. Chest. 2010;138:1093–100.
Hart RG, Pearce LA, Aguilar MI. Meta-analysis: antithrombotic therapy to prevent stroke in patients who have nonvalvular atrial fibrillation. Ann Intern Med. 2007;146:857–67.
Connolly SJ, Ezekowitz MD, Yusuf S, Eikelboom J, Oldgren J, Parekh A, Pogue J, Reilly PA, Themeles E, Varrone J, Wang S, Alings M, Xavier D, Zhu J, Diaz R, Lewis BS, Darius H, Diener HC, Joyner CD. Wallentin L; RE-LY steering committee and Investigators. N Engl J Med. 2009;361:1139–51. https://doi.org/10.1056/NEJMoa0905561.
Patel MR, Mahaffey KW, Garg J, Pan G, Singer DE, Hacke W, Breithardt G, Halperin JL, Hankey GJ, Piccini JP, Becker RC, Nessel CC, Paolini JF, Berkowitz SD, Fox KA, Califf RM, ROCKET AF Investigators. Rivaroxaban versus warfarin in nonvalvular atrial fibrillation. N Engl J Med. 2011;365:883–91.
Granger CB, Alexander JH, McMurray JJ, Lopes RD, Hylek EM, Hanna M, Al-Khalidi HR, Ansell J, Atar D, Avezum A, Bahit MC, Diaz R, Easton JD, Ezekowitz JA, Flaker G, Garcia D, Geraldes M, Gersh BJ, Golitsyn S, Goto S, Hermosillo AG, Hohnloser SH, Horowitz J, Mohan P, Jansky P, Lewis BS, Lopez-Sendon JL, Pais P, Parkhomenko A, Verheugt FW, Zhu J. Wallentin L; ARISTOTLE committees and Investigators. Apixaban versus warfarin in patients with atrial fibrillation. N Engl J Med. 2011;365:981–92.
Giugliano RP, Ruff CT, Braunwald E, Murphy SA, Wiviott SD, Halperin JL, Waldo AL, Ezekowitz MD, Weitz JI, Špinar J, Ruzyllo W, Ruda M, Koretsune Y, Betcher J, Shi M, Grip LT, Patel SP, Patel I, Hanyok JJ, Mercuri M, Antman EM, ENGAGE AF-TIMI 48 Investigators. Edoxaban versus warfarin in patients with atrial fibrillation. N Engl J Med. 2013;369:2093–104.
Norby FL, Bengtson LGS, Lutsey PL, Chen LY, MacLehose RF, Chamberlain AM, Rapson I, Alonso A. Comparative effectiveness of rivaroxaban versus warfarin or dabigatran for the treatment of patients with non-valvular atrial fibrillation. BMC Cardiovasc Disord. 2017;17:238. https://doi.org/10.1186/s12872-017-0672-5.
Kirchhof P, Benussi S, Kotecha D, Ahlsson A, Atar D, Casadei B, Castella M, Diener HC, Heidbuchel H, Hendriks J, Hindricks G, Manolis AS, Oldgren J, Popescu BA, Schotten U, Van Putte B, Vardas P, Agewall S, Camm J, Baron Esquivias G, Budts W, Carerj S, Casselman F, Coca A, De Caterina R, Deftereos S, Dobrev D, Ferro JM, Filippatos G, Fitzsimons D, Gorenek B, Guenoun M, Hohnloser SH, Kolh P, Lip GY, Manolis A, McMurray J, Ponikowski P, Rosenhek R, Ruschitzka F, Savelieva I, Sharma S, Suwalski P, Tamargo JL, Taylor CJ, Van Gelder IC, Voors AA, Windecker S, Zamorano JL, Zeppenfeld K. 2016 ESC guidelines for the management of atrial fibrillation developed in collaboration with EACTS. Eur Heart J 2016; 37:2893–2962. Epub 2016 Aug 27.
Camm AJ, Accetta G, Ambrosio G, Atar D, Bassand JP, Berge E, Cools F, Fitzmaurice DA, Goldhaber SZ, Goto S, Haas S, Kayani G, Koretsune Y, Mantovani LG, Misselwitz F, Oh S, Turpie AG, Verheugt FW, Kakkar AK, GARFIELD-AF Investigators. Evolving antithrombotic treatment patterns for patients with newly diagnosed atrial fibrillation. Heart. 2017;103:307–14. https://doi.org/10.1136/heartjnl-2016-309832. Epub 2016 Sep 19.
Prochaska JH, Göbel S, Keller K, Coldewey M, Ullmann A, Lamparter H, Jünger C, Al-Bayati Z, Baer C, Walter U, Bickel C, ten Cate H, Münzel T, Wild PS. Quality of oral anticoagulation with phenprocoumon in regular medical care and its potential for improvement in a telemedicine-based coagulation service--results from the prospective, multi-center, observational cohort study thrombEVAL. BMC Med. 2015;13:14. https://doi.org/10.1186/s12916-015-0268-9.
Heidbuchel H, Verhamme P, Alings M, Antz M, Diener HC, Hacke W, Oldgren J, Sinnaeve P, Camm AJ, Kirchhof P. ESC Scientific Document Group Updated European Heart Rhythm Association practical guide on the use of non-vitamin-K antagonist anticoagulants in patients with non-valvular atrial fibrillation: Executive summary. Eur Heart J. 2017;38:2137–49.
Hart RG, Eikelboom JW, Ingram AJ, Herzog CA. Anticoagulants in atrial fibrillation patients with chronic kidney disease. Nat Rev Nephrol. 2012;8:569–78. https://doi.org/10.1038/nrneph.2012.160.
Olesen JB, Lip GYH, Kamper A-L, Hommel K, Køber L, Lane DA, Lindhardsen J, Gislason GH, Torp-Pedersen C. Stroke and bleeding in atrial fibrillation with chronic kidney disease. N Engl J Med. 2012;367:625–35. https://doi.org/10.1056/NEJMoa1105594.
Friberg L, Benson L, Lip GYH. Balancing stroke and bleeding risks in patients with atrial fibrillation and renal failure: the Swedish atrial fibrillation cohort study. Eur Heart J. 2015;36:297–306. https://doi.org/10.1093/eurheartj/ehu139.
Goto S, Angchaisuksiri P, Bassand JP, Camm AJ, Dominguez H, Illingworth L, Gibbs H, Goldhaber SZ, Goto S, Jing ZC, Haas S, Kayani G, Koretsune Y, Lim TW, Oh S, Sawhney JPS, Turpie AGG, van Eickels M, Verheugt FWA, Kakkar AK, GARFIELD-AF Investigators. Management and 1-year outcomes of patients with newly diagnosed atrial fibrillation and chronic kidney disease: results from the prospective GARFIELD - AF registry. J Am Heart Assoc. 2019;8:e010510. https://doi.org/10.1161/JAHA.118.010510.
Hart RG, Pearce LA, Asinger RW, Herzog CA. Warfarin in atrial fibrillation patients with moderate chronic kidney disease. Clin J Am Soc Nephrol CJASN. 2011;6:2599–604. https://doi.org/10.2215/CJN.02400311.
Reddy VY, Sievert H, Halperin J, Doshi SK, Buchbinder M, Neuzil P, Huber K, Whisenant B, Kar S, Swarup V, Gordon N, Holmes D. PROTECT AF steering committee and Investigators. Percutaneous left atrial appendage closure vs warfarin for atrial fibrillation: a randomized clinical trial. JAMA. 2014;312:1988–98.
Holmes DR Jr, Kar S, Price MJ, Whisenant B, Sievert H, Doshi SK, Huber K, Reddy VY. Prospective randomized evaluation of the Watchman left atrial appendage closure device in patients with atrial fibrillation versus long-term warfarin therapy: the PREVAIL trial. J Am Coll Cardiol. 2014;64:1–12. https://doi.org/10.1016/j.jacc.2014.04.029.
Reddy VY, Möbius-Winkler S, Miller MA, Neuzil P, Schuler G, Wiebe J, Sick P, Sievert H. Left atrial appendage closure with the Watchman device in patients with a contraindication for oral anticoagulation: the ASAP study (ASA Plavix feasibility study with Watchman left atrial appendage closure technology). J Am Coll Cardiol. 2013;61:2551–6. https://doi.org/10.1016/j.jacc.2013.03.035.
Fastner C, Hoffmann L, Aboukoura M, Behnes M, Lang S, Borggrefe M, Akin I, Nienaber CA. Real-world experience comparing two common left atrial appendage closure devices. BMC Cardiovasc Disord. 2018;18:171. https://doi.org/10.1186/s12872-018-0899-9.
Ketteler M, Block GA, Evenepoel P, Fukagawa M, Herzog CA, McCann L, Moe SM, Shroff R, Tonelli MA, Toussaint ND, Vervloet MG, Leonard MB. Executive summary of the 2017 KDIGO chronic kidney disease-mineral and bone disorder (CKD-MBD) guideline update: what's changed and why it matters. Kidney Int. 2017;92:26–36. https://doi.org/10.1016/j.kint.2017.04.006.
Levey AS, Stevens LA, Schmid CH, Zhang YL, Castro AF 3rd, Feldman HI, Kusek JW, Eggers P, Van Lente F, Greene T, Coresh J. CKD-EPI (chronic kidney disease epidemiology collaboration). A new equation to estimate glomerular filtration rate. Ann Intern Med. 2009;150:604–12.
Luani B, Rauwolf T, Groscheck T, Tanev I, Herold J, Isermann B, Schmeisser A, Braun-Dullaeus RC. Serial assessment of natriuretic peptides in patients undergoing interventional closure of the left atrial appendage. Heart Lung Circ. 2018;27:828–34. https://doi.org/10.1016/j.hlc.2017.07.001. Epub 2017 Aug 1.
Lip GYH, Frison L, Halperin JL, Lane DA. Comparative validation of a novel risk score for predicting bleeding risk in anticoagulated patients with atrial fibrillation: the HAS-BLED (hypertension, abnormal renal/liver function, stroke, bleeding history or predisposition, labile INR, elderly, drugs/alcohol concomitantly) score. J Am Coll Cardiol. 2011;57:173–80. https://doi.org/10.1016/j.jacc.2010.09.024.
Reddy VY, Doshi SK, Kar S, Gibson DN, Price MJ, Huber K, Horton RP, Buchbinder M, Neuzil P, Gordon NT, Holmes DR Jr. PREVAIL and PROTECT AF Investigators. 5-year outcomes after left atrial appendage closure: from the PREVAIL and PROTECT AF trials. J Am Coll Cardiol. 2017;70:2964–75. https://doi.org/10.1016/j.jacc.2017.10.021 Epub 2017 Nov 4.
Boersma LV, Ince H, Kische S, Pokushalov E, Schmitz T, Schmidt B, Gori T, Meincke F, Protopopov AV, Betts T, Mazzone P, Foley D, Grygier M, Sievert H, De Potter T, Vireca E, Stein K. Bergmann MW; following investigators and institutions participated in the EWOLUTION study. Evaluating real-world clinical outcomes in atrial fibrillation patients receiving the WATCHMAN left atrial appendage closure technology. Circ Arrhythm Electrophysiol. 2019;12:e006841. https://doi.org/10.1161/CIRCEP.118.006841.
Shah M, Avgil Tsadok M, Jackevicius CA, Essebag V, Eisenberg MJ, Rahme E, Humphries KH, Tu JV, Behlouli H, Guo H, Pilote L. Warfarin use and the risk for stroke and bleeding in patients with atrial fibrillation undergoing dialysis. Circulation. 2014;129:1196–203. https://doi.org/10.1161/CIRCULATIONAHA.113.004777.
Wizemann V, Tong L, Satayathum S, Disney A, Akiba T, Fissell RB, Kerr PG, Young EW, Robinson BM. Atrial fibrillation in hemodialysis patients: clinical features and associations with anticoagulant therapy. Kidney Int. 2010;77:1098–106. https://doi.org/10.1038/ki.2009.477.
Holmes DR, Doshi SK, Kar S, Price MJ, Sanchez JM, Sievert H, Valderrabano M, Reddy VY. Left atrial appendage closure as an alternative to warfarin for stroke prevention in atrial fibrillation. A Patient-Level Meta-Analysis J Am Coll Cardiol. 2015;65:2614–23. https://doi.org/10.1016/j.jacc.2015.04.025.
Luani B, Groscheck T, Genz C, Tanev I, Rauwolf T, Herold J, Medunjanin S, Schmeisser A, Braun-Dullaeus RC. Left atrial enlargement and clinical considerations in patients with or without a residual interatrial shunt after closure of the left atrial appendage with the WATCHMAN™-device. BMC Cardiovasc Disord. 2017;17:294. https://doi.org/10.1186/s12872-017-0728-6.
Kaneko H, Neuss M, Weissenborn J, Butter C. Predictors of thrombus formation after percutaneous left atrial appendage closure using the WATCHMAN device. Heart Vessel. 2017;32:1137–43. https://doi.org/10.1007/s00380-017-0971-x.
Ketterer U, D’Ancona G, Siegel I, Ortak J, Ince H, Kische S. Percutaneous left atrial appendage occlusion: device thrombosis in clopidogrel non-responders. Int J Cardiol. 2016;204:196–7. https://doi.org/10.1016/j.ijcard.2015.11.160.
Pillarisetti J, Reddy YM, Gunda S, Swarup V, Lee R, Rasekh A, Horton R, Massumi A, Cheng J, Bartus K, Badhwar N, Han F, Atkins D, Bommana S, Earnest M, Nath J, Ferrell R, Bormann S, Dawn B, Di Biase L, Mansour M, Natale A, Lakkireddy D. Endocardial (Watchman) vs epicardial (lariat) left atrial appendage exclusion devices: understanding the differences in the location and type of leaks and their clinical implications. Heart Rhythm. 2015;12:1501–7. https://doi.org/10.1016/j.hrthm.2015.03.020. Epub 2015 Mar 14.
Saw J, Tzikas A, Shakir S, Gafoor S, Omran H, Nielsen-Kudsk JE, Kefer J, Aminian A, Berti S, Santoro G, Nietlispach F, Moschovitis A, Cruz-Gonzalez I, Stammen F, Tichelbäcker T, Freixa X, Ibrahim R, Schillinger W, Meier B, Sievert H, Gloekler S. Incidence and clinical impact of device-associated Thrombus and Peri-device leak following left atrial appendage closure with the Amplatzer cardiac plug. JACC Cardiovasc Interv. 2017;10:391–9. https://doi.org/10.1016/j.jcin.2016.11.029.
Hornung M, Gafoor S, Id D, Vaskelyte L, Hofmann I, Franke J, Sievert H, Bertog SC. Catheter-based closure of residual leaks after percutaneous occlusion of the left atrial appendage. Catheter Cardiovasc Interv. 2016;87:1324–30. https://doi.org/10.1002/ccd.26318. Epub 2015 Dec 23.
This research received no grants from any funding agencies in the public, commercial or not-for-profit sectors.
Availability of data and materials
Due to ethical and legal restrictions with respect to patient privacy, investigators are not authorized to make the clinical data publicly available. The data will be available to interested researchers upon request by contacting Dr. Blerim Luani (firstname.lastname@example.org).
Ethics approval and consent to participate
The study was approved by the ethics committee of the Otto-von-Guericke University Magdeburg on Mai 4th, 2015 (approval number 10/15). All patients gave their informed written consent before enrollment in the study. The Study was registered on 07.07.2016 in the German Clinical Trials Register (DRKS; ID: DRKS00010768).
Consent for publication
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
- LAA closure
- Watchman device
- Chronic kidney disease
- Oral anticoagulation
- Cerebrovascular events
- Bleeding complications
- Device related thrombi