- Research article
- Open Access
Deactivation vs. asynchronous pacing - prospective evaluation of a protocol for rhythm management in patients with magnetic resonance conditional pacemakers undergoing adenosine stress cardiovascular magnetic resonance imaging
BMC Cardiovascular Disorders volume 17, Article number: 142 (2017)
Cardiovascular Magnetic Resonance (CMR) imaging with adenosine stress is an important diagnostic tool in patients with known or suspected coronary artery disease (CAD). However, the method is not yet established for CAD patients with pacemakers (PM) in clinical practice. A possible reason is that no recommendations exist for PM setting (paused pacing or asynchronous mode) during adenosine stress. We elaborated a protocol for rhythm management in clinical routine for PM patients that considers heart rate changes under adenosine using a test infusion of adenosine in selected patients.
47 consecutive patients (mean age 72.3 ± 10,0 years) with MR conditional PM and known or suspected CAD who underwent CMR in clinical routine were studied in this prospective observational study. PM indications were sinus node dysfunction (SND, n = 19; 40,4%), atrioventricular (AV) block (n = 26; 55.3%) and bradyarrhythmia in permanent atrial fibrillation (AF, n = 2; 4.3%). In patients with SND, normal AV-conduction and resting HR >45 bpm at the time of CMR and in AF the PM was deactivated for the scan. In intermittent AV-block a test infusion of adenosine was given prior to the scan. All patients with permanent higher degree sinuatrial or AV-block or deterioration of AV-conduction in the adenosine test were paced asynchronously during CMR, in patients with preserved AV-conduction under adenosine the pacemaker was deactivated. CMR protocol included cine imaging, adenosine stress perfusion and late gadolinium enhancement.
The adenosine test was able to differentiate between mandatory PM stimulation during CMR and safe deactivation of the device. In patients with permanent sinuatrial or AV-block (n = 11; 23.4%) or deterioration of AV conduction in the adenosine test (n = 5, 10.6%) asynchronous pacing above resting heart rate did not interfere with intrinsic rhythm, no competitive stimulation was seen during the scan. 10 of 15 (66,7%) patients with intermittent AV-block showed preserved AV-conduction under adenosine. As in SND and AF deactivation of the PM showed to be safe during CMR, no bradycardia was observed.
Our protocol for rhythm management during adenosine stress CMR showed to be feasible and safe and may be recommended for pacemaker patients undergoing routine CMR.
CMR as a non-invasive imaging modality is firmly established in the clinical workup for patients with known or suspected CAD. Beside cine imaging and scar detection with Late Gadolinium Enhancement (LGE) ischemia detection plays a major role in CAD [1, 2]. It has a class Ia level A recommendation in case of intermediate pre test probability of CAD in latest guidelines . Hemodynamic relevance of stenoses in known CAD can be evaluated reliably .
In patients with cardiac conduction disorders underlying or concomitant CAD is common [4, 5]. However, while safety of CMR without stress agents in patients with MR conditional PM has been shown in a number of studies [6,7,8] no prospective data exists on rhythm management of those patients in adenosine stress CMR.
In MR conditional PM no inhibited mode can be chosen. Only deactivation (ODO-mode) or asynchronous pacing (DOO, AOO, VOO)  are available to avoid inhibition by electromagnetic interference or tracking of electromagnetic impulses. In SND and AV-block selecting an adequate pacing mode for routine adenosine stress CMR can be challenging because the effect of adenosine on heart rate (HR) has not been thoroughly studied in this entity of patients. Asynchronous mode (i.e. pacing at a fixed rate above baseline HR) could result in competitive pacing when HR accelerates under adenosine  reaching the pacing rate. This is due to the fact that sensing is deactivated in this mode and intrinsic rhythm cannot inhibit PM activity. PM stimuli falling in the vulnerable period of the cardiac cycle could then trigger arrhythmia . On the other hand deactivation of the PM in patients with normal HR under resting conditions could result in bradycardia or asystole under adenosine administration .
We recently published first data on adenosine stress CMR in pacemaker patients in a small retrospective study and found no relevant complications under a predefined pacing protocol . However, the number of patients with AV-block was limited and it remains unclear what pacing mode is preferable for patients with intermittent AV-block i.e. without higher degree AV-block at the time of CMR. To further investigate adenosine response and to decide whether the pacemaker can be deactivated during the scan we used a test infusion of adenosine for patients with intermittent AV-block and preserved AV-conduction at the time of CMR to screen for adenosine induced bradycardia.
In conclusion the diagnostic value of perfusion analysis by CMR in CAD is well known but still no recommendations exist for PM programming in MRI conditional devices during adenosine stress. We sought to test the safety of a protocol for rhythm management in those patients that is adapted to the adenosine response in PM patients with different underlying conduction disorders.
In this prospective observational study we analyzed 47 consecutive patients with MR conditional PM who underwent routine adenosine stress CMR including cine imaging, adenosine stress perfusion and LGE from april 2015 to december 2016. Inclusion criteria were presence of a MR conditional pacemaker system, indication for adenosine stress CMR for the evaluation of known or suspected CAD, age > 18 years. Exclusion criteria were implantation <6 weeks from the scan, epicardial or abandoned pacemaker leads, bronchial, asthma, impaired renal function, ejection fraction <35%, presence of an antitachycardia device.
Patients had a mean age of 72.1 ± 11.0 years. 19 (40.4%) had known CAD, 8 (17.0%) previous MI. All other patients had intermediate pretest probability of CAD . Echocardiography had shown preserved systolic left ventricular (LV) function in all subjects. Pacemaker indications were sinus node dysfunction (SND, 19; 40.4%), second or third degree AV-block (n = 26; 55.3%) and bradyarrhythmia in permanent AF (n = 2; 4.3%). Eight patients (17.0%) were PM dependent (HR <30 bpm). Impulse generator/lead models were Advisa (n = 1; 2.1%) and Ensura (n = 46; 97.9%) MRI SureScan/CapSureFix 5076 Novus (atrial), CapSureSense 4074 (ventricular) (Medtronic Inc., Minneapolis, MA, USA). For detailed baseline characteristics see Table 1.
Pacemaker programming and adenosine test
CMR was performed more than six weeks after PM implantation in all individuals according to ESC guidelines . Prior to and after CMR imaging battery status of the device, lead impedance, pacing capture thresholds and sensing amplitudes were measured. In patients with intermittent AV-block and preserved AV-conduction at the time of CMR PQ-interval was measured in resting ECG and the Wenckebach point was assessed by testing in AOO mode via the PM programmer. A Wenckebach point below 120 bpm was considered pathologic. Furthermore these patients received a test infusion of adenosine as 3-min infusion of 140μg/kg body weight/min according to the dose for CMR. The pacemaker was deactivated for the test (ODO). Possible deterioration of AV-conduction was recorded and classified in progression to second degree AV-block II Mobitz I, second degree AV-block Mobitz II and third degree AV-block. Reactivation of pacing was performed immediately in case of bradycardia. In SND the PQ interval and the Wenckebach point were measured to rule out occult AV-nodal disease; as in AF no adenosine test was performed prior to MR.
Devices were programmed to MR safe mode following manufacturer’s instructions immediately prior to the scan and reprogrammed immediately thereafter. Programming was performed according to a predefined protocol: To avoid interference of intrinsic rhythm with PM-stimulation in patients with SND and resting heart rate HR > 45 bpm no pacing (ODO)-mode was engaged during the scan - also when atrial fibrillation (AF) was present. In individuals with SND and HR ≤ 45 bpm the pacemaker was set to asynchronous atrial stimulation (AOO, 60 bpm). All patients with permanent second or third degree AV block and those with deterioration of AV-conduction in the adenosine test were continuously paced in asynchronous mode. Pacing rate was set 10 bpm above spontaneous heart rate with a minimum of 60 bpm. VOO mode at 60 bpm was chosen in AV block with sinus rate > 45 bpm to avoid competitive atrial stimulation, DOO mode at 60 bpm in AV block with sinus bradycardia ≤45 bpm. In patients with intermittent AV-block and preserved AV-conduction (i.e. no progression to second or third degree AV-block in the adenosine test) the PM was deactivated as in SND. Patients in AF at the time of the scan were paced VOO at 60 bpm if resting heart rate was ≤45 bpm. Table 2 shows the protocol used to select pacing modes for specific clinical constellations.
As previously described  patients were monitored during the scan with continuous electrocardiographic and visual supervision by a cardiologist present in the scanner room. Voice contact was maintained with the patient at all times of the scan. Advanced cardiac life support protocol was in effect. In the scanner the patient was placed on a carry sheet; medical staff was trained for rapid removal of the patient from the scanner room in the event of cardiopulmonary compromise. Thus immediate treatment of severe arrhythmia and reactivation of PM stimulation within seconds in non-paced patients was guaranteed. Atropine, adrenaline and theophylline injections were prepared ready for use in case of bradycardia. Two separate cubital venous cannulas were used for adenosine and gadolinium contrast agent respectively.
Cardiac magnetic resonance imaging
CMR scans were performed with a 1.5 T wide bore system (ESPREE – Siemens Healthcare, Erlangen, Germany) using a 4-channel body array and an 8-channel spine coil. Maximum gradient field was 33 mT/m (Z-Engine) with a slew rate of 100 T/m/S. maximum specific absorption rates were limited to 2.0 W/kg.
As previously published  our standard protocol meets the Society of Cardiovascular Magnetic Resonance (SCMR) guidelines for CMR . Cine steady-state free precession (SSFP) gradient-echo images were obtained in 10 to 12 short axis slices depending on the size of the ventricles and in 3 long axis planes corresponding to two, three and four chamber views. For stress perfusion-imaging adenosine was administered as 3-min infusion of 140μg/kg body weight/min. First-pass perfusion imaging was carried out with intravenous bolus administration of gadolinium (0.2 mmol/kg body weight) in a fast low angle shot (FLASH) sequence (3 to 4 slices). Late Gadolinium Enhancement (LGE) images were acquired fifteen minutes after injection of gadolinium as phase-sensitive inversion-recovery (PSIR) in short (10 to 12 slices) and long axis (3 planes) views.
Adenosine test prior to CMR
15 patients (31.9%) with intermittent AV-block and preserved AV-conduction at the time of CMR were tested. 5 of them (33.3%) showed deterioration of AV conduction with a drop of HR. All of those individuals had both prolonged PQ-interval (220 to 305 ms) and pathologic Wenckebach point (70 to 90 bpm). Two patients showed worsening of AV-conduction to AV-block II Mobitz II, three patients developed AV-block III. One patient had pathologic PQ-interval (230 ms) and normal Wenckebach point (130 bpm) and showed acceleration of HR from 70 to 85 bpm without progressive AV-block. One patient had normal AV-conduction (190 ms) and pathologic Wenckebach point (115 ms) and showed an increase in HR from 85 to 95 bpm. Seven of 15 patients with intermittent AV-block (46.6%%) had normal PQ-interval and normal Wenckebach point. None of them showed higher degree AV-block in the test, heart rate increased from 66 ± 8.6 to 75 ± 11.4 bpm (p < 0.001, paired t-test). Thus only in patients with both pathologic PQ-interval and pathologic Wenckebach point worsening of AV-conduction and drop of HR were observed. The adenosine test was well tolerated by the patients; no adverse reactions were seen. Figure 1 summarizes heart rate response under the test infusion of adenosine.
Effect and safety of adenosine administration for stress perfusion
There were no adenosine induced adverse events. Asynchronous stimulation of patients with permanent AV-block or progressive AV-block under adenosine 10 bpm above resting HR did not cause competitive pacing or arrhythmia during MR. Patients with preserved AV-conduction under adenosine and deactivated PM did not develop higher degree AV-block in CMR, HR increased from 67.6 ± 9.1 to 77.6 ± 11.3 bpm. Two patients with SND were in AF at the time of CMR. In patients with SND and normal AV-conduction i.e. normal PQ interval and normal Wenckebach point (n = 17; 36.2%) at the time of the scan with deactivated (ODO) pacemaker adenosine administration accelerated HR from 60.1 ± 9.1 to 76.0 ± 9.3 bpm (p < 0.001, Wilcoxon signed rank test). AV-conduction was not significantly influenced by adenosine; no higher degree AV block occurred. When sinus rate was <45 bpm (n = 1; 2.1%) AOO pacing at 60 bpm led to permanent capture, no acceleration of HR under adenosine was noticed. Of 4 patients in AF in 3 HR remained unchanged during stress perfusion, one patient had acceleration of HR from 110 to 125 bpm.
Device integrity was not compromised by the CMR scan. No patient showed relevant alterations of lead impedance, pacing capture threshold, sensing amplitude or battery voltage. Table 3 summarizes device parameters pre and post CMR.
Adenosine stress CMR in patients with MR conditional PM is complicated by the absence of an inhibited pacing mode in these devices. Choosing an adequate PM setting (deactivation vs. asynchronous pacing) in specific conduction disorders can be challenging due to possible alterations of HR and heart rhythm under adenosine. Until now no guidelines for clinical routine have been established addressing this issue. We therefore elaborated a protocol for rhythm management intending to minimize the risk of arrhythmia due to competitive PM stimulation on the one hand and possible bradycardia on the other. In this study our protocol proved to be safe and feasible. No complications of adenosine stress CMR related to the presence of a PM or the underlying cardiac conduction disorder occurred.
The present study is in line with our previous findings  that in SND with normal resting HR and normal PQ interval paused PM stimulation (ODO mode) is suitable for adenosine stress perfusion. We found predominance of the sympatho-excitatory effect of adenosine  that overrides cardiac inhibition comparable to patients without SND resulting in a significant increase in HR. We selected DOO pacing only in patients with sinusbradycardia <45 bpm to prevent competitive atrial stimulation which can induce AF , also considering the fact that adenosine may promote AF by shortening the atrial action potential and refractory period . Avoiding competitive atrial stimulation by pacing above intrinsic HR is not useful due to the acceleration of HR under adenosine (up to 40 bpm in tis study). Pacing far above baseline HR for a longer time could cause discomfort or even circulatory compromise in PM patients adapted to relative bradycardia . VOO mode was avoided because ventricular tachycardia can be induced by R-on-T PM stimulation . The risk of proarrhythmia with asynchronous ventricular pacing for PM interrogation is considered low . However, routine adenosine stress CMR in CAD requires a long period in MR conditional mode because sequences for localization, cine imaging, first pass perfusion and LGE are necessary. Asynchronous ventricular PM stimulation might be hazardous under these conditions. In patients with higher degree AV-block or bradycardia induced by adenosine in the preceding test the risk of R-on-T stimulation considered low because the patient is paced above intrinsic HR at rest and under adenosine. Prolonged asystole under adenosine stress imaging in occult SND has been reported , therefore immediate reactivation of the PM in case of persistent bradycardia must be guaranteed. For further clinical practice we recommend deactivation of the device under careful monitoring in SND without AV-block.
The adenosine test in patients with intermittent AV-block was useful to differentiate patients that can be examined with deactivated devices from those with mandatory PM stimulation. One could argue that asynchronous pacing for all patients with AV-block may be chosen. However, with intact AV-conduction under adenosine the risk of proarrhythmia due to competitive ventricular stimulation cannot be excluded. Deterioration of AV-conduction was only observed in patients with both prolonged PQ interval and pathologic Wenckebach point. Larger studies have to show if CMR can be performed without previous testing in patients without first degree AV-block at rest and normal Wenckebach point. We believe that screening for occult AV-dysfunction by measuring the PQ-interval and Wenckebach point prior to CMR in SND is useful because SND patients may develop binodal disease  that does not become clinically apparent due to the PM therapy. We recommend adenosine test for SND patients with pathologic AV-conduction at the time of CMR, however all individuals in this study had normal AV-conduction. Further studies have to show if adenosine testing is adequate for patients with SND and impaired AV-conduction unmasked prior to CMR.
The investigated pacing protocol showed to be feasible and safe. However, adenosine stress CMR has to be compared to other diagnostic strategies, namely when the high supervisory expense in this setting is considered. We encourage prospective randomized studies to clarify which imaging strategy is the best choice for PM patients in terms of safety and clinical value.
This study is limited by the relatively sample size. Adverse effects may only appear in a larger cohort of patients. Furthermore we did not test a high dose adenosine protocol, which may be more effective in ischemia detection .
Our protocol for rhythm management in patients with MR conditional PM appears to be feasible and safe and may be used for adenosine CMR in clinical routine until larger studies exist.
Coronary artery disease
Cardiovascular Magnetic Resonance
European Society of Cardiology
Fast low angle shot
Late Gadolinium Enhancement
Pacing capture threshold
Phase-sensitive inversion recovery
Society for Cardiovascular Magnetic Resonance
Sinus node dysfunction
Steady-state free precession
Short-tau inversion recovery
Nandalur KR, Dwamena BA, Choudhri AF, Nandalur MR, Carlos RC. Diagnostic performance of stress cardiac magnetic resonance imaging in the detection of coronary artery disease: a meta-analysis. J am Coll Cardiol. 2007;50:1343–53.
Authors/Task Force m, Windecker S, Kolh P, Alfonso F, Collet JP, Cremer J, et al. 2014 ESC/EACTS guidelines on myocardial revascularization: the Task Force on myocardial revascularization of the European Society of Cardiology (ESC) and the European Association for Cardio-Thoracic Surgery (EACTS) developed with the special contribution of the European Association of Percutaneous Cardiovascular Interventions (EAPCI). Eur Heart J. 2014;35:2541–619.
Li M, Zhou T, Yang LF, Peng ZH, Ding J, Sun G. Diagnostic accuracy of myocardial magnetic resonance perfusion to diagnose ischemic stenosis with fractional flow reserve as reference: systematic review and meta-analysis. JACC Cardiovasc Imaging. 2014;7:1098–105.
Alonso A, Jensen PN, Lopez FL, Chen LY, Psaty BM, Folsom AR, et al. Association of sick sinus syndrome with incident cardiovascular disease and mortality: the atherosclerosis risk in communities study and cardiovascular health study. PLoS One. 2014;9:e109662.
Hsueh CW, Lee WL, Chen YT, Ting CT. The incidence of coronary artery disease in patients with symptomatic bradyarrhythmias. Jpn Heart J. 2001;42:417–23.
Wollmann CG, Thudt K, Kaiser B, Salomonowitz E, Mayr H, Globits S. Safe performance of magnetic resonance of the heart in patients with magnetic resonance conditional pacemaker systems: the safety issue of the ESTIMATE study. J Cardiovasc Magn Reson. 2014;16:30.
Gimbel JR, Bello D, Schmitt M, Merkely B, Schwitter J, Hayes DL, et al. Randomized trial of pacemaker and lead system for safe scanning at 1.5 Tesla. Heart Rhythm. 2013;10:685–91.
Schwitter J, Kanal E, Schmitt M, Anselme F, Albert T, Hayes DL, et al. Impact of the Advisa MRI pacing system on the diagnostic quality of cardiac MR images and contraction patterns of cardiac muscle during scans: Advisa MRI randomized clinical multicenter study results. Heart Rhythm. 2013;10:864–72.
Bernstein AD, Daubert JC, Fletcher RD, Hayes DL, Luderitz B, Reynolds DW, et al. The revised NASPE/BPEG generic code for antibradycardia, adaptive-rate, and multisite pacing. North American Society of Pacing and Electrophysiology/British pacing and electrophysiology group. Pacing Clin Electrophysiol. 2002;25:260–4.
Biaggioni I, Killian TJ, Mosqueda-Garcia R, Robertson RM, Robertson D. Adenosine increases sympathetic nerve traffic in humans. Circulation. 1991;83:1668–75.
Sweeney MO, Porkolab FL. Ventricular fibrillation induced by double premature ventricular pacing stimuli in a dual-chamber pacemaker with auto capture. Heart Rhythm. 2009;6:429–32.
Hesse B, Tagil K, Cuocolo A, Anagnostopoulos C, Bardies M, Bax J, et al. EANM/ESC procedural guidelines for myocardial perfusion imaging in nuclear cardiology. Eur J Nucl med Mol Imaging. 2005;32:855–97.
Klein-Wiele O, Garmer M, Urbien R, Busch M, Kara K, Mateiescu S, et al. Feasibility and safety of adenosine cardiovascular magnetic resonance in patients with MR conditional pacemaker systems at 1.5 Tesla. J Cardiovasc Magn Reson. 2015;17:112.
Task Force M, Montalescot G, Sechtem U, Achenbach S, Andreotti F, Arden C, et al. 2013 ESC guidelines on the management of stable coronary artery disease: the Task Force on the management of stable coronary artery disease of the European Society of Cardiology. Eur Heart J. 2013;34:2949–3003.
Brignole M, Auricchio A, Baron-Esquivias G, Bordachar P, Boriani G, Breithardt OA, et al. 2013 ESC guidelines on cardiac pacing and cardiac resynchronization therapy: the Task Force on cardiac pacing and resynchronization therapy of the European Society of Cardiology (ESC). Developed in collaboration with the European heart rhythm Association (EHRA). Eur Heart J. 2013;34:2281–329.
American College of Cardiology Foundation Task Force on Expert Consensus D, Hundley WG, Bluemke DA, Finn JP, Flamm SD, Fogel MA, et al. ACCF/ACR/AHA/NASCI/SCMR 2010 expert consensus document on cardiovascular magnetic resonance: a report of the American College of Cardiology Foundation Task Force on expert consensus documents. J am Coll Cardiol. 2010;55:2614–62.
Costa F, Biaggioni I. Adenosine activates afferent fibers in the forearm, producing sympathetic stimulation in humans. J Pharmacol exp Ther. 1993;267:1369–74.
Furman S, Cooper JA. Atrial fibrillation during A-V sequential pacing. Pacing Clin Electrophysiol. 1982;5:133–5.
Kanei Y, Hanon S, Van-Tosh A, Schweitzer P. Adenosine-induced atrial fibrillation during pharmacologic stress testing: report of eight cases and review of the literature. Int J Cardiol. 2008;129:e15–7.
Tops LF, Schalij MJ, Bax JJ. The effects of right ventricular apical pacing on ventricular function and dyssynchrony implications for therapy. J am Coll Cardiol. 2009;54:764–76.
Oupadia P, Ramaswamy K. Images in clinical medicine. “R-on-T” phenomenon. N Engl J med. 1998;338:1812.
Nowak B, Hemmer W, Israel CW, Kramer LI, Neuzner J, Pfeiffer D, et al. Statement of the working Group of the Germany Society on the safety of asynchronous ventricular pacemaker stimulation. Clin res Cardiol. 2006;95:57–60.
Pennell DJ, Mahmood S, Ell PJ, Underwood SR. Bradycardia progressing to cardiac arrest during adenosine thallium myocardial perfusion imaging in occult sino-atrial disease. Eur J Nucl med. 1994;21:170–2.
Rosenqvist M. Atrial pacing for sick sinus syndrome. Clin Cardiol. 1990;13:43–7.
Karamitsos TD, Ntusi NA, Francis JM, Holloway CJ, Myerson SG, Neubauer S. Feasibility and safety of high-dose adenosine perfusion cardiovascular magnetic resonance. J Cardiovasc Magn Reson. 2010;12:66.
We acknowledge Michael Pilz for editing the graphic content of the manuscript.
Availability of data and materials
OK conceived of the study, performed the PM interrogations and tests and wrote the manuscript. MG was involved in the conception of the study. GB and RU collected clinical data. MB helped in statistical analysis and literature review and revised the manuscript. KK helped in the conception of the study and in drafting the manuscript. HS was involved in the conception of the study and data analysis. MS was involved in the conception of the study. BH was involved in the conception of the study and revised the manuscript. DG helped in the coordination of the study and revised the manuscript. All authors read and approved the final manuscript.
OK: Cardiologist, member of the German Cardiac Society (DGK), SCMR and DGK Level III qualification for CMR, DGK-approved for interventional cardiology and special rhythmology. MG: Radiologist, member of the German Röntgen Society, Q2 certificate for CMR. GB: Doctorand, RU: Cardiologist, DGK member. MB: Physicists, research-, software-, and application specialist for MR. KK: Cardiologist, DGK member, approved for interventional cardiology. HS: Cardiologist, member of the German Cardiac Society (DGK), DGK-approved for interventional cardiology and special rhythmology. MS: Cardiologist, DGK-member, approved for interventional cardiology, SCMR and DGK level III qualification for CMR. BH: Cardiologist, DGK-member, approved for interventional cardiology and special rhythmology. DG: Radiologist, member of the German Röntgen Society, specialist in MR guided interventional therapy.
The authors declare that they have no competing interests.
Consent for publication
Ethics approval and consent to participate
The study complies with the Declaration of Helsinki and was approved by the ethics committee of the University of Witten/Herdecke. All individuals gave their consent to participate prior to inclusion.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Klein-Wiele, O., Garmer, M., Barbone, G. et al. Deactivation vs. asynchronous pacing - prospective evaluation of a protocol for rhythm management in patients with magnetic resonance conditional pacemakers undergoing adenosine stress cardiovascular magnetic resonance imaging. BMC Cardiovasc Disord 17, 142 (2017). https://doi.org/10.1186/s12872-017-0579-1
- MRI conditional
- Cardiovascular magnetic resonance
- Adenosine stress
- Sinus node dysfunction
- Atrioventricular block
- Asynchronous pacing