- Research article
- Open Access
- Open Peer Review
This article has Open Peer Review reports available.
The use of implantable cardioverter defibrillators in Iceland: a retrospective population based study
© Leosdottir et al; licensee BioMed Central Ltd. 2006
Received: 24 January 2006
Accepted: 24 May 2006
Published: 24 May 2006
Indications for implantable cardioverter defibrillator (ICD) implantation have expanded considerably in recent years, resulting in steadily growing numbers of ICD recipients worldwide. The aim of this study was to review the overall experience with ICDs in Iceland.
This was a retrospective single centre study set at the University Hospital in Iceland. Data on all ICD implantations in Iceland from the first implantation in 1992 till the end of 2002 was reviewed.
Sixty-two patients (71% male) received an ICD during this period. There was an increase in the number of implants by year and the number of new implants in 2001 and 2002 amounted to 56 and 38 per million, respectively. The mean age at implantation was 58 (+/-14) years. Forty patients (65%) had coronary artery disease. The most common indications for ICD implantation were cardiac arrest, 32 (52%) and another 26 (42%) had experienced ventricular tachycardia without cardiac arrest. The most common adverse event was inappropriate shocks. Twenty-eight patients (45%) received therapy from their ICDs, with the majority receiving appropriate therapy. Of the thirteen patients deceased before or during the study period, no case of sudden arrhythmic death was observed.
This study shows that the experience with ICDs in Iceland is in most respects similar to other Western countries.
The use of implantable cardioverter defibrillators (ICDs) for management of ventricular arrhythmias has become an increasingly popular option since Mirowsky and colleagues first introduced the device in 1980 . Following the publication of several large multi-center trials in the last few years, indications for implantation have expanded considerably and now include prophylactic implantation for patients with coronary artery disease and a severely depressed ejection fraction [2–4]. This has resulted in a rise in implantation rates and thus steadily growing numbers of ICD recipients worldwide, especially in the United States and Western Europe .
The objective of this study was to review the experience with ICDs in Iceland where the first ICD implant was performed in 1992. On account of Iceland's small population of approximately 300.000, all of these patients attend the same outpatient clinic at the University Hospital in Reykjavik. Therefore, we were able to review information on every implantation that had been done in the country from the beginning, and as such, examine the experience with ICD therapy in a whole population setting.
Design and subjects
This was a retrospective study, set at the University Hospital in Reykjavik, reviewing all ICD implantations in Iceland from the first implantation in 1992 to the end of year 2002. The study was reviewed and approved by the Institutional Review Committee of Landspitali University Hospital and the Icelandic Data Protection Authority, and complied with the declaration of Helsinki. Informed consent was received from all patients who were alive at the time of the study.
Data collection and analysis
The following data was collected: Age at implantation; gender; presence or absence of structural heart disease; indication for implantation; ejection fraction (EF) prior to implantation; length of the surgical procedure; type of procedure; surgical complications; adverse events; length of follow up; therapy delivered by the device; malfunctions; reasons for reoperations and causes of death for deceased patients. Information was retrieved from the University Hospital medical records and the ICD outpatient clinic records. Information on EF was retrieved from the latest echocardiogram or ventriculogram performed within one year prior to implantation. When EF was recorded by echocardiography only an estimate was given instead of a single number. For this reason we categorized the EF into four categories: less than 20%, 21–40%, 41–60% and more than 60%. Causes of death were retrieved from the National Death Registry.
When evaluating the cause of delivered therapy by the ICD, stored documentation of shocks and antitachycardia pacing-therapy (ATP) was reviewed. Stored electrograms where available in the the devices implanted in Iceland after1994. As such the majority of ICDs in the patients studied had stored electrograms available. Therapy was categorized as appropriate, inappropriate or indeterminate. Therapy on account of malfunction of the device or documented tachyarrhythmia of atrial origin was defined as inappropriate. Therapy was defined as indeterminate if it could not be categorized as either appropriate or inappropriate. All patients on whom events were evaluated had either stored electrograms or printouts of or event recordings showing a plot of RR intervals during an event. If neither was present the episode was classified as indeterminate with regards to outcome. A single investigator evaluated the stored electrograms and RR plots, with consultation from a second investigator when deemed neccessary.
Therapy was considered effective if the ventricular rate after shock/ATP delivery was <65% of the ventricular rate before shock/ATP delivery. Therapy was classified as causing acceleration of the arrhythmia if ventricular rate was ≥ than 20% faster after shock/ATP delivery than prior to therapy. All ATP given for the same episode of a ventricular tachyarrhythmia was considered a single therapy. Assessment of effectiveness in such episodes was confined to outcome of the last ATP-therapy given. All data are presented as means ± SD.
Patient demographic and clinical data
Age at implantation (years)
58 +/- 14
Underlying cardiac disease
Coronary artery disease (CAD)
VT/VF without any known cardiac illness
Suspected drug related ventricular arrhythmia
Coronary artery spasm resulting in VT
Indication for implantation
Cardiac arrest due to VT or VF*
VT without loss of consciousness
Syncope, VT/VF induced at EPS*
EF prior to implantation*
Indications for ICD implantations, classified in accordance to AHA/ACC Implantation of Pacemaker and Antiarrhythmia Guidelines, were cardiac arrest due to ventricular fibrillation (VF) or ventricular tachycardia (VT) in thirty-two patients (52%), VT without cardiac arrest in twenty-six patients (42%) and syncope with inducible VT/VF at an electrophysiological study (EPS) in four patients (6%) . No primary prophylactic implantation was performed.
The majority of the patients 40 (65%) had coronary artery disease (CAD). Seven patients (11%) had long QT-syndrome (LQTS) and six (10%) had dilated cardiomyopathy. The following cardiac disease was present in one patient each: hypertrophic cardiomyopathy, coronary artery spasm (refractory to vasodilator therapy and with polymorphic VT) and cardiac amyloidosis. Seven patients (11%) suffered from VT or VF without any demonstrable cardiac illness. Thirty-six patients (60%) had an EF over 40%, including 24 who had an EF >60%. Two deceased patients had no retrievable records of EF.
The length of the operations decreased during the study period. In the years 1992–1997 the mean operating time were 2:21 hours, compared to 1:20 hours in the years 1998–2002.
All operations were done with a non-thoracotomy approach. Of the sixty-one primary implantations and ten re-operations where a new device was implanted, sixty-four (90%) were implanted subpectorally and seven (10%) under the rectus abdominis muscle. The tranvenous lead insertion was usually via the subclavian vein (89%) but on rare occasions the lead was inserted through the cephalic vein (11%).
Indications for reoperations.
Related to leads:
Decreased sensing in leads
Lead or lead insulation breakdown
Dislodgement of lead
Inappropriate shocks without apparent damage in leads/device
Loss of connection to device after shock
Surgical complications and long-term adverse events
Short term (less than 30 days post operatively) and long term (more than 30 days post operatively) adverse events (AE).
Short term AE (less than 30 days post operatively):
no. of patients
Dislodgement of lead
Long term AE (more than 30 days post operatively):
due to atrial fibrillation/flutter
due to lead malfunction
without apparent cause
Decreased sensing/increased pacing threshold
Dislodgement of device
Discomfort around implant site
Loss of connection to device after shock
The most common long-term adverse events were inappropriate shocks. In the cases of inappropriate shocks without apparent cause or because of lead malfunction, a new device and/or leads were implanted. Other long-term adverse events were decreased sensing or increased pacing threshold (8), dislodgement of the device (migrated to the axilla) (1), transient discomfort around implant site (1) and in one case there was loss of communication with the device via the programmer wand. All of these cases except the case of discomfort around the implant site resulted in re-operations.
There were 17 dual chamber devices were (24%) and 54 single chamber devices (76%) implanted. All the patients were followed every 3 months at the same device clinic. The mean follow up time was 29.1 +/-18.5 months (range 0.2–154.9 months).
The majority (84%) of the 25 patients had appropriate therapy at some time from their device. When reviewing effectiveness, 69% of the shocks and 48% of the ATP treatments were effective in reducing the ventricular rate by ≥ 35% which we defined as successful. None of the appropriate shocks caused acceleration of the arrhythmia, but 5% of the appropriate ATP treatments caused acceleration, eventually leading to a shock.
Out of the twenty-five patients receiving treatment, ten (40%) at some point received inappropriate shocks and/or ATP-treatment. The median number of inappropriate shocks was 4 (range 1–33). Four out of these ten patients received inappropriate shocks because of either device or lead malfunction. The remaining six received inappropriate shocks because of atrial arrhythmias. Three of those six patients had dual chamber ICDs. Three patients received the vast majority of inappropriate shocks; 33, 30 and 17 shocks each (83% of the total number of inappropriate shocks). Two received multiple inappropriate shocks because of atrial fibrillation, including one with a dual chamber device. That patient underwent AV-nodal ablation and has not received an inappropriate shock after this. The third patient received 33 shocks on the same day because of device malfunction.
In the majority of cases, inappropriate treatments due to atrial arrhythmias did not result in cardioversion to sinus rhythm. On one occasion, an inappropriate shock for atrial fibrillation resulted in ventricular fibrillation. Three inappropriate ATP-treatments lead to shocks by acceleration of the ventricular rate in atrial fibrillation into the VF zone.
Thirteen patients with an ICD died prior to the end of 2002. Eight patients (62%) had a cardiac cause of death; three from acute myocardial infarction resulting in acute heart failure and five from end stage heart failure. No case of sudden arrhythmic death was observed. Two patients died from cancer, two from subarachnoidal haemorrhage and one from bacterial sepsis, unrelated to the device.
In this retrospective study we reviewed sixty-one primary ICD implantations and nineteen re-operations performed in Iceland from the first implantation in April 1992 till the end of December 2002. Major complications and serious adverse events were rare, with the most common long-term adverse event being inappropriate shocks. The majority of patients that received therapy from the device had such appropriately, with 69% of the shocks and 48% of the ATP treatments effectively terminating the arrhythmia.
The main strengths of this study are that it examines the experience with ICD treatment in the Icelandic population as a whole, including every single ICD implantation in the country from the first implantation in 1992 till the end of year 2002. This gives an opportunity to evaluate the effectiveness of ICD therapy in this small community. At the same time the study's major limitation is probably its small sample size and retrospective design.
The age distribution of our patient population was similar to previous studies on this subject [6–9]. The percentage of women was somewhat higher than observed in other studies, or 29% compared to 10–20% [6–9]. Sixty-two percent of our patient population had coronary artery disease as an underlying cardiac illness, which is somewhat lower than observed in other studies [6, 7, 9–11]. This might be explained by the relatively many cases of long QT syndrome, which is more common in women  and usually diagnosed at a younger age, but in our study six patients (10%) had this diagnosis.
The mean EF of our population seemed to be somewhat higher than observed in previous studies [7, 8, 11]. Forty percent of our patient population had normal ejection fractions and only two patients had an EF lower than 20%.
Distribution of indications according to the AHA/ACC guidelines was similar to previous experiences [6–8, 10]. Implantation rates increased rapidly over the ten-year period as was to be expected. When comparing implantation rates in Iceland to other European countries, implants in Iceland account to 56 and 38 per million population in the last two years studied, respectively, which is comparable to the current implant rates of around 45 per million in Western Europe [5, 13].
In our study, all of the operations were done by non-thoracotomy approach. No serious infections resulting in device explantation were seen during the study period. Infection rates in ICD patients have been reported to be from 0–8% [14–16]. The fact that all the operations in this study were performed in an operating theatre, as opposed to an electrophysiology laboratory and a rigorous prophylactic antibiotics scheme applied may help explain the low infection rate seen here.
The incidence of haematomas and pleural effusions was somewhat higher than observed in previous studies [6, 8, 14]. All of the devices were implanted subpectorally or under the rectus abdominis, and this may possibly explain the higher incidence of haematomas and the low infection rate.
The majority of re-operations were performed because of leads problems, most often because of decreased sensing of the lead. In some instances lead malfunction led to inappropriate shocks. Our experience is consistent with previous studies, indicating that lead-related problems account for the large proportion of long term ICD complications [8, 15–17].
Inappropriate shocks were the most common long-term adverse event encountered in our study, but 40% (10 out of 25) of the patients receiving shocks, received inappropriate therapy at some time (shocks and/or ATP-therapy). As well as sometimes causing serious physical discomfort for the patient, adverse psychological consequences of multiple shocks can occur. Previous studies have reported worse outcomes on measures of anxiety, depression and quality of life for ICD patients receiving frequent shocks compared to those receiving no or few shocks [19, 20]. On the other hand, quality of life in the ICD patient population as a whole has been shown to be comparable to or even superior to medically treated patients with malignant arrhythmias and patients with pacemakers [19, 21].
In this study three out of six patients receiving inappropriate therapy because of atrial arrhythmias had dual chamber devices. Studies comparing single chamber devices to dual chamber devices with regard to inappropriate therapy for atrial arrhythmias have shown ambiguous results [22, 23]. Currently, a large prospective multicentre randomised trial is underway, designed to analyse the ability of dual chamber ICDs to reduce adverse events, including inappropriate therapy, compared with single chamber devices .
Advanced detection algorhythms were activated in the devices during implant in those with known preexisting paroxysmal atrial arrhythmias. These patients all had a dual chamber device implanted. For those who had no prior history of atrial arrhythmias, advanced detection algorhythms were not activated at implant. For those who received inappropriate shocks due to atrial arrhythmias during follow up, advanced detection criteria were activated if the patients had a dual chamber device. Additionally this led to anti-arrhythmic drug therapy if the patient had not been taking such drugs prior to the inappropriate shock.
The vast majority of treated patients received appropriate therapy at some time, with 69% of the shocks and 48% of the ATP treatments being effective in terminating the ventricular arrhythmia. This success rate is somewhat less than previous studies have shown, where as high as 83% success for shock treatment and 78–94% for ATP treatment has been reported [25–28]. However, patient selection, length of follow up, methods of classifying appropriateness and success of therapy vary considerably between studies, making it difficult to reliably compare results.
No death was observed on account of sudden arrhythmic death. While this may indicate the effectiveness of the device to treat life threatening arrhythmias it must also be taken into consideration that that only thirteen patients died during or were deceased before the study period, which may limit the extent to which conclusions can be drawn regarding this.
In this retrospective study reviewing all ICD implantations in Iceland over a ten year period, we observed that implantation rates have been increasing to a similar extent as in other Western European countries. Therapy received was most often appropriate, with an acceptable success rate in terminating the malignant arrhythmia. Taken together, this study indicates that experience with ICDs in Iceland is comparative to other Western countries, with the devices delivering effective and potentially life-saving therapy to many of the patients. As such, the current study supports the expanding use of ICD therapy for patients suffering from malignant arrhythmias. At the same time we observed that lead problems are not infrequent and inappropriate therapy remains a problem. This underscores the need to focus on improving lead function quality as well as increasing the devices' arrhythmia analysing capability, to minimize the risk of patients receiving inappropriate therapy.
The authors would like to thank the staff at the pacemaker and ICD clinic at Landspitali University Hospital, for all their help. The study was supported by grants from Landspitali University Hospital Science Fund, Reykjavik, Iceland and the Sigurlidi Kristjansson and Helga Jonsdottir Memorial Foundation.
- Mirowski M, Reid PR, Mower MM, Watkins L, Gott VL, Schauble JF, Langer A, Heilman MS, Kolenik SA, Fischell RE, Weisfeldt ML: Termination of malignant ventricular arrythmias with an implanted automatic defibrillator in human beings. N Engl J Med. 1980, 303: 322-324.View ArticlePubMedGoogle Scholar
- ACC/AHA Task Force: ACC/AHA/NASPE 2002 Guideline update for implantation of cardiac pacemakers and antiarrhythmia devices: Summary article. Circulation. 2002, 106: 2145-2161. 10.1161/01.CIR.0000035996.46455.09.View ArticleGoogle Scholar
- Moss AJ, Zareba W, Hall WJ, Klein H, Wilber DJ, Cannom DS, Daubert JP, Higgins SL, Brown MW, Andrews ML, Multicenter Automatic Defibrillator Implantation Trial II Investigators: Prophylactic implantation of a defibrillator in patients with myocardial Infarction and reduced ejection fraction (MADIT II). N Engl J Med. 2002, 346: 877-883. 10.1056/NEJMoa013474.View ArticlePubMedGoogle Scholar
- Heidenreich PA, Keeffe B, McDonald KM, Hlatky MA: Overview of randomized trials of antiarrythmic drugs and devices for the prevention of sudden cardiac death. Am Heart J. 2002, 144: 422-430. 10.1067/mhj.2002.125499.View ArticlePubMedGoogle Scholar
- Seidl K, Senges J: Worldwide utilization of implantable cardioverter/defibrillators now and in the future. Card Electrophysiol Rev. 2003, 7: 5-13. 10.1023/A:1023603118860.View ArticlePubMedGoogle Scholar
- Nielsen JC, Mortensen PT, Pedersen AK: Long-term follow-up of patients treated with implantable cardioverter defibrillators in a Danish centre. Scand Cardiovasc J. 1997, 31: 151-156.View ArticlePubMedGoogle Scholar
- Greenberg D, Katz A, Epstein M, Golovchiner G, Ilia R, Caspi A: Implantable cardioverter defibrillators in Israel: utilization and implantation trends. Int J Cardiol. 2002, 82: 17-23. 10.1016/S0167-5273(01)00563-0.View ArticlePubMedGoogle Scholar
- Pfeiffer D, Jung W, Fehske W, Korte T, Manz M, Moosdorf R, Luderitz B: Complications of pacemaker-defibrillator devices: Diagnosis and management. Am Heart J. 1994, 127: 1073-1080. 10.1016/0002-8703(94)90090-6.View ArticlePubMedGoogle Scholar
- Trappe H-J, Wenzlaff P, Pfitzner P, Fieguth H-G: Long term follow up of patients with implantable cardioverter-defibrillators and mild, moderate, or severe impairment of left ventricular function. Heart. 1997, 78: 243-249.View ArticlePubMedPubMed CentralGoogle Scholar
- Palatianos GM, Thurer RJ, Cooper DK, Georgakakis GE, Tucker RL, Fernandez P, Interian A, Myerburg RJ: The implantable cardioverter-defibrillator: clinical results. Pacing Clin Electrophysiol. 1991, 14: 297-301.View ArticlePubMedGoogle Scholar
- Geelen P, Lorga Filho A, Primo J, Wellens F, Brugada P: Experience with implantable cardioverter defibrillator therapy in elderly patients. Eur Heart J. 1997, 18: 1339-1342.View ArticlePubMedGoogle Scholar
- Engelstein ED: Long QT syndrome: a preventable cause of sudden death in women. Curr Womens Health Rep. 2003, 3: 126-134.PubMedGoogle Scholar
- Higgins SL: Impact of the MADIT on indication trends. Am J Cardiol. 1999, 83: 79D-82D. 10.1016/S0002-9149(98)01007-8.View ArticlePubMedGoogle Scholar
- Rosenqvist M, Beyer T, Block M, den Dulk K, Minten J, Lindemans F: Adverse events with transvenous implantable cardioverter-defibrillators: A prospective multicenter study. Circulation. 1998, 98: 663-670.View ArticlePubMedGoogle Scholar
- Pavia S, Bruce W: The management of surgical complications of pacemaker and implantable cardioverter-defibrillators. Curr Opin Cardiol. 2001, 16: 66-71. 10.1097/00001573-200101000-00010.View ArticlePubMedGoogle Scholar
- Mela T, McGovern BA, Garan H, Vlahakes GJ, Torchiana DF, Ruskin J, Galvin JM: Long-term infection rates associated with the pectoral versus abdominal approach to cardioverter-defibrillator implants. Am J Cardiol. 2001, 88: 750-753. 10.1016/S0002-9149(01)01845-8.View ArticlePubMedGoogle Scholar
- Kron J, Herre J, Renfroe EG, Rizo-Patron C, Raitt M, Halperin B, Gold M, Goldner B, Wathen M, Wilkoff B, Olarte A, Yao Q: Lead- and device-related complications in the antiarrythmic versus implantable defibrillators trial. Am Heart J. 2001, 141: 92-88. 10.1067/mhj.2001.111261.View ArticlePubMedGoogle Scholar
- DiMarco JP: Implantable Cardioverter-Defibrillators. N Eng J Med. 2003, 349: 1836-1847. 10.1056/NEJMra035432.View ArticleGoogle Scholar
- McCready MJ, Exner DV: Quality of life and psychological impact of implantable cardioverter defibrillators: Focus on randomized controlled trial data. Card Electrophysiol Rev. 2003, 7: 63-70. 10.1023/A:1023699225221.View ArticlePubMedGoogle Scholar
- Dougherty CM: Psychological reactions and family adjustment in shock versus no shock groups after implantation of internal cardioverter defibrillator. Heart Lung. 1995, 24: 281-291. 10.1016/S0147-9563(05)80071-8.View ArticlePubMedGoogle Scholar
- Duru F, Klaghofer R, Mattmann H, Sensky T, Buddeberg C, Candinas R: How different from pacemaker patients are recipients of implantable cardioverter-defibrillators with respect to psychosocial adaption, affective disorders, and quality of life?. Heart. 2001, 85: 375-379. 10.1136/heart.85.4.375.View ArticlePubMedPubMed CentralGoogle Scholar
- Deisenhofer I, Kolb C, Ndrepepa G, Schreieck J, Karch M, Schmieder S, Zrenner B, Schmitt C: Do current dual chamber cardioverter defibrillators have advantages over conventional single chamber cardioverter defibrillators in reducing inappropriate therapies? A randomized, prospective study. J Cardiovasc Electrophysiol. 2001, 12: 134-142. 10.1046/j.1540-8167.2001.00134.x.View ArticlePubMedGoogle Scholar
- Kouakam C, Kacet S, Hazard JR, Ferraci A, Mansour H, Defaye P, Davy JM, Lambiez M, Ventak AV Investigators: Performance of a dual-chamber implantable defibrillator algorithm for discrimination of ventricular from supraventricular tachycardia. Europace. 2004, 6: 32-42. 10.1016/j.eupc.2003.09.007.View ArticlePubMedGoogle Scholar
- Quesada A, Almendral J, Arribas F, Ricci R, Wolpert C, Adragao P, Cobo E, Navarro X, DATAS Investigators: The DATAS rationale and design: a controlled, randomized trial to assess the clinical benefit of dual chamber (DDED) defibrillator. Europace. 2004, 6: 142-150. 10.1016/j.eupc.2003.11.011.View ArticlePubMedGoogle Scholar
- Klein RC, Raitt MH, Wilkoff BL, Beckman KJ, Coromilas J, Wyse DG, Friedman PL, Martins JB, Epstein AE, Hallstrom AP, Ledingham RB, Belco KM, Greene HL, AVID Investigators.: Analysis of implantable cardioverter defibrillator therapy in the Antiarrhythmics Versus Implantable Defibrillators (AVID) trial. J Cardiovasc Electrophysiol. 2003, 14: 940-948. 10.1046/j.1540-8167.2003.01554.x.View ArticlePubMedGoogle Scholar
- Grosse-Meininghaus D, Siebels J, Wolpert Ch, Schulte B, Vogt J, Korte T, Duckeck W, Langes K, German Ventritex MD-Investigators: Efficacy of antitachycardia pacing confirmed by stored electrograms. A retrospective analysis of 613 stored electrograms in implantable defibrillators. Z Kardiol. 2002, 91: 396-403. 10.1007/s00392-002-0780-8.View ArticlePubMedGoogle Scholar
- Nasir N, Pacifico A, Doyle TK, Earle NR, Hardage ML, Henry PD: Spontaneous ventricular tachycardia treated by antitachycardia pacing. Cadence Investigators. Am J Cardiol. 1997, 79: 820-822. 10.1016/S0002-9149(96)00881-8.View ArticlePubMedGoogle Scholar
- Trappe H-J, Achtelik M, Pfitzner P, Voight B, Weismüller P: Single-chamber versus dual-chamber implantable cardioverter defibrillators: Indications and clinical results. Am J Cardiol. 1999, 83: 8-16. 10.1016/S0002-9149(98)01037-6.View ArticleGoogle Scholar
- The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2261/6/22/prepub
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.