This article has Open Peer Review reports available.
Flow mediated dilation of the brachial artery: an investigation of methods requiring further standardization
© Peretz et al; licensee BioMed Central Ltd. 2007
Received: 12 December 2006
Accepted: 21 March 2007
Published: 21 March 2007
In order to establish a consistent method for brachial artery reactivity assessment, we analyzed commonly used approaches to the test and their effects on the magnitude and time-course of flow mediated dilation (FMD), and on test variability and repeatability. As a popular and noninvasive assessment of endothelial function, several different approaches have been employed to measure brachial artery reactivity with B-mode ultrasound. Despite some efforts, there remains a lack of defined normal values and large variability in measurement technique.
Twenty-six healthy volunteers underwent repeated brachial artery diameter measurements by B-mode ultrasound. Following baseline diameter recordings we assessed endothelium-dependent flow mediated dilation by inflating a blood pressure cuff either on the upper arm (proximal) or on the forearm (distal).
Thirty-seven measures were performed using proximal occlusion and 25 with distal occlusion. Following proximal occlusion relative to distal occlusion, FMD was larger (16.2 ± 1.2% vs. 7.3 ± 0.9%, p < 0.0001) and elongated (107.2 s vs. 67.8 s, p = 0.0001). Measurement of the test repeatability showed that differences between the repeated measures were greater on average when the measurements were done using the proximal method as compared to the distal method (2.4%; 95% CI 0.5–4.3; p = 0.013).
These findings suggest that forearm compression holds statistical advantages over upper arm compression. Added to documented physiological and practical reasons, we propose that future studies should use forearm compression in the assessment of endothelial function.
Impaired endothelial function is recognized as an early and modulating process in the pathophysiology of atherosclerotic cardiovascular disease. Endothelial function is often quantified by flow-mediated dilation (FMD), which represents the endothelium-dependent relaxation of a conduit artery-typically the brachial artery – due to an increased blood flow. Brachial artery reactivity is a frequently used non-invasive ultrasonographic assessment of FMD that indicates endothelium-dependent response to shear stress. This measure is a marker for increased cardiovascular risk, and correlates with impaired endothelium-dependent relaxation in the coronary arteries. Due to lack of a standardized method to measure brachial artery reactivity[5, 6], we sought to evaluate the impact of different circulatory occlusion sites (upper arm and forearm occlusion locations) and timing of measurements, in order to establish a consistent method for brachial artery reactivity assessment in a future study of environmental health effects. We assessed the magnitude and temporal characteristics of the response as well as for sources and levels of variability and repeatability of measurements. This study leads to further clarification of impacts of brachial artery reactivity technique and analysis.
Twenty-six volunteers participated in the study. Participants were age 18–49, healthy, normotensive, nonsmokers. The study was approved by the University of Washington Human Subjects Review Committee; subjects provided informed consent for all procedures.
In the first 19 participants, FMD was measured twice on two separate days, at least two weeks apart. In the remaining seven participants, FMD was measured on two separate days, on two occasions on each day separated by one hour for all individuals.
All measurements of brachial artery diameter and FMD were performed in the morning, in a quiet and dark room and at controlled ambient temperatures between 20°C and 26°C. The electrocardiogram (ECG) was continuously monitored. Female participants were examined during the follicular phase of their menstrual cycle. Studies were conducted after an overnight fast of at least 10 hours (water was permitted), with the subjects supine and after 10 minutes of rest. The subject's right arm was comfortably immobilized in the extending position, allowing for ultrasound scanning of the brachial artery 5–10 cm above the antecubital fossa.
In each examination, recording of vessel images were followed by inflation of a cuff to suprasystolic pressure (40 to 50 mmHg above systolic pressure) for 5 minutes as suggested by findings in Corretti et al. Then the brachial artery diameter was imaged and recorded for 3 minutes. The first 19 participants were randomly assigned to cuff inflation either on the forearm, or upper arm. In the remaining seven participants the cuff was inflated on the forearm on the first day and upper arm on the second day. Due to the location of the occluding cuff, the transducer was positioned in a slightly lower position in the studies conducted with proximal occlusion compared to those with distal occlusions.
The brachial artery was scanned in the longitudinal section using an HDI 5000 Ultrasound Instrument (Philips Medical Systems, Bothell, WA) with a 5–12 MHz linear array transducer. One experienced sonographer collected all images. Images were digitized from the video output of the ultrasound machine using a frame grabber under control of custom software on a personal computer. Image acquisition was gated with an ECG signal so that images were captured at end diastole in each cardiac cycle.
Brachial artery diameter was assessed by a single trained operator using an automated, beat-by-beat image processing software package (Vascular Tools 4.6, Medical Imaging Applications, USA). This analysis technique was previously introduced  and independently validated[9, 10]. Briefly, the operator defined a vascular region of interest, which was then applied automatically to identify the media-to-media diameters in each frame over time. We used two measures of quality control: we excluded from the analysis frames with a detected vessel border smaller than 70% of the width of the region of interest (confidence index > 70%) and, for post-cuff deflation sessions only, frames with diameters that differ by more than 1 standard deviation from a polynomial fit.
Descriptive data are presented as mean ± standard error (SE) unless specified otherwise. Dmax, time-to-peak FMD, and Dend were compared between cuff positions using ANOVA with correction for gender and the level to which the blood pressure cuff was inflated (i.e. 40 or 50 mmHg above systolic blood pressure).
Variability and repeatability of the test were measured and compared between the two cuff positions. Assessment of variability included: calculation of the coefficient of variation (CV) of the baseline diameters from accepted frames to determine the within-session variability; mean CV from 2 or more baseline measurements for each of the first 19 participants as a day-to-day variability. Both descriptors were calculated for the two cuff locations and then compared by independent t test.
Repeatability for FMD analysis for each cuff location was calculated using intraclass correlation coefficient (ICC) in a two-way mixed effects model, and Bland & Altman plots with calculation of the repeatability coefficient. SPSS 12.0 (Chicago, Illinois) statistical software and Microsoft Excel were used.
In addition, a linear model was run to assess whether the absolute values of the difference in FMD between test and retest were different using the proximal versus the distal circulatory occlusion. To account for the correlation between distal and proximal comparisons done on same subjects, a generalized estimating equations (GEE) approach was used, assuming an exchangeable correlation structure. S-plus 7.0 (Insightful Corp. 2005) statistical software was used. Results from all tests were considered statistically significant if p <0.05.
A total of 62 ultrasound examinations of the brachial artery followed by assessment of endothelium-dependent flow mediated vasodilation were analyzed in this paper, 37 with upper arm cuff location, and 25 with forearm cuff location. Participants' mean age was 28.4 (range, 20–42 yr). More male (N = 16) than female (N = 10) participated in the study.
Overall, mean baseline brachial artery diameter from 37 sessions with upper arm occlusion was smaller (3.8 ± 0.1) than that from 25 sessions acquired with forearm occlusion (4.2 ± 0.2) (p = 0.058). This difference between baseline diameters appears to be related to the slightly different location of the ultrasound transducer on the participants' arm. In fact, similar results were obtained after limiting analysis to those participants that were evaluated with both proximal and distal occlusion (data not shown). Mean number of frames obtained for baseline diameter assessment was similar for both proximal and distal cuff locations (81.7 vs. 81.5 respectively).
Flow Mediated Dilation
Pooling all the brachial artery reactivity tests together, we found that peak FMD using the upper arm (N = 37) was greater than that for the forearm (N = 25) (16.2 ± 1.2% vs. 7.3 ± 0.9, p < 0.0001). The gender and the pressure to which the cuff was inflated did not alter these results (data not shown).
On average, the time-to-peak FMD was 107.2 ± 6 seconds for upper arm cuff inflation and 67.8 ± 8.9 seconds for forearm cuff inflation (p = 0.0001). After peak dilation, the artery diameter decrements were slower with proximal occlusion than distal occlusion. At the end of the three-minute image capture period after cuff release, Dend was 11.3 ± 1.2% with proximal cuff occlusion, and 3.3 ± 0.8% with the distal cuff occlusion (p = 0.0001).
Variability and Repeatability
For the 62 studies for which ~80 baseline frames were taken in a single session, the within-session diameter variability was small (CV = 0.85 ± 0.04%); there was no statistically significant difference between the upper arm cuff location (CV= 0.85 ± 0.05%) and the more distal location (CV= 0.84 ± 0.06%). Day-to-day variability of baseline diameter was also small for both the distal cuff placement (CV = 3.4 ± 1.3%) and the proximal one (CV = 5.1 ± 1.7%), with no statistically significant difference between them.
The intraclass correlation coefficient (ICC) between peak-FMD of two repeated measures was 0.6 (95%CI -0.2:0.8) for the upper arm occlusions, and 0.6 (95%CI -0.7:0.9) for the forearm occlusion. Using the GEE approach we found that the magnitude of the differences between the repeated measures were on average 2.4% (95% CI 0.5–4.3; p = 0.013) greater when the measurements were done using the proximal method as compared to the distal method.
In this study we evaluated statistical reasons to support application of specific conditions to FMD testing on the brachial artery. While assessment of FMD of the brachial artery either with proximal (upper arm), or distal (forearm) occlusion has been found to be correlated with coronary endothelial function[12, 13], we evaluated whether other test characteristics may provide a basis for selecting one approach over the other.
We have confirmed the well established finding that the maximal FMD is greater and delayed with upper arm occlusion[6, 14–20]. We showed a significantly greater and delayed FMD following proximal occlusion (16.2% and 107.2 seconds, respectively) compared to distal occlusion (7.3% and 67.8 seconds). The difference in the magnitude of FMD between occlusion sites (8.9%) is greater than the difference reported in a recent meta-analysis (2.4%). This discrepancy may be due to three major reasons. First, the measures were made in populations with different baseline endothelial function. Second, differences in baseline diameters in each of the two occlusion sites, may have affected the shear stress, the stimulus for endothelium-dependent dilation. In this study, shear stress was not analyzed. Finally, our use of an automated method of analysis with all available frames. Others studies that demonstrated greater FMD in response to proximal occlusion rather than distal occlusion [14–20] captured brachial artery images continuously, but then examined frames from specific time-points to define the artery's diameter and the frequency of these intervals was not consistent between studies.
The time to maximal FMD may vary not only by occlusion site, but also by the method used to detect the diameter from captured frames. For example, adhering to the recommended time interval to maximal FMD at 60 seconds after the release of the occlusive cuff  would have resulted in missing the true peak-FMD of 40% of the measurements we obtained with forearm occlusion (figure 3b), and about 95% of the cases of upper arm occlusion (figure 3a). Random measurement errors and underestimation of the actual FMD are less likely when examination of brachial artery dilation is done continuously over the entire course of the session and analyzed from all available images with automated computer based edge detection.
The assessment of FMD of the brachial artery is a functional bioassay for in vivo endothelial function in humans. It has been stated that the primary mechanism responsible for vessel dilation with forearm occlusion is endothelial release of nitric oxide(NO). Doshi et al. demonstrated that most of the dilatory response to proximal occlusion is NO-independent. Additional mechanisms, such as direct effect of hypoxia on the smooth muscle, altered myogenic response, or altered release of other mediators involved in vessel reactivity, may be involved in the dilation of the brachial artery following upper arm occlusion. NO-mediated vasodilation occurs ~60–80 seconds following reperfusion[24, 25]. Therefore, in the absence of a pharmacologic intervention, the increased time to maximal FMD and the prolonged dilatory response (Dend 11.3% vs. 3.3% p <0.0001) for proximal occlusion relative to distal occlusion may reflect other underlying mechanisms. NO is not the only mediator involved in FMD following proximal and distal occlusions; other endothelium-derived factors are probably involved differently with the two occlusion locations. It is essential to conduct validation studies in order to understand the role of different components of endothelial function in one approach or the other of the test.
Many studies have assessed the variability of the brachial artery diameter measurements, both at baseline and during the dilatory response. While the baseline diameter was shown to have relatively small variability (CV range 1–13%), the maximal FMD was associated with a much greater range of variability (CV range 1–84%). Various factors may contribute to the variability of brachial artery diameter, and these can be largely divided into technical and physiological factors. Ultrasound of the brachial artery is technically challenging, and measurement errors – even in magnitude of fraction of millimeters – especially when added to effect of intrinsic factors affecting FMD, may affect the accuracy of the test. Therefore, it is essential to apply a rigorous methodology to the test. In doing so, we utilized a single ultrasound machine, all studies were performed by one technician, and all images were read by one off-line reader. We found relatively low within-session variability and day-to-day variability of baseline diameter with no difference between proximal and distal placements. Similar day-to-day variability was found by Uehata et al. Greater variability of peak FMD is shown by the relatively high fraction of participants (more than a third, in both occlusion sites) that did not reach their maximal FMD by the average time-to-peak FMD (figure 3). The small impact of technical factors on baseline brachial artery diameter variability may indicate that physiological factors contribute most of peak FMD variability as suggested previously.
In this study we did not quantify the shear stress. It has been observed that the shear stress computed as area under the curve, is the major contributor to the magnitude of FMD. Therefore it is plausible that variability in shear stress may affect variability and repeatability of FMD, and understanding this interplay requires additional study.
We also assessed the repeatability of FMD, an important quality of this outcome as a potential tool in clinical studies where repeated measures are needed, and as a potential indicator for the risk of future coronary disease. Previous studies have shown inconsistent results of brachial artery FMD repeatability. Malik at al reported poor repeatability of FMD with an intraclass correlation coefficient of 0.1. In contrast, Welsch et al reported an intraclass correlation coefficient of 0.92 comparing day-to-day FMD. We demonstrated a reasonable repeatability of FMD measurements. Repeatability of FMD was better for the distal cuff location as shown by the smaller coefficient of repeatability (CR = 10.1% vs. 14.1%) and the greater difference on average of two repeated measurements with upper arm cuff placement (2.4%). Even with accurate methodology, our repeatability was only modest; suggesting that some degree of measurement error is inevitable in any setting. However, it is reasonable to assume that studies that use more than one sonographer or are performed in more than one center may be subject to a greater degree of measurement error than observed in this study. Hence, such studies must be extremely well coordinated, using highly trained technical staff, and probably employ a larger sample size than single site studies. The use of duplicate measurements may be important in reducing measurement error, as evidence show that repetitive reactive hyperemia does not effect FMD measurements , though the minimum time between measurements has not been clearly identified.
For brachial artery reactivity to be a widespread clinical and research tool, the flow mediated dilation measurement of the brachial artery should be standardized, and centers should adhere to a more consistent analysis protocol. Our data suggest that the choice of the location of cuff inflation may have statistical implications that favor the forearm approach. There are established physiological reasons that support employing one approach over the other. While FMD following distal occlusion is mainly NO dependent, other – not as well elucidated – factors are involved in the dilatory response to upper arm occlusion. In future studies we will study the effects of an environmental insult on FMD of the brachial artery, while measuring different components of endothelial function. In the absence of a well characterized mechanism involved in the flow mediated dilation of the brachial artery with upper arm occlusion, based on our results, we suggest that the forearm occlusion be used. The cuff should be inflated five cm distal to the elbow for five minutes, then, in the post-hyperemic phase of the test the brachial artery should be scanned continuously. Post-cuff deflation examination should be continued for three minutes in healthy adults, whereas, in adults at risk for CAD or with known disease, it is likely necessary to capture the artery's images for longer periods. Last, we suggest also that the images be analyzed with an automated computer-based edge detection algorithm. This will allow for a time series analysis of all brachial artery diameters and identification of the true maximal FMD. More studies performed with standardized method are needed to confirm our findings and allow for potential future use of the test for clinical purposes.
Support for this study was provided by grant R830954 and R827355 from the Environmental Protection Agency and grant M01RR-00037 from the National Institute of Health and ES013195 from the National Institute for Environmental Health Sciences.
- Asselbergs FW, van der Harst P, Jessurun GA, Tio RA, van Gilst WH: Clinical impact of vasomotor function assessment and the role of ACE-inhibitors and statins. Vascul Pharmacol. 2005, 42 (3): 125-140. 10.1016/j.vph.2005.01.009.View ArticlePubMedGoogle Scholar
- Celermajer DS, Sorensen KE, Gooch VM, Spiegelhalter DJ, Miller OI, Sullivan ID, Lloyd JK, Deanfield JE: Non-invasive detection of endothelial dysfunction in children and adults at risk of atherosclerosis. Lancet. 1992, 340 (8828): 1111-1115. 10.1016/0140-6736(92)93147-F.View ArticlePubMedGoogle Scholar
- Kuvin JT, Patel AR, Sliney KA, Pandian NG, Rand WM, Udelson JE, Karas RH: Peripheral vascular endothelial function testing as a noninvasive indicator of coronary artery disease. J Am Coll Cardiol. 2001, 38 (7): 1843-1849. 10.1016/S0735-1097(01)01657-6.View ArticlePubMedGoogle Scholar
- Takase B, Uehata A, Akima T, Nagai T, Nishioka T, Hamabe A, Satomura K, Ohsuzu F, Kurita A: Endothelium-dependent flow-mediated vasodilation in coronary and brachial arteries in suspected coronary artery disease. Am J Cardiol. 1998, 82 (12): 1535-9, A7-8. 10.1016/S0002-9149(98)00702-4.View ArticlePubMedGoogle Scholar
- Kuvin JT, Patel AR, Karas RH: Need for standardization of noninvasive assessment of vascular endothelial function. Am Heart J. 2001, 141 (3): 327-328. 10.1067/mhj.2001.113221.View ArticlePubMedGoogle Scholar
- Roman MJ, Naqvi TZ, Gardin JM, Gerhard-Herman M, Jaff M, Mohler E: Clinical application of noninvasive vascular ultrasound in cardiovascular risk stratification: a report from the American Society of Echocardiography and the Society of Vascular Medicine and Biology. J Am Soc Echocardiogr. 2006, 19 (8): 943-954. 10.1016/j.echo.2006.04.020.View ArticlePubMedGoogle Scholar
- Corretti MC, Plotnick GD, Vogel RA: Technical aspects of evaluating brachial artery vasodilatation using high-frequency ultrasound. Am J Physiol. 1995, 268 (4 Pt 2): H1397-404.PubMedGoogle Scholar
- Sonka M, Liang W, Lauer RM: Automated analysis of brachial ultrasound image sequences: early detection of cardiovascular disease via surrogates of endothelial function. IEEE Trans Med Imaging. 2002, 21 (10): 1271-1279. 10.1109/TMI.2002.806288.View ArticlePubMedGoogle Scholar
- Mancini GB, Yeoh E, Abbott D, Chan S: Validation of an automated method for assessing brachial artery endothelial dysfunction. Can J Cardiol. 2002, 18 (3): 259-262.PubMedGoogle Scholar
- Preik M, Lauer T, Heiss C, Tabery S, Strauer BE, Kelm M: Automated ultrasonic measurement of human arteries for the determination of endothelial function. Ultraschall Med. 2000, 21 (5): 195-198. 10.1055/s-2000-7989.View ArticlePubMedGoogle Scholar
- Bland JM, Altman DG: Statistical methods for assessing agreement between two methods of clinical measurement. Lancet. 1986, 1 (8476): 307-310.View ArticlePubMedGoogle Scholar
- Anderson TJ, Uehata A, Gerhard MD, Meredith IT, Knab S, Delagrange D, Lieberman EH, Ganz P, Creager MA, Yeung AC, et al: Close relation of endothelial function in the human coronary and peripheral circulations. J Am Coll Cardiol. 1995, 26 (5): 1235-1241. 10.1016/0735-1097(95)00327-4.View ArticlePubMedGoogle Scholar
- Matsuo S, Matsumoto T, Takashima H, Ohira N, Yamane T, Yasuda Y, Tarutani Y, Horie M: The relationship between flow-mediated brachial artery vasodilation and coronary vasomotor responses to bradykinin: comparison with those to acetylcholine. J Cardiovasc Pharmacol. 2004, 44 (2): 164-170. 10.1097/00005344-200408000-00004.View ArticlePubMedGoogle Scholar
- Doshi SN, Naka KK, Payne N, Jones CJ, Ashton M, Lewis MJ, Goodfellow J: Flow-mediated dilatation following wrist and upper arm occlusion in humans: the contribution of nitric oxide. Clin Sci (Lond). 2001, 101 (6): 629-635.View ArticleGoogle Scholar
- Berry KL, Skyrme-Jones RA, Meredith IT: Occlusion cuff position is an important determinant of the time course and magnitude of human brachial artery flow-mediated dilation. Clin Sci (Lond). 2000, 99 (4): 261-267.View ArticleGoogle Scholar
- Betik AC, Luckham VB, Hughson RL: Flow-mediated dilation in human brachial artery after different circulatory occlusion conditions. Am J Physiol Heart Circ Physiol. 2004, 286 (1): H442-8. 10.1152/ajpheart.00314.2003.View ArticlePubMedGoogle Scholar
- Agewall S, Doughty RN, Bagg W, Whalley GA, Braatvedt G, Sharpe N: Comparison of ultrasound assessment of flow-mediated dilatation in the radial and brachial artery with upper and forearm cuff positions. Clin Physiol. 2001, 21 (1): 9-14. 10.1046/j.1365-2281.2001.00302.x.View ArticlePubMedGoogle Scholar
- Bots ML, Westerink J, Rabelink TJ, de Koning EJ: Assessment of flow-mediated vasodilatation (FMD) of the brachial artery: effects of technical aspects of the FMD measurement on the FMD response. Eur Heart J. 2005, 26 (4): 363-368. 10.1093/eurheartj/ehi017.View ArticlePubMedGoogle Scholar
- Mannion TC, Vita JA, Keaney JF, Benjamin EJ, Hunter L, Polak JF: Non-invasive assessment of brachial artery endothelial vasomotor function: the effect of cuff position on level of discomfort and vasomotor responses. Vasc Med. 1998, 3 (4): 263-267.View ArticlePubMedGoogle Scholar
- Vogel RA, Corretti MC, Plotnick GD: A comparison of brachial artery flow-mediated vasodilation using upper and lower arm arterial occlusion in subjects with and without coronary risk factors. Clin Cardiol. 2000, 23 (8): 571-575.View ArticlePubMedGoogle Scholar
- Pyke KE, Tschakovsky ME: The relationship between shear stress and flow-mediated dilatation: implications for the assessment of endothelial function. J Physiol. 2005, 568 (Pt 2): 357-369. 10.1113/jphysiol.2005.089755.View ArticlePubMedPubMed CentralGoogle Scholar
- Corretti MC, Anderson TJ, Benjamin EJ, Celermajer D, Charbonneau F, Creager MA, Deanfield J, Drexler H, Gerhard-Herman M, Herrington D, Vallance P, Vita J, Vogel R: Guidelines for the ultrasound assessment of endothelial-dependent flow-mediated vasodilation of the brachial artery: a report of the International Brachial Artery Reactivity Task Force. J Am Coll Cardiol. 2002, 39 (2): 257-265. 10.1016/S0735-1097(01)01746-6.View ArticlePubMedGoogle Scholar
- Tschakovsky ME, Pyke KE: Counterpoint: Flow-mediated dilation does not reflect nitric oxide-mediated endothelial function. J Appl Physiol. 2005, 99 (3): 1235-7; discussion 1237-8. 10.1152/japplphysiol.00607.2005.View ArticlePubMedGoogle Scholar
- Joannides R, Haefeli WE, Linder L, Richard V, Bakkali EH, Thuillez C, Luscher TF: Nitric oxide is responsible for flow-dependent dilatation of human peripheral conduit arteries in vivo. Circulation. 1995, 91 (5): 1314-1319.View ArticlePubMedGoogle Scholar
- Lieberman EH, Gerhard MD, Uehata A, Selwyn AP, Ganz P, Yeung AC, Creager MA: Flow-induced vasodilation of the human brachial artery is impaired in patients <40 years of age with coronary artery disease. Am J Cardiol. 1996, 78 (11): 1210-1214. 10.1016/S0002-9149(96)00597-8.View ArticlePubMedGoogle Scholar
- West SG, Wagner P, Schoemer SL, Hecker KD, Hurston KL, Likos Krick A, Boseska L, Ulbrecht J, Hinderliter AL: Biological correlates of day-to-day variation in flow-mediated dilation in individuals with Type 2 diabetes: a study of test-retest reliability. Diabetologia. 2004, 47 (9): 1625-1631. 10.1007/s00125-004-1502-8.View ArticlePubMedGoogle Scholar
- Moens AL, Goovaerts I, Claeys MJ, Vrints CJ: Flow-mediated vasodilation: a diagnostic instrument, or an experimental tool?. Chest. 2005, 127 (6): 2254-2263. 10.1378/chest.127.6.2254.View ArticlePubMedGoogle Scholar
- Uehata A, Lieberman EH, Gerhard MD, Anderson TJ, Ganz P, Polak JF, Creager MA, Yeung AC: Noninvasive assessment of endothelium-dependent flow-mediated dilation of the brachial artery. Vasc Med. 1997, 2 (2): 87-92.PubMedGoogle Scholar
- Hijmering ML, Stroes ES, Pasterkamp G, Sierevogel M, Banga JD, Rabelink TJ: Variability of flow mediated dilation: consequences for clinical application. Atherosclerosis. 2001, 157 (2): 369-373. 10.1016/S0021-9150(00)00748-6.View ArticlePubMedGoogle Scholar
- Pyke KE, Tschakovsky ME: Peak vs. total reactive hyperemia: which determines the magnitude of flow mediated dilation?. J Appl Physiol. 2006Google Scholar
- Malik J, Wichterle D, Haas T, Melenovsky V, Simek J, Stulc T: Repeatability of noninvasive surrogates of endothelial function. Am J Cardiol. 2004, 94 (5): 693-696. 10.1016/j.amjcard.2004.05.049.View ArticlePubMedGoogle Scholar
- Welsch MA, Allen JD, Geaghan JP: Stability and reproducibility of brachial artery flow-mediated dilation. Med Sci Sports Exerc. 2002, 34 (6): 960-965. 10.1097/00005768-200206000-00009.View ArticlePubMedGoogle Scholar
- Harris RA, Padilla J, Rink LD, Wallace JP: Variability of flow-mediated dilation measurements with repetitive reactive hyperemia. Vasc Med. 2006, 11 (1): 1-6. 10.1191/1358863x06vm641oa.View ArticlePubMedGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2261/7/11/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.