- Research article
- Open Access
- Open Peer Review
Incidence and timing of potentially high-risk arrhythmias detected through long term continuous ambulatory electrocardiographic monitoring
© Solomon et al. 2016
- Received: 12 November 2015
- Accepted: 2 February 2016
- Published: 17 February 2016
Ambulatory electrocardiographic (ECG) monitoring is the standard to screen for high-risk arrhythmias. We evaluated the clinical utility of a novel, leadless electrode, single-patient-use ECG monitor that stores up to 14 days of a continuous recording to measure the burden and timing of potentially high-risk arrhythmias.
We examined data from 122,815 long term continuous ambulatory monitors (iRhythm ZIO® Service, San Francisco) prescribed from 2011 to 2013 and categorized potentially high-risk arrhythmias into two types: (1) ventricular arrhythmias including non-sustained and sustained ventricular tachycardia and (2) bradyarrhythmias including sinus pauses >3 s, atrial fibrillation pauses >5 s, and high-grade heart block (Mobitz Type II or third-degree heart block).
Of 122,815 ZIO® recordings, median wear time was 9.9 (IQR 6.8–13.8) days and median analyzable time was 9.1 (IQR 6.4–13.1) days. There were 22,443 (18.3 %) with at least one episode of non-sustained ventricular tachycardia (NSVT), 238 (0.2 %) with sustained VT, 1766 (1.4 %) with a sinus pause >3 s (SP), 520 (0.4 %) with a pause during atrial fibrillation >5 s (AFP), and 1486 (1.2 %) with high-grade heart block (HGHB). Median time to first arrhythmia was 74 h (IQR 26–149 h) for NSVT, 22 h (IQR 5–73 h) for sustained VT, 22 h (IQR 7–64 h) for SP, 31 h (IQR 11–82 h) for AFP, and 40 h (SD 10–118 h) for HGHB.
A significant percentage of potentially high-risk arrhythmias are not identified within 48-h of ambulatory ECG monitoring. Longer-term continuous ambulatory ECG monitoring provides incremental detection of these potentially clinically relevant arrhythmic events.
- Ambulatory monitoring
- Diagnostic testing
Ambulatory electrocardiographic (ECG) monitoring is the standard of care to screen symptomatic outpatient adults for high-risk ventricular and atrial arrhythmias [1–3]. However, there is marked variation in the technological features and patient compliance among different ECG monitoring systems [4, 5]. Traditional 24-h monitoring devices (i.e., Holter monitors) often do not detect symptomatic or clinically meaningful arrhythmias [6, 7].
Recent technological advances have allowed for higher fidelity recording and larger storage capacities that are able to capture full disclosure ECG recordings beyond the traditional 24- or 48-h monitoring periods. Furthermore, innovative device designs aim to increase patient convenience and patient compliance. Emerging evidence suggests that longer wear times yield greater arrhythmia detection in selected at-risk patients that could impact clinical decision-making and outcomes . Although there has been very limited evaluation of this approach outside of detecting the presence of atrial fibrillation, longer monitoring periods are emerging as a new standard of care for selected patients.
To understand the applicability in day-to-day clinical practice, we evaluated contemporary results from a novel, long-term ambulatory ECG monitoring system to measure the burden and timing of potentially high-risk arrhythmias, including ventricular tachycardia, high-grade heart block and clinically significant pauses in atrioventricular conduction.
Data and study population
We analyzed data for all the ZIO® Service long-term continuous ambulatory ECG monitors (ZIO® Service, iRhythm Technologies, Inc., San Francisco, California) that were prescribed from November 2011 to December 2013 (N = 128,401). The ZIO® Patch is a lightweight, lead-wire free, single-patient-use ECG monitor that adheres to the left upper chest and records and stores up to 14 days of continuous, beat-to-beat ECG. Patients have the option of pressing a trigger button on the device and filling out a log to document symptomatic events during their wear duration, which allows for symptom-rhythm correlation in the ECG report. After a patient completes their 14-day recording, the ZIO® Patch is removed from the chest and mailed to iRhythm Technologies, Inc., where the up to 14-day single-channel recording is analyzed using a combination of proprietary algorithms and review by Certified Cardiac Technicians (CCT). The findings are then reported to the ordering physician in a report that includes information on several standard arrhythmias, including atrial fibrillation and flutter, ventricular tachycardia, supraventricular tachycardia, atrioventricular pauses, heart block, atrial and ventricular ectopic beats, and other identified arrhythmias. All components of the device are recycled after data downloading. Further details on the ZIO® Service and its analytic algorithms have been described previously [5, 9].
We applied standard quality control techniques to assemble a cleaned, analytic dataset of basic patient information and detailed information on detected arrhythmias. This included removing outliers that contained likely erroneous data, including records with heart rates >300 beats or <20 per minute, along with excluding patients <18 years old, records with start times outside our study dates, and records with wear-time or analyzable-time of less than 24 h. Analyzable time was calculated as the amount of time that the ECG patch was recording (enrollment period) minus the amount of time of unanalyzable ECG signal due to artifact.
We categorized potentially high-risk arrhythmias into two types: (1) ventricular arrhythmias, including non-sustained and sustained ventricular tachycardia; and (2) bradyarrhythmias including sinus pauses >3 s, atrial fibrillation pauses >5 s, and high grade heart block including Mobitz Type II or third-degree heart block. Sustained ventricular tachycardia (VT) included VT that lasted greater than 30 s. Mobitz II heart block and third degree heart block were identified by the manufacturer according to their FDA-cleared algorithms and 100 % data curation and quality review by CCT’s trained in advanced arrhythmia detection. Symptomatic pauses were defined as a pause (greater than 3 s for sinus rhythm and greater than 5 s for those in atrial fibrillation) that occurred within 45 s of a patient trigger.
All analyses were performed at the Kaiser Permanente Northern California Division of Research using SAS statistical software, version 9.3 (Cary, N.C.). Continuous variables were reported as means with standard deviations and categorical variables as frequencies and proportions. We calculated the proportion and associated 95 % confidence interval of patients with each arrhythmia overall and the cumulative yield per additional day of monitoring. We used chi-squared tests to compare the proportion of high-risk arrhythmias detected at 1-, 2-, and 7-days versus 14-days.
A research exemption was obtained from the institutional review board of the Kaiser Foundation Research Institute given that the analyses were completed on a fully de-identified dataset provided by iRhythm Technologies, Inc, and therefore no formal ethics approval was required for this study.
Study sample and distribution of wear and analyzable time
Cumulative detection of potentially high-risk arrhythmias
Characteristics of 122,815 eligible continuous ambulatory ECG monitoring records between November 2011 and December 2013
Non-sustained ventricular tachycardia
Sustained ventricular tachycardia
Atrial fibrillation pause
High-grade heart blocka
N = 122,815
N = 22,443
N = 238
N = 1,766
N = 521
N = 249
N = 1,468
Age, years, N (%)
Women, N (%)
Mean (Standard Deviation)
Mean (Standard Deviation)
More than half (53 %) of the recorders were worn by women; but for nearly all arrhythmias except symptomatic pauses, there were more detected arrhythmias among men than women. Although nearly half (49.8 %) of the analyzed patients were for patients aged less than 65 years old, the majority of all detected arrhythmias were among patients aged 65 years or older.
Timing of detection of potentially high-risk arrhythmias
Within a very large, contemporary study analyzing nearly 123,000 long-term continuous ambulatory cardiac monitors that were prescribed between 2011 and 2013, we found a moderate burden of potentially high-risk arrhythmias, including both ventricular arrhythmias and bradyarrhythmias. Patient compliance with extended monitoring was high, with at least 25 % of patients achieving greater than 13 days of continuous monitoring. For all arrhythmias examined, longer monitoring times significantly increased the yield of detected arrhythmias. While the gains in arrhythmia yield were particularly marked in the first 7 days of monitoring, it is notable that the gains continued to increase from days 7 to 14. The relatively high wear and analyzable time for the longer term continuous monitors suggests that outpatient ECG monitoring using this approach is feasible and can have significant yield of clinically important arrhythmias beyond atrial fibrillation. Our study examined a more recent time period than prior investigations of long-term continuous monitors [5, 9] and suggests that changes in the device technology and accumulated operator experience may have resulted in improved patient compliance.
Prior research suggests that traditional 24-h Holter monitoring is not sufficiently long enough to detect many types of arrhythmias [10–13], and recent evidence has demonstrated that longer monitoring may be useful to detect arrhythmias in high-risk patient populations, such as those with a recent history of cryptogenic ischemic stroke [8, 14], although the majority of these studies have focused primarily on finding atrial fibrillation. For example, in a registry of 239 patients who wore 30-day loop recorders after discharge for cryptogenic ischemic stroke, researchers found that 24 % of all detected cases of occult atrial fibrillation were found in the final 10 days of 30-day monitoring (i.e., between days 20 and 30) . Similarly, in a larger controlled trial of a similar patient population where 24-h Holter monitoring was compared to 30-day monitoring, 17 % of all cases of atrial fibrillation were detected in the final week of monitoring . Although conventional wisdom suggests that longer monitoring may be useful for detecting rarer, potentially high-risk arrhythmias, such as ventricular arrhythmias and bradyarrhythmias, there is little empirical evidence on the impact and diagnostic yield of longer continuous monitoring for other clinically meaningful arrhythmias outside of atrial fibrillation. One advantage to the studied technology compared to typical 24- or 48-h Holter monitor systems is its longer continuous wear time up to 14 days, as well as its application without any long wires attached to distant electrodes. For longer monitoring periods, loop or event recorders have typically been the preferred technology, with the main disadvantage being that recordings are only stored if they meet pre-defined algorithms or for symptomatic triggers. A post-hoc investigation of the patient’s rhythm pre-or post-event cannot be done. Implantable loop recorders are occasionally used for very rare arrhythmia events, but these have the same limitations as loop and event monitors and also require a small surgical procedure to implant the device with its attendant risks.
For ventricular arrhythmias, we found that although the majority of arrhythmias are identified in the first 7 days, a significant proportion of arrhythmias were still detected in the 7 to 14-day monitoring window. This was more pronounced for non-sustained VT than sustained VT, with more than 20 % of non-sustained VT being identified in the 7–14 day window. Although we did not have detailed clinical characteristics for our patient population, in high-risk patients, such as those with cardiomyopathy, non-sustained VT can be a high risk marker that may warrant a change in treatment such as the consideration of an implantable cardioverter-defibrillator in certain patient populations (i.e., hypertrophic cardiomyopathy). Depending upon the clinical circumstances, both nonsustained and sustained VT often support the need for further diagnostic testing, such as the evaluation for structural heart disease or for cardiac ischemia. Similarly, although potentially high-risk bradyarrhythmias were less common, if they are not appropriately identified and treated, patients may suffer significant morbidity and excess mortality. The consideration of therapeutic interventions such as permanent pacemaker implantation is recommended by the joint American College of Cardiology, American Heart Association, and Heart Rhythm Society guidelines for the high-risk bradyarrhythmias evaluated in our study .
Our study had certain limitations. We did not have data on any changes in clinical management or patient outcomes following monitoring, so we were unable to delineate the direct clinical impact from the detection of arrhythmias found from the monitors in our study. We did not have information on all symptomatic triggers, and thus did not analyze the proportion of all symptomatic triggers that correlate to true arrhythmias. Patient information was limited to demographic characteristics, and data were unavailable on patients’ comorbidities, which could potentially help further risk stratify patients and allow for predictive modeling to help identify those most at-risk for high-risk arrhythmias. In addition, some bradyarrhythmias, such as asymptomatic sinus and AF pauses, may occur nocturnally in normal subjects. Further, we did not validate the data on the clinical indication for the ordered monitors, and differences among providers’ thresholds for ordering the monitors could have an impact on patient selection and arrhythmia yield. Finally, while average wear time was high, at least a quarter of patients wore the device for less than 7 days (25th percentile of 6.8 days), thus artificially reducing the yield of detected arrhythmias from 7 to 14 days. Thus, the actual yield of detected arrhythmias from days 7 to 14 or monitoring may be even higher than we observed.
In sum, our study suggests that longer term monitoring up to 14 days resulted in high patient compliance, and greater detection of high-risk arrhythmias than 24- or 48-h monitoring strategies. We observed similar findings across gender and age subgroups. Although the bulk of arrhythmias were detected within the first 7 days, longer-term monitoring between 7 and 14 days yielded a significant number of likely clinically meaningful, potentially high-risk arrhythmias. Future research should examine the clinical utility of improved high-risk arrhythmia detection in targeted patient groups and its impact on patient management and associated clinical outcomes.
This project was sponsored by a research grant from iRhythm Technologies, Inc., San Francisco, California.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Crawford MH, Bernstein SJ, Deedwania PC, DiMarco JP, Ferrick KJ, Garson Jr A, et al. ACC/AHA guidelines for ambulatory electrocardiography: executive summary and recommendations. A report of the American College of Cardiology/American Heart Association task force on practice guidelines (committee to revise the guidelines for ambulatory electrocardiography). Circulation. 1999;100(8):886–93.View ArticlePubMedGoogle Scholar
- Katritsis DG, Siontis GC, Camm AJ. Prognostic significance of ambulatory ECG monitoring for ventricular arrhythmias. Prog Cardiovasc Dis. 2013;56(2):133–42.View ArticlePubMedGoogle Scholar
- Liao J, Khalid Z, Scallan C, Morillo C, O’Donnell M. Noninvasive cardiac monitoring for detecting paroxysmal atrial fibrillation or flutter after acute ischemic stroke: a systematic review. Stroke. 2007;38(11):2935–40.View ArticlePubMedGoogle Scholar
- Enseleit F, Duru F. Long-term continuous external electrocardiographic recording: a review. Europace. 2006;8(4):255–66.View ArticlePubMedGoogle Scholar
- Barrett PM, Komatireddy R, Haaser S, Topol S, Sheard J, Encinas J, et al. Comparison of 24-hour Holter monitoring with 14-day novel adhesive patch electrocardiographic monitoring. Am J Med. 2014;127(1):95. e11-97.View ArticlePubMedGoogle Scholar
- Healey JS, Connolly SJ, Gold MR, Israel CW, Van Gelder IC, Capucci A, et al. Subclinical atrial fibrillation and the risk of stroke. N Engl J Med. 2012;366(2):120–9.View ArticlePubMedGoogle Scholar
- Reiffel JA, Schwarzberg R, Murry M. Comparison of autotriggered memory loop recorders versus standard loop recorders versus 24-hour Holter monitors for arrhythmia detection. Am J Cardiol. 2005;95(9):1055–9.View ArticlePubMedGoogle Scholar
- Gladstone DJ, Spring M, Dorian P, Panzov V, Thorpe KE, Hall J, et al. Atrial fibrillation in patients with cryptogenic stroke. N Engl J Med. 2014;370(26):2467–77.View ArticlePubMedGoogle Scholar
- Turakhia MP, Hoang DD, Zimetbaum P, Miller JD, Froelicher VF, Kumar UN, et al. Diagnostic utility of a novel leadless arrhythmia monitoring device. Am J Cardiol. 2013;112(4):520–4.View ArticlePubMedGoogle Scholar
- Seet RC, Friedman PA, Rabinstein AA. Prolonged rhythm monitoring for the detection of occult paroxysmal atrial fibrillation in ischemic stroke of unknown cause. Circulation. 2011;124(4):477–86.View ArticlePubMedGoogle Scholar
- Higgins P, MacFarlane PW, Dawson J, McInnes GT, Langhorne P, Lees KR. Noninvasive cardiac event monitoring to detect atrial fibrillation after ischemic stroke: a randomized, controlled trial. Stroke. 2013;44(9):2525–31.View ArticlePubMedGoogle Scholar
- Schreiber D, Sattar A, Drigalla D, Higgins S. Ambulatory cardiac monitoring for discharged emergency department patients with possible cardiac arrhythmias. West J Emerg Med. 2014;15(2):194–8.PubMed CentralView ArticlePubMedGoogle Scholar
- Rothman SA, Laughlin JC, Seltzer J, Walia JS, Baman RI, Siouffi SY, et al. The diagnosis of cardiac arrhythmias: a prospective multi-center randomized study comparing mobile cardiac outpatient telemetry versus standard loop event monitoring. J Cardiovasc Electrophysiol. 2007;18(3):241–7.View ArticlePubMedGoogle Scholar
- Flint AC, Banki NM, Ren X, Rao VA, Go AS. Detection of Paroxysmal Atrial Fibrillation by 30-Day Event Monitoring in Cryptogenic Ischemic Stroke: The Stroke and Monitoring for PAF in Real Time (SMART) Registry. Stroke. 2012.Google Scholar
- Epstein AE, DiMarco JP, Ellenbogen KA, Estes III NA, Freedman RA, Gettes LS, et al. ACC/AHA/HRS 2008 Guidelines for device-based therapy of cardiac rhythm abnormalities: a report of the American College of Cardiology/American Heart Association Task Force on Practice Guidelines (Writing Committee to Revise the ACC/AHA/NASPE 2002 guideline update for implantation of cardiac pacemakers and antiarrhythmia devices): developed in collaboration with the American Association for Thoracic Surgery and Society of Thoracic Surgeons. Circulation. 2008;117(21):e350–408.View ArticlePubMedGoogle Scholar