- Research article
- Open Access
- Open Peer Review
Wearable defibrillator use in heart failure (WIF): results of a prospective registry
BMC Cardiovascular Disorders volume 12, Article number: 123 (2012)
Heart failure (HF) patients have a high risk of death, and implantable cardioverter defibrillators (ICDs) are effective in preventing sudden cardiac death (SCD). However, a certain percentage of patients may not be immediate candidates for ICDs, particularly those having a short duration of risk or an uncertain amount of risk. This includes the newly diagnosed patients, as well as those on the cardiac transplant list or NYHA class IV heart failure patients who do not already have an ICD. In these patients, a wearable cardioverter defibrillator (WCD) may be used until long term risk of SCD is defined. The purpose of this study was to determine the incidence of SCD in this population, and the efficacy of early defibrillation by a WCD.
Ten enrolling centers identified 89 eligible HF patients who were either listed for cardiac transplantation, diagnosed with dilated cardiomyopathy, or receiving inotropic medications. Data collected included medical history, device records, and outcomes (including 90 day mortality).
Out of 89 patients, final data on 82 patients has been collected. Patients wore the device for 75±58 days. Mean age was 56.8±13.2, and 72% were male. Most patients (98.8%) were diagnosed with dilated cardiomyopathy with a low ejection fraction (<40%) and twelve were listed for cardiac transplantation. Four patients were on inotropes. There were no sudden cardiac arrests or deaths during the study. Interestingly, 41.5% of patients were much improved after WCD use, while 34.1% went on to receive an ICD.
In conclusion, the WCD monitored HF patients until further assessment of risk. The leading reasons for end of WCD use were improvement in left ventricular ejection fraction (LVEF) or ICD implantation if there was no significant improvement in LVEF.
Sudden cardiac death (SCD) is believed to account for 40 to 70% of the mortality associated with congestive heart failure (HF), with progressive pump failure causing the majority of the remaining fatalities [1, 2]. Current figures estimate that 20% of HF patients will die within one year of diagnosis , which represents an improvement over prior decades . Thus, the sudden death rate in the first year of diagnosis is about 10%, mostly resulting from terminal ventricular arrhythmias. Further complicating the situation, some medications used in the treatment of HF (ie, inotropes and diuretics) can also contribute to ventricular tachycardia (VT) in patients . Implantable cardioverter-defibrillators (ICDs) are effective in preventing SCD in HF due to ischemic and non-ischemic cardiomyopathy when cardiac function, as measured by ejection fraction (EF), is ≤ 35% [6–9]. HF patients on heart transplant lists have also been shown to benefit from ICDs [10–12]. However, ICDs may not be the most cost-effective , or the best method of reducing the incidence of sudden death in all HF patients, particularly in patients having a short (or uncertain) duration of risk or those with drug refractory NYHA class IV HF . According to AHA/ACC/HRS guidelines for ICDs, newly diagnosed nonischemic dilated cardiomyopathy (NIDCM) patients should be medically optimized, a process that may take months, before implantation . Conversely, wearable cardioverter defibrillators (WCD) such as the LifeVest® device (manufacturer: ZOLL, Pittsburgh, PA) can be used immediately to protect against SCD until long-term risk of sudden death is established or cardiac function improves.
The WCD (LifeVest®, ZOLL, Pittsburgh, PA) was FDA-approved in 2001 as an outpatient device after a clinical trial involving 285 out-of-hospital patients . WCDs are intended for use in adult patients that have an increased risk of SCD if an ICD is not warranted. Since approval, more than 60,000 patients have used the LifeVest® on an outpatient basis for a wide variety of conditions. Like ICDs, WCDs are designed to detect and treat ventricular tachyarrhythmias without the need for bystander intervention. Unlike ICDs, they are completely external to the body and do not require surgical intervention. Thus, they are ideal for shorter-term applications where the risk of SCA is changing or uncertain. WCD use in the HF population is limited [16, 17]. In order to gain further clinical experience with WCD use in this population, ZOLL sponsored the WIF (Wearable defibrillator use In heart Failure) registry. The primary purpose of the WIF registry was to determine the incidence of SCA in this population, the cause of SCA (bradyarrhythmias or tachyarrhythmias), and the efficacy of early defibrillation given by the WCD. Three patient groups were followed; patients listed for heart transplantation, patients with dilated cardiomyopathy (DCM) and patients using inotropes known to cause ventricular tachyarrhythmias.
The WIF study was designed to collect SCA events, WCD defibrillation efficacy, and WCD usage data in HF patients. The WCD usage criteria was defined as met if median daily use was at least 85% of the day in those patients that wore the device for at least 7 days. WCD prescription length was not prespecified, but was anticipated to be 3 months. The WCD (LifeVest®, ZOLL, Pittsburgh, PA) is FDA approved for use in this population, and is reimbursed by Medicare and other insurers. The study was sponsored by the manufacturer (ZOLL, Pittsburgh, PA).
Ten centers with WCD prescribing physicians were selected for the study based upon expressed interest. All sites were in the US and included both academic and community centers.
Patients were enrolled from July 1, 2007 through February 1, 2010. For the purposes of this registry, HF patients were screened if they were: 1) listed (or being considered) for heart transplantation, and/or, 2) they had DCM (with VT or EF ≤ 40%), and/or, 3) the patient was receiving inotropes. HF patients were excluded from the study if they had an active ICD or if they were impaired such that they could not use the device. Age, pregnancy, and time since HF diagnosis were not exclusion criteria. The device’s default VT and VF detection rates were set at 150 and 200 beats per minute, respectively, and the default shock energy was set at 150 joules. However, physicians had the ability to change these settings. A sample size of 500 was originally specified in order to capture multiple SCA events based on an estimated event rate of one or two percent over the three month study period. However, due to lower than expected site participation the registry was terminated after three years.
The protocol, informed consent document, and relevant supporting information was submitted and approved for each study site by their local institutional review board, for Saint Luke’s Hospital of Kansas City, Albert Einstein Health Care Network, and The University of Iowa, or a national institutional review board (Schulman Associates) before the study was initiated in accordance with ethical principles that have their origin in the Declaration of Helsinki and consistent with Good Clinical Practice and applicable regulatory requirements. Patient demographics, limited medical history, medications, laboratory results, and cardiac testing were collected upon entry. WCD device records were interrogated to determine compliance with use, defibrillation events, and arrhythmia detection. WCD ECG recordings, whether patient initiated or automatic, were also collected. Reasons for WCD device removal, cardiac events during the study, medications, and cardiac testing data were collected at the end of WCD use. Patient survival during WCD use was determined three months after the patient began participation. The social security death index (SSDI) was used to determine all cause-mortality after WCD use. The duration of follow-up was computed from the time of WCD start until death for those who died, and to the final date of data collection (June 9, 2010) for those who were still alive.
Descriptive statistics were utilized to describe this population based on data collected at the time of referral for WCD therapy or after therapy. All values are reported as means ± standard deviations. Student’s t test (unpaired) for continuous data, or Fisher’s exact test for categorical data, was used where a p value of 0.05 was considered statistically significant. The Pearson’s product–moment correlation coefficient was calculated using the free online academic software at http://www.wessa.net to test whether a correlation existed between factors impacting WCD use.
The WIF registry began enrolling patients in mid July 2007. At the conclusion of the study in May 2010, 89 patients had been enrolled from ten centers. This report is based on the analysis of data from 82 patients who completed the study. Of the seven who did not complete the study, 4 were lost to study follow-up after completing WCD use, and 3 dropped out after wearing the WCD for a couple of hours. The cohort was predominantly male (72%) with a mean age of 56.8 ± 13.2 years (range: 25–82). Most patients were diagnosed with ischemic or non-ischemic dilated cardiomyopathy (DCM) with a low ejection fraction (EF) (98.8%), and twelve were listed for cardiac transplantation. Four patients were on inotropes, and three of these patients were also listed for transplantation. Baseline characteristics are shown in Table 1. The majority of patients had NYHA class II- III HF, and only one patient started out with an EF >45% (transplant rejection with VT). The etiology of the cardiomyopathy was idiopathic in 39% of patients, ischemic in 36% of patients, and a variety of other non-ischemic causes (Table 2). Co-morbidities included hypertension (68%), diabetes (44%), smoking (56%), and obesity (42%). Seventy-eight patients (95.1%) were on beta-blockers, 63 (76.8%) on ACE inhibitors, 16 (19.5%) on angiotensin II receptor blockers, and seven (8.5%) on amiodarone. Only 5 (6.1%) patients had a prior history of sustained ventricular dysrhythmias. Time since first HF diagnosis was not collected.
Compliance with the WCD
The average period of WCD usage for all patients was 79.5±57.8 days (median: 79, range: 1–277). Two patients were still wearing the device at the end of the study. Compliance with wearing the WCD was then calculated based on patients who wore the device for 7 days or greater (n=75). The average daily device use was 19.5±4.6 hr/day (median: 21.8; range: 3.7-23.7) over an average of 75.1±57.7 days (median: 64; range: 7–277). Average daily use was significantly correlated to total days worn (r=0.300, p<0.01).
There were no SCA events or deaths during the study, and 90- day survival after WCD fitting was 100%. There were also no adverse events or inappropriate shocks by the WCD. Two patients (2.7%) complained of palpitations, and five patients (6.8%) complained of lightheadedness or fainting during WCD use, with no sustained ventricular arrhythmias or asystoles detected by the WCD. Three patients had a CABG procedure during WCD use but none had percutaneous coronary intervention. Medication use at the end of the study was similar to baseline, with 95.7% on beta-blockers, 72.5% on ACE inhibitors, 20.3% on angiotensin II receptor blockers, and 15.9% on amiodarone. Four patients underwent an electrophysiology (EP) study after discharge. One patient had an inconclusive EP result, one had a positive result, and two had a negative test. The first two patients then underwent ICD implantation.
At the time of WCD discontinuation, 34 (41.5%) patients were considered much improved and no longer needed the WCD due to either an improved EF (defined as EF ≥ 35% at the time use ended) (31 out of 34), acute allograft rejection resolved (1 of 34), feeling better (1 of 34), and one unknown reason. Twenty-eight (34.1%) went on to receive an ICD implant. Six (7.3%) refused to wear the device due to discomfort or other reasons, seven (8.5%) patients had an unknown/other reason for removing their WCD. Two (2.4%) transitioned to “do not resuscitate” status, one (1.2%) received a heart transplant, one (1.2%) had aortic/mitral/tricuspid valve repair, one (1.2%) had frequent inappropriate detections, and two (2.4%) patients were still using the WCD at the close of the study (Figure 1). Although a greater proportion of presumed non-ischemic patients ended WCD use due to improvement as compared to ischemic patients (50% vs. 34%), it was not statistically different (p=0.267). Likewise, 32% of non-ischemic patients received ICD after WCD use vs. 45% of ischemic patients.
Improved LVEF during WCD use
The average final EF for all patients was 37.2%±14.6% (n=70), with an average EF increase per patient of 13.5%±15.7%. This was a statistically significant improvement of EF since initial enrollment (p<0.001). Fifteen patients had an EF >50% at the end of WCD use. Baseline EF versus final EF per patient is shown in Figure 2(a,b). Those patients with lower baseline EF experienced the greatest improvement in EF (Figure 2b). Only one patient with a high baseline EF was enrolled by virtue of transplant rejection with VT. The average length of prescription to ICD implant was 77±55 days (median:76), while the average length of time between prescription to EF improvement was 82±58 days (median:85). Figure 2c shows change in EF according to outcome (improved, received ICD, or other). The average final EF before ICD implant was 25.5%±9.5% (n=20). The final EF of those patients who improved was 46.4%±9.8% compared to a baseline of 25.5% ± 11.1% (n=33, p<0.0001). Those experiencing EF improvement had an average 20.9% (or 1.8 fold) increase in their EF% from baseline. There was no difference in baseline EF between those who improved and those who did not. New York Heart Association class also improved by 0.5±1.0 points since initial enrollment (n=59, p<0.005). There were no significant differences in beta-blocker, ACE-Inhibitor, or angiotensin-II receptor blocker usage between those patients who improved, and those who received an ICD (data not shown). After WCD use, six patients (7.3%) died from unknown causes 26, 56, 229, 254, 304, and 553 days (average 288±176 days) after WCD discontinuation, according to the social security death index. The patient who died 26 days after WCD use received an ICD (EF=27.5%). The patients who died at 56 and 229 days post- WCD use had an EF improvement (EF= 43% and 50%, respectively at the time of WCD end of use). The patient who died 304 days after WCD use had stopped the WCD due to discomfort. The patient who died 553 days after WCD ended use was due to a lack of improvement (EF=31%) and switch to a “do not resuscitate” status.
Patients on transplant list and inotropes
Twelve patients on the heart transplant list completed the study. Four patients were able to remove the WCD after their condition improved, four received ICDs, one had aortic/mitral/tricuspid valve repair, one received a heart transplant, and two ended WCD use for an unknown reasons. Ninety day survival was 100%, although one patient died 56 days after WCD discontinuation for improved EF, according to the social security death index. Of the four patients on inotropes; two received ICDs, one received a heart transplant, and one had an improvement in EF but died 56 days after WCD discontinuation (same patient as above).
The intent of the WIF registry was to observe the use of the WCD in a population of HF patients over a 90 day period. The enrolled group represented a mix of dilated cardiomyopathy patients (Tables 1 and 2), with the majority of patients having an idiopathic (39%) or ischemic (36%) etiology of HF. Twelve patients (14.6%) were listed for cardiac transplantation, and four (4.9%) were on inotropes. As one may expect from the small sample size that resulted from lower than anticipated enrollment, there were no SCA events in the 82 patients who completed the study. Current WCD event statistics suggest an event rate for new cardiomyopathy patients wearing the WCD to be approximately 0.00013 VT/VF events per patient per day . Therefore, the chance of not having an event was approximately 42%. Regardless, the WCD bridged the DCM patients until long-term therapy decisions could be made. There were also no adverse events or unnecessary shocks by the WCD during the study. Daily compliance with the WCD was a median of 21.8 hrs, which met our goal of at least 85% median daily wear time. This compliance is very similar to that of the current general commercial population (median=21.7 hr/day) . Seventy percent of patients wore the device for average ≥ 80% of the day. Although assessing and comparing compliance is difficult, compliance with the WCD is similar to compliance with medications in HF populations .
Interestingly, 34 (41.5%) patients were considered much improved at the end of the study and no longer needed the WCD, mostly due to an improved EF (EF ≥ 35%). This may have been due to the high rate of beta-blocker use, or possibly because some patients had reversible causes of disease. However, 33% of the 12 transplant-listed patients also showed improvement. To our knowledge, few recent peer-reviewed studies have analyzed improvement in ventricular function 60–90 days post-diagnosis or acute decompensation of DCM [20, 21]. The IMAC trial, which studied intravenous immune globulin in recent onset DCM, also saw mean LVEF increase significantly in patients from 25% to 41% over a 6 month period regardless of treatment . It should also be emphasized that improvement in LV function should not be taken as a “cure” and medications should be continued or monitored closely, as relapses may occur . Although we do not know the cause of death, two patients who were discontinued from the WCD due to EF improvement (EF=43% and 50%) died (56 and 229 days) after removing the WCD. There was an overall 7.3% mortality rate over 288±176 days after being fit with the WCD. Unfortunately, this time period was beyond the scope of the study and the reasons for death are unknown. The mortality rate is similar to that seen in the SCD-HeFT and DEFINITE (non-ischemic) trials [7, 9].
The benefits of prophylactic ICD implantation remains contentious in DCM patients . Although ICDs have been shown to be beneficial in this group of patients [7, 8], it is not without risk, and the event rate remains low (especially for those without prior MI) . For this reason, risk stratification of these patients remains extremely important. A recent study showed a yearly increase in out-of-hospital mortality rates in Medicare HF patients, possibly due to shortened hospital stays . It may be beneficial to the patient (and society) to give the patient enough time to potentially stabilize on medication and recover LV function, while still keeping them protected and out of the hospital. It may also be beneficial to avoid ICD placement in the transplant-listed patient, if their wait is not expected to be extremely long. These goals can be accomplished with the WCD.
There were several limitations to this study. First, due to the smaller than expected enrollment, the study did not meet its primary objective of collecting SCA incidence data. Also, 8.5% of those who were initially enrolled were lost to study follow-up. Second, factors affecting EF improvement could not be analyzed fully due to the heterogeneic etiologies and unknown duration of disease. Finally, the mortality rate in our study was determined using the Social Security Death Index and may not have captured all deaths in our population, leading to erroneous conclusions based on this data
In conclusion, the WCD monitored HF patients until further assessment of risk. The leading reasons for end of WCD use were improvement in left ventricular ejection fraction (LVEF) or ICD implantation if there was no significant improvement in LVSF. Due to the smaller than anticipated sample size, there were no SCD events in the 82 patients who completed the study (≤90 days). The overall 7.3% mortality rate over 288±176 days after being fit with the WCD is similar to that seen in other trials.
Disclosures of financial support
This research was sponsored by ZOLL Cardiac Management Solutions, Pittsburgh, PA.
Goldman S, Johnson G, Cohn JN, Cintron G, Smith R, Francis G: Mechanism of death in heart failure. The vasodilator-heart failure trials. The V-HeFT VA cooperative studies group. Circulation. 1993, 87: VI24-31.
Nagele H, Rodiger W: Sudden death and tailored medical therapy in elective candidates for heart transplantation. J Heart Lung Transplant. 1999, 18: 869-876. 10.1016/S1053-2498(99)00040-6.
Lloyd-Jones D, Adams RJ, Brown TM, Carnethon M, Dai S, De Simone G, Ferguson TB, Ford E, Furie K, Gillespie C, et al: Heart disease and stroke statistics–2010 update: a report from the American Heart Association. Circulation. 2010, 121: e46-e215.
Levy D, Kenchaiah S, Larson MG, Benjamin EJ, Kupka MJ, Ho KK, Murabito JM, Vasan RS: Long-term trends in the incidence of and survival with heart failure. N Engl J Med. 2002, 347: 1397-1402. 10.1056/NEJMoa020265.
Packer M: Sudden unexpected death in patients with congestive heart failure: a second frontier. Circulation. 1985, 72: 681-685. 10.1161/01.CIR.72.4.681.
Moss AJ, Zareba W, Hall WJ, Klein H, Wilber DJ, Cannom DS, Daubert JP, Higgins SL, Brown MW, Andrews ML: Prophylactic implantation of a defibrillator in patients with myocardial infarction and reduced ejection fraction. N Engl J Med. 2002, 346: 877-883. 10.1056/NEJMoa013474.
Bardy GH, Lee KL, Mark DB, Poole JE, Packer DL, Boineau R, Domanski M, Troutman C, Anderson J, Johnson G, et al: Amiodarone or an implantable cardioverter-defibrillator for congestive heart failure. N Engl J Med. 2005, 352: 225-237. 10.1056/NEJMoa043399.
Packer DL, Prutkin JM, Hellkamp AS, Mitchell LB, Bernstein RC, Wood F, Boehmer JP, Carlson MD, Frantz RP, McNulty SE, et al: Impact of implantable cardioverter-defibrillator, amiodarone, and placebo on the mode of death in stable patients with heart failure: analysis from the sudden cardiac death in heart failure trial. Circulation. 2009, 120: 2170-2176. 10.1161/CIRCULATIONAHA.109.853689.
Kadish A, Dyer A, Daubert JP, Quigg R, Estes NA, Anderson KP, Calkins H, Hoch D, Goldberger J, Shalaby A, et al: Prophylactic defibrillator implantation in patients with nonischemic dilated cardiomyopathy. N Engl J Med. 2004, 350: 2151-2158. 10.1056/NEJMoa033088.
Saba S, Atiga WL, Barrington W, Ganz LI, Kormos RL, MacGowan GA, Mathier MA, McNamara DM, Obioha-Ngwu O, Murali S: Selected patients listed for cardiac transplantation may benefit from defibrillator implantation regardless of an established indication. J Heart Lung Transplant. 2003, 22: 411-418. 10.1016/S1053-2498(02)00573-9.
Sandner SE, Wieselthaler G, Zuckermann A, Taghavi S, Schmidinger H, Pacher R, Ploner M, Laufer G, Wolner E, Grimm M: Survival benefit of the implantable cardioverter-defibrillator in patients on the waiting list for cardiac transplantation. Circulation. 2001, 104: I171-176.
Ermis C, Zadeii G, Zhu AX, Fabian W, Collins J, Lurie KG, Sakaguchi S, Benditt DG: Improved survival of cardiac transplantation candidates with implantable cardioverter defibrillator therapy: role of beta-blocker or amiodarone treatment. J Cardiovasc Electrophysiol. 2003, 14: 578-583. 10.1046/j.1540-8167.2003.02590.x.
Chen L, Hay JW: Cost-effectiveness of primary implanted cardioverter defibrillator for sudden death prevention in congestive heart failure. Cardiovasc Drugs Ther. 2004, 18: 161-170.
Stevenson LW, Desai AS: Selecting patients for discussion of the ICD as primary prevention for sudden death in heart failure. J Card Fail. 2006, 12: 407-412. 10.1016/j.cardfail.2006.06.001.
Centers for Medicare & Medicaid Services: Coverage determinations: implantable automatic defibrillators. Medicare National Coverage Determinations Manual. 2010, Baltimore, Md. CMS-Pub, 100-3; Pt 1, section 20.4. Available at: http://www.cms.hhs.gov/medicare-coverage-database/details/ncd-details.aspx?NCDId=110&ncdver=3&bc=AgAAQAAAAAAA&. Accessed December 23
Feldman AM, Klein H, Tchou P, Murali S, Hall WJ, Mancini D, Boehmer J, Harvey M, Heilman MS, Szymkiewicz SJ, Moss AJ: Use of a wearable defibrillator in terminating tachyarrhythmias in patients at high risk for sudden death: results of the WEARIT/BIROAD. Pacing Clin Electrophysiol. 2004, 27: 4-9. 10.1111/j.1540-8159.2004.00378.x.
Klein HU, Meltendorf U, Reek S, Smid J, Kuss S, Cygankiewicz I, Jons C, Szymkiewicz S, Buhtz F, Wollbrueck A, et al: Bridging a temporary high risk of sudden arrhythmic death. Experience with the wearable cardioverter defibrillator (WCD). Pacing Clin Electrophysiol. 2010, 33: 353-367. 10.1111/j.1540-8159.2009.02590.x.
Chung MK, Szymkiewicz SJ, Shao M, Zishiri E, Niebauer MJ, Lindsay BD, Tchou PJ: Aggregate national experience with the wearable cardioverter-defibrillator: event rates, compliance, and survival. J Am Coll Cardiol. 2010, 56: 194-203. 10.1016/j.jacc.2010.04.016.
Brown SB, Shannon RP: Improving medication compliance in patients with heart failure. Am J Cardiol. 2008, 101: 274-277. 10.1016/j.amjcard.2007.07.073.
Binkley PF, Lesinski A, Ferguson JP, Hatton PS, Yamokoski L, Hardikar S, Cooke GE, Leier CV: Recovery of normal ventricular function in patients with dilated cardiomyopathy: predictors of an increasingly prevalent clinical event. Am Heart J. 2008, 155: 69-74. 10.1016/j.ahj.2007.08.010.
Moon J, Ko YG, Chung N, Ha JW, Kang SM, Choi EY, Rim SJ: Recovery and recurrence of left ventricular systolic dysfunction in patients with idiopathic dilated cardiomyopathy. Can J Cardiol. 2009, 25: e147-150. 10.1016/S0828-282X(09)70497-0.
McNamara DM, Holubkov R, Starling RC, Dec GW, Loh E, Torre-Amione G, Gass A, Janosko K, Tokarczyk T, Kessler P, et al: Controlled trial of intravenous immune globulin in recent-onset dilated cardiomyopathy. Circulation. 2001, 103: 2254-2259. 10.1161/01.CIR.103.18.2254.
Jessup M, Abraham WT, Casey DE, Feldman AM, Francis GS, Ganiats TG, Konstam MA, Mancini DM, Rahko PS, Silver MA, et al: 2009 focused update: ACCF/AHA Guidelines for the Diagnosis and Management of Heart Failure in Adults: a report of the American College of Cardiology Foundation/American Heart Association Task Force on Practice Guidelines: developed in collaboration with the International Society for Heart and Lung Transplantation. Circulation. 2009, 119: 1977-2016.
Bueno H, Ross JS, Wang Y, Chen J, Vidan MT, Normand SL, Curtis JP, Drye EE, Lichtman JH, Keenan PS, et al: Trends in length of stay and short-term outcomes among medicare patients hospitalized for heart failure, 1993–2006. JAMA. 2010, 303: 2141-2147. 10.1001/jama.2010.748.
The pre-publication history for this paper can be accessed here:http://www.biomedcentral.com/1471-2261/12/123/prepub
We would like to thank all the investigators and centers that participated in the WIF study: Andrew C. Kao, M.D., St. Luke’s Hospital, Kansas City, MO; Steven W. Krause, D.O., Blessing Hospital, Quincy, IL; Rajiv Handa M.D., Metro Heart Group of St. Louis, St. Louis; Darshak Karia, M.D., Albert Einstein Medical Center, Philadelphia; Guillermo Reyes, M.D., Cardiology of San Antonio, San Antonio; Frances Johnson, M.D., University of Iowa, Iowa City; Alvin Cacho, M.D. San Joaquin Cardiology Medical, Lodi, CA; Lewis Horvitz, M.D., Hamilton Cardiology Associates, Trenton, NJ; Mohammad Mortada, M.D., Comprehensive Cardiovascular Care of Wisconsin, Milwaukee, WI.
NB and SS are employees of ZOLL. ZOLL sponsored this study, including the article-processing charge of this manuscript, and contributed to study design, implementation, and data monitoring.
AK and SS contributed to study design. All authors contributed to the acquisition of data and analysis of the results. AK, DK, NB, and SS were involved in drafting the manuscript. All authors revised the manuscript critically and approved the final version.
About this article
Cite this article
Kao, A.C., Krause, S.W., Handa, R. et al. Wearable defibrillator use in heart failure (WIF): results of a prospective registry. BMC Cardiovasc Disord 12, 123 (2012). https://doi.org/10.1186/1471-2261-12-123
- Heart failure
- Wearable cardioverter defibrillator
- Sudden cardiac death